MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, AURANGABAD. (M.S.)


 Martina Parks
 10 months ago
 Views:
Transcription
1 MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, AURANGABAD. (M.S.) DEPARTMENT OF CIVIL ENGINEERING FLUID MECHANICS I LAB MANUAL Prepared By Prof. L. K. Kokate Lab Incharge Approved By Dr. S. B. Shinde H.O.D. Civil JNEC CIVIL/FMI/LKK/Aug 2014 Page 1
2 Vision of Civil Department The department of Civil Engineering strives to produce qualified engineers, researchers and professionals to serve the society with sustainable development. Mission of Civil Department To provide quality education and prepare competitive graduates for successful career in Civil Engineering. To develop research opportunities that creates competent professionals who are trained in the design and development of environment friendly Civil Engineering system. Program Educational Objectives I. Graduates of Civil Engineering Program will be prepared to take the challenges in the field of Civil Engineering. II. To provide Graduates with a sound Knowledge in mathematical, scientific and Civil Engineering fundamentals required to solve engineering problems and also to pursue higher studies. III. To train students with good scientific and engineering breadth in Construction industry & many field of Civil Engineering. IV. To build the confidence of students leading to professional and ethical integrity, effective communication skill, leadership, so that they can apply engineering knowledge for betterment of society. V. To provide a good competitive learning environment so that graduates of Civil Engineering will be ready to meet the needs of Indian and multinational construction industries. The students are able to demonstrate: Program specific outcomes 1. The knowledge of planning and designing of the system components for building planning, transportation, water resources, estimating, costing and scheduling the construction processes. 2. The fundamental knowledge of analysis and design of various structures with an understanding of associated safety, quality and economy. 3. The knowledge of field data collection and material characterization to provide constructive and creative engineering solutions that reflect social and environmental sensitivities. JNEC CIVIL/FMI/LKK/Aug 2014 Page 2
3 FLUID MECHANICS I EXPERIMENTS CLASS:  Second Year Civil Engineering LIST OF EXPERIMENTS Sr. No. Name of Experiment Page No. From To I Study of pressure measuring devices. II Determination of meta centric height. 3 5 III Calibration of Bernoulli s equation. 6 8 IV Calibration of Venturimeter V Determination of Hydraulic coefficient for orifices VI Determination of coefficient of discharge for mouthpiece VII Calibration of Rectangular notch VIII Calibration of Triangular Notch IX Study of electrical analogy method for plotting of flow nets. Time Allotted for each Practical Session = 02 Hrs. JNEC CIVIL/FMI/LKK/Aug 2014 Page 3
4 EXPERIMENT NO: II  To Determine the Metacentric Height of a Cargo / War Ship JNEC CIVIL/FMI/LKK/Aug 2014 Page 4
5 AIM: To Determine the Metacentric Height of a Cargo / War Ship INTRODUCTION: Metacenter is defined as, the point about which the body starts oscillating when it is tilted (inclined) by a small angle. Metacenter may also be defined as, the point at which the line of action of force of buoyancy will meet the normal axis of the body when the body is given a small angular displacement. Metacentric Height is defined as, the distance between the Metacenter of a floating body & center of gravity. DESCRIPTION: The ship model is approximately 37 cm size square in plan and is about 23 cm high. The model is floated on water. The ship is tilted by moving a small weight at the level of the deck of the ship. To note down the tilt of the ship, a plumb is provided which records the tilt on a graduated arc of a circle. An arrangement is made to load the ship as a War ship or Cargo ship. PROCEDURE: Sr. No. For Cargo Ship For War Ship 1 Place suitable symmetrical weights at the bottom of the ship and load it as a Cargo Ship. Place suitable symmetrical weights at the deck level of the ship and load it as a War Ship. 2 Float the ship on the water. Float the ship on the water. 3 Adjust the balancing weights on both the sides of the ship so that the Plumb indicates zero reading on the graduated arc. 4 Keep the Moving (Hanging) Load/Weight at a distance of 3.5 cm off the centre on left side. Adjust the balancing weights on both the sides of the ship so that the Plumb indicates zero reading on the graduated arc. Keep the Moving (Hanging) Load/Weight at a distance of 3.5 cm off the centre on left side. 5 Note down the tilt of the ship in degrees. Note down the tilt of the ship in degrees. 6 Go on shifting the Hanging Load towards left & note down the distance of the centre, & tilt of the ship. 7 Repeat the procedure by shifting the load on the right hand side of the centre. Go on shifting the Hanging Load towards left & note down the distance of the centre, & tilt of the ship. Repeat the procedure by shifting the load on the right hand side of the centre. JNEC CIVIL/FMI/LKK/Aug 2014 Page 5
6 OBSERVATION : W 1 = Weight of the ship including balancing weight in grams.= 6200 gms W 2 = Total weight added to make it as a Cargo / War Ship.= 500 X 2 = 100 gms W 3 = Weight of the Hanging Load in grams.= 500 gms OBSERVATION TABLE: Sr. No Distance off the centre to the left X in cms Tilt of the Ship θ in degrees Metacentric Height=MG 1 in cms. Distance off the centre to the left X in cms Tilt of the Ship θ in degrees Metacentric Height=MG 2 in cms Average MG in cms SPECIMEN CALCULATIONS: W = (w 1 + w 2 ) in grams. MG 1 or MG 2 = Metacentric Heights in centimeters. = (W1 x X) / (W x tan θ 0 ) Average MG = (MG 1 + MG 2 ) / 2 Metacentric Height of a Cargo Ship (MG c ) =..cms. Metacentric Height of a War Ship (MGw) =..cms. RESULT: CONCLUSION: As the angle of tilt (θ 0 ) increases, Metacentric Height (MG or GM) also increases / decreases. JNEC CIVIL/FMI/LKK/Aug 2014 Page 6
7 EXPERIMENT NO: III  to Verify Bernoulli s Theorem JNEC CIVIL/FMI/LKK/Aug 2014 Page 7
8 AIM: To verify the Bernoulli s theorem. APPARATUS: Bernoulli s Set Up, Stop Watch, & Meter Scale. DESCRIPTION: Bernoulli s Theorem states that, in steady, ideal flow of an in compressible fluid, the total energy at any point of the fluid is constant. The total energy consists of Pressure Energy, Kinetic Energy, & Potential Energy (Datum Energy). The energy per unit weight of the fluid is Pressure Energy. Therefore, Pressure Energy = P / ρg Kinetic Energy = V 2 / 2g & Datum Energy = Z The applications of Bernoulli s theorem are: 1) Venturi Meter 2) Orifice Meter 3) Pilot Tube Description: The equipment is designed as a self sufficient unit; it has a sump tank, measuring tank, & 0.5 HP monoblock pump for water circulation. The apparatus consists of Supply Tank & Delivery Tank, which are connected to a Perspex flow channel. The channel tapers for a length of 25 cm & then piezometer tubes are fixed at a distance of 5 cm, centre to centre for measurement of pressure head. PROCEDURE: 1. Keep the bypass valve open & start the pump & slowly start closing the valve. 2. The water shall start flowing through the flow channel. The level in the piezometer tubes shall start rising. 3. Open the valve at the delivery tank side, & adjust the head in piezometer tubes to a steady position. 4. Measure the heads at all the points and also discharge with the help of Diversion Pan in the measuring tank. 5. Change the discharge & repeat the procedure. 6. Do the necessary calculations using the readings noted down before. JNEC CIVIL/FMI/LKK/Aug 2014 Page 8
9 SPECIFICATIONS: Tube No. C/S Area OBSERVATION TABLE: SPECIMEN CALCULATIONS: RESULT: 1) At discharge..liters / second, Total head is..centimeters. 2) At discharge..liters / second, Total head is..centimeters. JNEC CIVIL/FMI/LKK/Aug 2014 Page 9
10 EXPERIMENT NO: IV  FLOW THROUGH VENTURIMETER JNEC CIVIL/FMI/LKK/Aug 2014 Page 10
11 AIM: To determine the coefficient (K) of the Venturimeter. DESCRIPTION: Venturimeter is a device, used to measure the discharge of any liquid flowing through a pipe line. The pressure difference between the inlet and the throat of the Venturimeter is recorded using a mercury differential manometer, and the time is recorded for a measured discharge. Venturimeters are used to measure the flow rate of fluid in a pipe. It consists of a short length of pipe tapering to a narrow throat in the middle and then diverging gradually due to the reduced area and hence there is a pressure drop. By measuring the pressure drop with a manometer, the flow rate can be calculated by applying Bernoulli s equation. The meters are fitted in the piping system with sufficiently long pipe lengths (greater than 10 mm diameter) upstream of the meters. Each pipe has the respective Venturimeter with quick action cocks for pressure tappings. These pressure tappings are connected to a common middle chamber, which in turn is connected to a differential manometer. Each pipe line is provided with a flow control water is collected in an M.S. collecting tank of cross sectional are 0.4 m x 0.4 m provided with gauge scale fitting and drain valve. PROCEDURE: 1. The diameters of the inlet and throat are recorded and the internal plan dimensions of the collecting tank are measured. 2. Keeping the outlet valve closed, the inlet valve is opened fully. 3. The outlet vale is opened slightly and the manometric heads in both the limbs (h 1 and h 2 ) are noted. 4. The outlet valve of the collecting tank is closed tightly and the time t required for H rise of water in the collecting tank is observed using a stop watch. 5. The above procedure is repeated by gradually increasing the flow and observing the required readings. 6. The observations are tabulated and the coefficient of the Venturimeter is computed. Constant of Venturimeter, K = Where, a 1 = area of inlet a 2 = area of throat h = Venturi head in terms of flowing liquid = h 1 = Manometric head in one limb of the manometer h 2 = Manometric head in other limb of the manometer JNEC CIVIL/FMI/LKK/Aug 2014 Page 11
12 S m = Specific gravity of following liquid S 1 = Specific gravity of following liquid g = Acceleration due to gravity Actual Discharge (Q a ) = OBSERVATIONS: Diameter of inlet, d 1 =.mm Diameter of inlet, d 2 =.mm Internal plan dimensions of collecting tank Length, l =.mm Breadth, b =.mm OBSERVATION TABLE: Sr. No. Manometric Readings (mm) of Water H h 2 Difference X=(h 1 h 2 ) Venturi head in terms of flowing fluid (h) mm h Time for H =100mm rise t Sec. Trials Avg. Actual Discharge (mm 3 /sec) Coefficient of Venturimeter Mean Value of C d =. JNEC CIVIL/FMI/LKK/Aug 2014 Page 12
13 SPECIMEN CALCULATIONS: Area of inlet of Venturimeter a 1 = 2 πd 1 /4 ( mm 2 ) Area of throat of Venturimeter a 2 = 2 πd 2 /4 ( mm 2 ) Internal plan area of collecting tank = l x b (mm 2 ) Actual discharge, Q a =??? (mm 3 /s) Coefficient of Meter, (K) = Q a / C. h GRAPH: Q a vs. h h on Xaxis RESULT: Average Coefficient of the Venturimeter, C d = JNEC CIVIL/FMI/LKK/Aug 2014 Page 13
14 EXPERIMENT NO: IV DETERMINATION OF HYDRAULIC COEFFICIENT FOR ORIFICES JNEC CIVIL/FMI/LKK/Aug 2014 Page 14
15 AIM: The objective of this experiment is to find and study the variations of hydraulic coefficients under different operating conditions of the given orifice. APPARATUS: orifice apparatus, measuring tanks, a stopwatch & sliding hook gauge to measure the coordinates of the moving jet. DISCRIPTION: Orifice is a at sharp edged small circular hole fitted in one side of a reservoir containing fluid. It may be classified on the basis of their size, shape, upstream edge and the discharge conditions. Most commonly used are circular and rectangular orifices. It is used to determine the discharge through a tank. The thickness of the wall is assumed to be small compared to the diameter of the orifice. Because of the convergence of the streamlines approaching the orifice, the cross section of the jet decreases slightly until the pressure is equalized over the crosssection, and the velocity profile is nearly rectangular. This point of minimum area is called the vena contracta. Beyond the vena contracta, friction with the Fluid outside the jet (air) slows it down, and the cross section increases. This divergence is usually quite small, and the jet is nearly cylindrical with a Constant velocity. The jet is held together by surface tension. The ratio of the area of vena contracta to the orifice area is called the coefficient of contraction. The theoretical discharge through the orifice will be, where, a 0 = area of the orifice (cm 2 ) h = Constant Head (cm) g = acceleration due to gravity, 981 (cm/ sec 2 ) Q th = a 0 x (2 g h) The actual discharge is less than the theoretical discharge due to friction loss and the loss due to expansion of the jet. The actual discharge through the orifice can be determined using the collecting tank and stopwatch setup. Qac = (A x R) / t where, A = area of the collecting tank (cm 2 ). R = height difference of the water column in the piezometer, (cm) t = time taken to rise R meters, (sec) The coefficient of discharge CD is defined as the ratio of the actual discharge to theoretical discharge. The actual velocity of the jet through the orifice is found out by measuring the coordinates of the moving jet and applying Newton s laws. Refer figure below. The coordinates x and y of any point on the jet are measured using the sliding hook gauge. JNEC CIVIL/FMI/LKK/Aug 2014 Page 15
16 The coefficient of velocity C v :????????? The coefficient of contraction C C :??????? PROCEDURE: 1. Note the dimensions of the discharge measuring tank, orifice. 2. Check that the zero of the scale of the inlet tank is the same level as the Center line of the mouthpiece or orifice. If not, measure the difference in elevation and take it as zero error. 3. Adjust the opening of the inlet valve till the water level in the supply tank become steady. 4. Note down the head. 5. using the hook gauge arrangement measure the coordinates of the jet in a convenient point. 6. Using collecting tank and stop watch setup measure the actual discharge. 7. Repeat the experiment for different inlet valve openings and tabulate the readings. 8. Plot the characteristics CD Vs h, CC Vs h and CV Vs h. JNEC CIVIL/FMI/LKK/Aug 2014 Page 16
17 OBSERVATIONS : 1. Diameter of Orifice = 0.8 cm 2. Dimension of measuring tank= 30cm x 40 cm 4. Area of measuring tank (A)= 5. Area of Orifice (a o )= OBSERVATION TABLE : Sr No H (cm) X (cm) Y (cm) I. R. (cm) Measuring Tank Reading F. R. (cm) Diff 'R ' (cm) t (sec) Qact (cm 3 /sec ) Qth (cm 3 /sec ) C d C v C c SPECIMEN CALCULATIONS: RESULT: Hydraulic coefficients, CD = Cv = Cc = JNEC CIVIL/FMI/LKK/Aug 2014 Page 17
18 EXPERIMENT NO: VII DETERMINATION OF COEFFICIENT OF DISCHARGE FOR MOUTHPIECE. JNEC CIVIL/FMI/LKK/Aug 2014 Page 18
19 AIM: The objective of this experiment is to find and study the variations of hydraulic coefficients of the given mouthpiece. APPARATUS: mouthpiece apparatus and measuring tanks with a stopwatch for measuring the actual flow rate. DISCRIPTION: A mouthpiece is a short pipe of length not more than two or three times its diameter, connected to an orifice of same size provided in the wall of a reservoir containing fluid. It is an extension of the orifice and through which the fluid is discharged The theoretical discharge through the mouthpiece will be, where, a 0 = area of the mouthpiece (cm 2 ) h = Constant Head (cm) g = acceleration due to gravity, 981 (cm/ sec 2 ) Q th = a 0 x (2 g h) The actual discharge is less than the theoretical discharge due to friction loss and the loss due to expansion of the jet. The actual discharge through the orifice can be determined using the collecting tank and stopwatch setup. Q ac = (A x R) / t where, A = area of the collecting tank (cm 2 ). R = height difference of the water column in the piezometer, (cm) t = time taken to rise R meters, (sec) The coefficient of discharge CD is defined as the ratio of the actual discharge to theoretical discharge. Procedure 1. Note the dimensions of the discharge measuring tank, mouthpiece. 2. Check that the zero of the scale of the inlet tank is the same level as the center line of the mouthpiece. If not, measure the difference in elevation and take it as zero error. 3. Adjust the opening of the inlet valve till the water level in the supply tank become steady. 4. Note down the head. 5. Using collecting tank and stop watch setup measure the actual discharge. 7. Repeat the experiment for different inlet valve openings and tabulate the readings. JNEC CIVIL/FMI/LKK/Aug 2014 Page 19
20 OBSERVATIONS : 1. Diameter of mouthpiece = 2.5 cm 2. Dimension of measuring tank= 30 cm X 40 cm 4. Area of measuring tank (A)= 5. Area of Orifice (a o )= OBSERVATION TABLE : Sr No H (cm) Measuring Tank Reading I. R. (cm) F. R. (cm) Diff 'R ' (cm) t (sec) Qact (cm 3 /sec ) Qth (cm 3 /sec ) C d SPECIMEN CALCULATIONS: RESULT: Hydraulic coefficients, CD = JNEC CIVIL/FMI/LKK/Aug 2014 Page 20
21 EXPERIMENT NO: VII CALIBRATION OF RECTANGULAR NOTCHES JNEC CIVIL/FMI/LKK/Aug 2014 Page 21
22 AIM: To Determine the coefficient of discharge of the given Rectangular notch for different rates of flow. APPARATUS: Rectangular notch fitted on an open channel, hook gauge and measuring tank with stopwatch. DISCRIPTION: In open channel flows, weirs are commonly used to either regulate or to measure the volumetric flow rate. They are of particular use in large scale situations such as irrigation schemes, canals and rivers. For small scale applications, weirs are often referred to as notches and are sharp edged and manufactured from thin plate material. The basic principle is that discharge is directly related to the water depth above the crotch (bottom) of the notch. This distance is called head over the notch. Due to the minimal installation costs flow rate measurement with a notch is very less expensive. The rectangular notch is the most commonly used thin plate weir. The flow pattern over a notch or weir is complex and there is no analytical Solution to the relationship between discharge and head so that a semiempirical Approach has to be used. The expression for theoretical discharge over a rectangular notch is given by, where, L = width of the notch, (m) h= head of water over the notch, (m) g= acceleration due to gravity (m/s 2 ) Water is allowed to pass through the given notch at different flow rates. Actual discharge through the channel can be determined using the collecting tank and stopwatch setup. Where, a = area of the collecting tank. (m 2 ) H = height difference of the water column in the piezometer, (m) t = time taken to rise H meters, (sec) The coefficient of discharge CD is defined as the ratio of actual discharge obtained experimentally to the theoretical discharge.?????????? JNEC CIVIL/FMI/LKK/Aug 2014 Page 22
23 The calibration equation is stated as, where, Qac = K h n K and n are constants depending on the geometry of the notch. Taking logarithm on both sides we get, logqac = log k+n log h which is the equation of a straight line, where, log k is the y intercept and n is its slope. The graph logqac Vs. logh is to be plotted to find k and n. Procedure 1. Check the experimental setup for leaks. Measure the dimensions of collecting tank and the notch. 2. Observe the initial reading of the hook gauge and make sure there is no discharge. Note down the sill level position of the hook gauge. 3. Open the inlet valve of the supply pipe for a slightly increased discharge. Wait for sometime till the flow become steady. 4. Adjust the hook gauge to touch the new water level and note down the reading. Difference of this hook gauge reading with initial still level reading is the head over the notch (h). 5. Collect the water in the collecting tank and observe the time t to collect H height of water. 6. Repeat the above procedure for different flow rates by adjusting the inlet valve opening and tabulate the readings. 7. Complete the tabulation and find the mean value of CD. 8. Draw the necessary graphs and calibrate the the notch. JNEC CIVIL/FMI/LKK/Aug 2014 Page 23
24 Observations: Length of the rectangular notch = m Angle of the triangular notch = deg Collecting tank area = m 2 Observation table: Hook Guage Reading Measuring Tank Reading Vol. Q act Q th Sr. No C.B. w.l. Diff. H (cm) H (m) I.R. F.R. Diff. R (cm) R. (m) t (sec) (m 3 ) (m 3 / sec) (m 3 / sec) C d K Log H Log Q act SPECIMEN CALCULATIONS: K = Cd.(2/3). (2g) L For first reading: Q act = m 3 /sec Q theo = m 3 /sec C d = Q/ Q theo K = Should be n =~ (3/2 ) JNEC CIVIL/FMI/LKK/Aug 2014 Page 24
25 If we take the log for the two sides of equation : log Q = log K + n log H, where n : the power of H = ( the slope.) from table. log k = from graph k = C d =. Results : The average coefficient of discharge of the given for Triangular notch, Cd = Cd From Graph=. JNEC CIVIL/FMI/LKK/Aug 2014 Page 25
26 EXPERIMENT NO: VIII CALIBRATION OF TRIANGULAR NOTCHES JNEC CIVIL/FMI/LKK/Aug 2014 Page 26
27 AIM: To Determine the coefficient of discharge of the given Triangular notch for different rates of flow. APPARATUS: Triangular notch fitted on an open channel, hook gauge and measuring tank with stopwatch. DISCRIPTION: In open channel flows, weirs are commonly used to either regulate or to measure the volumetric flow rate. They are of particular use in large scale situations such as irrigation schemes, canals and rivers. For small scale applications, weirs are often referred to as notches and are sharp edged and manufactured from thin plate material. The basic principle is that discharge is directly related to the water depth above the crotch (bottom) of the notch. This distance is called head over the notch. Due to the minimal installation costs flow rate measurement with a notch is very less expensive. The V notch or triangular notch design causes small changes in discharge to have a large change in depth allowing more accurate head measurement than with a rectangular notch. The flow pattern over a notch or weir is complex and there is no analytical solution to the relationship between discharge and head so that a semiempirical approach has to be used. The expression for Theoretical discharge over a triangular notch is given by, where, θ= angle of the notch, (deg) h= head of water over the notch, (m) g= acceleration due to gravity (m/s 2 ) Water is allowed to pass through the given notch at different flow rates. Actual discharge through the channel can be determined using the collecting tank and stopwatch setup. Where, a = area of the collecting tank. (m 2 ) H = height difference of the water column in the piezometer, (m) t = time taken to rise H meters, (sec) The coefficient of discharge CD is defined as the ratio of actual discharge obtained experimentally to the theoretical discharge. i.e.?????????? JNEC CIVIL/FMI/LKK/Aug 2014 Page 27
28 The calibration equation is stated as, where, Qac = K h n K and n are constants depending on the geometry of the notch. Taking logarithm on both sides we get, logqac = log k+n log h which is the equation of a straight line, where, log k is the y intercept and n is its slope. The graph logqac Vs. logh is to be plotted to find k and n. Procedure 1. Check the experimental setup for leaks. Measure the dimensions of collecting tank and the notch. 2. Observe the initial reading of the hook gauge and make sure there is no discharge. Note down the sill level position of the hook gauge. 3. Open the inlet valve of the supply pipe for a slightly increased discharge. Wait for sometime till the flow become steady. 4. Adjust the hook gauge to touch the new water level and note down the reading. Difference of this hook gauge reading with initial still level reading is the head over the notch (h). 5. Collect the water in the collecting tank and observe the time t to collect H height of water. 6. Repeat the above procedure for different flow rates by adjusting the inlet valve opening and tabulate the readings. 7. Complete the tabulation and find the mean value of CD. 8. Draw the necessary graphs and calibrate the the notch. JNEC CIVIL/FMI/LKK/Aug 2014 Page 28
29 Observations : Length of the rectangular notch = m Angle of the triangular notch = deg Collecting tank area = m 2 Observation table : Hook Guage Reading Measuring Tank Reading Vol. Q act Q th Sr. No C.B. w.l. Diff. H (cm) H (m) I.R. F.R. Diff. R (cm) R. (m) t (sec) (m 3 ) (m 3 / sec) (m 3 / sec) C d K Log H Log Q act SPECIMEN CALCULATIONS: K = C d.(8/15). (2g) tan (θ/2) For first reading: Q act = m 3 /sec Q theo = m 3 /sec C d = Q/ Q theo K = Should be n =~ (3/2 ) JNEC CIVIL/FMI/LKK/Aug 2014 Page 29
30 If we take the log for the two sides of equation : log Q = log K + n log H, where n : the power of H = ( the slope.) from table. log k = from graph k = C d =. Results : The average coefficient of discharge of the given for Triangular notch, Cd = Cd From Graph= JNEC CIVIL/FMI/LKK/Aug 2014 Page 30
MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, FLUID MECHANICS LABORATORY MANUAL
MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, AURANGABAD. (M.S.) DEPARTMENT OF CIVIL ENGINEERING FLUID MECHANICS LABORATORY MANUAL Prepared By Mr. L. K. Kokate Lab Incharge Approved By
More informationEXPERIMENT No.1 FLOW MEASUREMENT BY ORIFICEMETER
EXPERIMENT No.1 FLOW MEASUREMENT BY ORIFICEMETER 1.1 AIM: To determine the coefficient of discharge of the orifice meter 1.2 EQUIPMENTS REQUIRED: Orifice meter test rig, Stopwatch 1.3 PREPARATION 1.3.1
More informationIf a stream of uniform velocity flows into a blunt body, the stream lines take a pattern similar to this: Streamlines around a blunt body
Venturimeter & Orificemeter ELEMENTARY HYDRAULICS National Certificate in Technology (Civil Engineering) Chapter 5 Applications of the Bernoulli Equation The Bernoulli equation can be applied to a great
More informationExperiment (4): Flow measurement
Experiment (4): Flow measurement Introduction: The flow measuring apparatus is used to familiarize the students with typical methods of flow measurement of an incompressible fluid and, at the same time
More informationBRCM COLLEGE OF ENGINEERING & TECHNOLOGY Practical Experiment Instructions Sheet
Exp. Title FLUID MECHANICS I LAB Syllabus FMI Semester4 th Page No. 1 of 1 Internal Marks: 25 L T P External Marks: 25 0 0 2 Total Marks: 50 1. To determine the met centric height of a floating body
More informationDARSHAN INSTITUTE OF ENGINEERING AND TECHNOLOGY, RAJKOT FLUID MECHANICS ( )
DARSHAN INSTITUTE OF ENGINEERING AND TECHNOLOGY, RAJKOT FLUID MECHANICS (2141906) Sr. No. Experiment Start Date End Date Sign Remark 1. To understand pressure measurement procedure and related instruments/devices.
More informationUNIT I FLUID PROPERTIES AND STATICS
SIDDHARTH GROUP OF INSTITUTIONS :: PUTTUR Siddharth Nagar, Narayanavanam Road 517583 QUESTION BANK (DESCRIPTIVE) Subject with Code : Fluid Mechanics (16CE106) Year & Sem: IIB.Tech & ISem Course & Branch:
More informationExperiment To determine the coefficient of impact for vanes. Experiment To determine the coefficient of discharge of an orifice meter.
SUBJECT: FLUID MECHANICS VIVA QUESTIONS (M.E 4 th SEM) Experiment To determine the coefficient of impact for vanes. Q1. Explain impulse momentum principal. Ans1. Momentum equation is based on Newton s
More informationMass of fluid leaving per unit time
5 ENERGY EQUATION OF FLUID MOTION 5.1 Eulerian Approach & Control Volume In order to develop the equations that describe a flow, it is assumed that fluids are subject to certain fundamental laws of physics.
More information5 ENERGY EQUATION OF FLUID MOTION
5 ENERGY EQUATION OF FLUID MOTION 5.1 Introduction In order to develop the equations that describe a flow, it is assumed that fluids are subject to certain fundamental laws of physics. The pertinent laws
More informationDETERMINATION OF DISCHARGE AND HEAD LOSS USING A FLOWMEASURING APPARATUS
DETERMINATION OF DISCHARGE AND HEAD LOSS USING A FLOWMEASURING APPARATUS 1. INTRODUCTION Through use of the FlowMeasuring Apparatus, this experiment is designed to accustom students to typical methods
More informationFLUID MECHANICS AND MACHINERY LABORATORY
FLUID MECHANICS AND MACHINERY LABORATORY STUDENTS REFERENCE MANUAL K.L.UNIVERSITY DEPARTMENT OF CIVIL ENGINEERING Compiled By P.SUNDARA KUMAR, M.Tech (PhD) Associate Professor 1 Preface In most of the
More informationLABORATORY MANUAL FLUID MECHANICS ME216F
LABORATORY MANUAL FLUID MECHANICS ME216F LIST OF THE EXPERIMENT SNO NAME OF THE EXPERIMENT PAGE NO FROM TO 1. To determine the coefficient of impact for vanes. 2. To determine the coefficient of discharge
More informationFluid Mechanics Lab (ME216F) List of Experiments
Fluid Mechanics Lab (ME216F) List of Experiments 1. To determine the coefficient of discharge C d, velocity C v, and contraction C c of various types of orifices 2. Determine of discharge coefficients
More informationCOURSE CODE : 3072 COURSE CATEGORY : B PERIODS/ WEEK : 5 PERIODS/ SEMESTER : 75 CREDIT : 5 TIME SCHEDULE
COURSE TITLE : FLUID MECHANICS COURSE CODE : 307 COURSE CATEGORY : B PERIODS/ WEEK : 5 PERIODS/ SEMESTER : 75 CREDIT : 5 TIME SCHEDULE MODULE TOPIC PERIOD 1 Properties of Fluids 0 Fluid Friction and Flow
More informationR09. d water surface. Prove that the depth of pressure is equal to p +.
Code No:A109210105 R09 SET1 B.Tech II Year  I Semester Examinations, December 2011 FLUID MECHANICS (CIVIL ENGINEERING) Time: 3 hours Max. Marks: 75 Answer any five questions All questions carry equal
More informationTeacher s Signature. S. No. Experiment marks. 3 To determine the coefficient of discharge of Notch (V and Rectangular types)
S. No. Index Name of experiment Date of performance 1. To determine the coefficient of impact for vanes. 2 To determine coefficient of discharge of an orificemeter. 3 To determine the coefficient of discharge
More informationFLOW MEASUREMENT IN PIPES EXPERIMENT
University of Leicester Engineering Department FLOW MEASUREMENT IN PIPES EXPERIMENT Page 1 FORMAL LABORATORY REPORT Name of the experiment: FLOW MEASUREMENT IN PIPES Author: Apollin nana chaazou Partner
More informationChapter 4 DYNAMICS OF FLUID FLOW
Faculty Of Engineering at Shobra nd Year Civil  016 Chapter 4 DYNAMICS OF FLUID FLOW 41 Types of Energy 4 Euler s Equation 43 Bernoulli s Equation 44 Total Energy Line (TEL) and Hydraulic Grade Line
More informationChapter 3 Bernoulli Equation
1 Bernoulli Equation 3.1 Flow Patterns: Streamlines, Pathlines, Streaklines 1) A streamline, is a line that is everywhere tangent to the velocity vector at a given instant. Examples of streamlines around
More informationExperiment No.4: Flow through Venturi meter. Background and Theory
Experiment No.4: Flow through Venturi meter Background and Theory Introduction Flow meters are used in the industry to measure the volumetric flow rate of fluids. Differential pressure type flow meters
More informationLecture 3 The energy equation
Lecture 3 The energy equation Dr Tim Gough: t.gough@bradford.ac.uk General information Lab groups now assigned Timetable up to week 6 published Is there anyone not yet on the list? Week 3 Week 4 Week 5
More informationCALIBRATION OF NOTCHES
CALIBRATION OF NOTCHES A notch may be defined as an overflow opening provided in the side of a tank or a small channel such that the liquid surface in the tank or channel is below the top edge of the Opening.
More informationUniversity of Engineering and Technology, Taxila. Department of Civil Engineering
University of Engineering and Technology, Taxila Department of Civil Engineering Course Title: CE201 Fluid Mechanics  I Prerequisite(s): None Credit Hours: 2 + 1 Contact Hours: 2 + 3 Text Book(s): Reference
More informationCH.1 Overview of Fluid Mechanics/22 MARKS. 1.1 Fluid Fundamentals.
Content : 1.1 Fluid Fundamentals. 08 Marks Classification of Fluid, Properties of fluids like Specific Weight, Specific gravity, Surface tension, Capillarity, Viscosity. Specification of hydraulic oil
More informationApproximate physical properties of selected fluids All properties are given at pressure kn/m 2 and temperature 15 C.
Appendix FLUID MECHANICS Approximate physical properties of selected fluids All properties are given at pressure 101. kn/m and temperature 15 C. Liquids Density (kg/m ) Dynamic viscosity (N s/m ) Surface
More informationVENTURIMETER EXPERIMENT
ENTURIMETER EXERIMENT. OBJECTİE The main objectives of this experiment is to obtain the coefficient of discharge from experimental data by utilizing venturi meter and, also the relationship between Reynolds
More informationLecture23. Flowmeter Design.
Lecture23 Flowmeter Design. Contents of lecture Design of flowmeter Principles of flow measurement; i) Venturi and ii) Orifice meter and nozzle Relationship between flow rate and pressure drop Relation
More informationI. To find the coefficient of discharge for vcnturi meter. 2. To find the coefficient of discharge for ori rice meter.
Flow Measurement by Venturi and Orifice Meters Objectives: I. To find the coefficient of discharge for vcnturi meter. 2. To find the coefficient of discharge for ori rice meter. Venturi and Orificemeters
More informationMAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, AURANGABAD. (M.S.)
MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, AURANGABAD. (M.S.) DEPARTMENT OF CIVIL ENGINEERING SURVEYING  II LAB MANUAL Prof.R.V.Wanjule Approved By Lab Incharge Dr. S. B. Shinde H.O.D.
More informationS.E. (Mech.) (First Sem.) EXAMINATION, (Common to Mech/Sandwich) FLUID MECHANICS (2008 PATTERN) Time : Three Hours Maximum Marks : 100
Total No. of Questions 12] [Total No. of Printed Pages 8 Seat No. [4262]113 S.E. (Mech.) (First Sem.) EXAMINATION, 2012 (Common to Mech/Sandwich) FLUID MECHANICS (2008 PATTERN) Time : Three Hours Maximum
More informationUNIT IV. Flow through Orifice and Mouthpieces and Flow through Notchs and Weirs
SIDDHARTH GROUP OF INSTITUTIONS :: PUTTUR Siddharth Nagar, Narayanavanam Road 517583 QUESTION BANK (DESCRIPTIVE) Subject with Code : FM(15A01305) Year & Sem: IIB.Tech & ISem Course & Branch: B.Tech 
More informationINSTITUTE OF AERONAUTICAL ENGINEERING Dundigal, Hyderabad AERONAUTICAL ENGINEERING QUESTION BANK : AERONAUTICAL ENGINEERING.
Course Name Course Code Class Branch INSTITUTE OF AERONAUTICAL ENGINEERING Dundigal, Hyderabad  00 0 AERONAUTICAL ENGINEERING : Mechanics of Fluids : A00 : III B. Tech Year : 0 0 Course Coordinator
More informationBACHELOR OF TECHNOLOGY IN MECHANICAL ENGINEERING (COMPUTER INTEGRATED MANUFACTURING)
No. of Printed Pages : 6 BME028 BACHELOR OF TECHNOLOGY IN MECHANICAL ENGINEERING (COMPUTER INTEGRATED MANUFACTURING) TermEnd Examination December, 2011 00792 BME028 : FLUID MECHANICS Time : 3 hours
More informationApplied Fluid Mechanics
Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and
More informationENGINEERING FLUID MECHANICS. CHAPTER 1 Properties of Fluids
CHAPTER 1 Properties of Fluids ENGINEERING FLUID MECHANICS 1.1 Introduction 1.2 Development of Fluid Mechanics 1.3 Units of Measurement (SI units) 1.4 Mass, Density, Specific Weight, Specific Volume, Specific
More informationFLUID MECHANICES LAB:I
Force Area Length FLUID MECHANICES LAB:I Experiment:0 Measurement of viscosity by Redwood viscometer. Aim:  To determine the kinematic viscosity of a liquid and its variation with temperature. Apparatus:
More informationRepublic of Iraq Ministry of Higher Education and Scientific Research University of Technology Electromechanical Department
RepublicofIraq MinistryofHigherEducation andscientificresearch UniversityofTechnology Electromechanical Department  2014 1436 Experiment No. 1 Calibration of Bourdon Gauge Description: Figure (1) shows
More informationThe general rules of statics (as applied in solid mechanics) apply to fluids at rest. From earlier we know that:
ELEMENTARY HYDRAULICS National Certificate in Technology (Civil Engineering) Chapter 2 Pressure This section will study the forces acting on or generated by fluids at rest. Objectives Introduce the concept
More informationExperimental and CFD analysis of flow through venturimeter to determine the coefficient of discharge
Experimental and CFD analysis of flow through venturimeter to determine the coefficient of discharge Nikhil Tamhankar Amar Pandhare Ashwinkumar Joglekar Vaibhav Bansode Abstract The pressure distribution
More informationEXPERIMENT NO. 4 CALIBRATION OF AN ORIFICE PLATE FLOWMETER MECHANICAL ENGINEERING DEPARTMENT KING SAUD UNIVERSITY RIYADH
EXPERIMENT NO. 4 CALIBRATION OF AN ORIFICE PLATE FLOWMETER MECHANICAL ENGINEERING DEPARTMENT KING SAUD UNIVERSITY RIYADH Submitted By: ABDULLAH IBN ABDULRAHMAN ID: 13456789 GROUP A EXPERIMENT PERFORMED
More informationLecture 13 Flow Measurement in Pipes. I. Introduction
Lecture 13 Flow Measurement in Pipes I. Introduction There are a wide variety of methods for measuring discharge and velocity in pipes, or closed conduits Many of these methods can provide very accurate
More informationFluid Mechanics. du dy
FLUID MECHANICS Technical English  I 1 th week Fluid Mechanics FLUID STATICS FLUID DYNAMICS Fluid Statics or Hydrostatics is the study of fluids at rest. The main equation required for this is Newton's
More informationTHE SCREW GAUGE. AIM: To learn to use a Screw Gauge and hence use it to find the dimensions of various regular materials given.
EXPERIMENT NO: DATE: / / 0 THE SCREW GAUGE AIM: To learn to use a Screw Gauge and hence use it to find the dimensions of various regular materials given. APPARUTUS: Given a Screw Gauge, cylindrical glass
More informationTOPICS. Density. Pressure. Variation of Pressure with Depth. Pressure Measurements. Buoyant ForcesArchimedes Principle
Lecture 6 Fluids TOPICS Density Pressure Variation of Pressure with Depth Pressure Measurements Buoyant ForcesArchimedes Principle Surface Tension ( External source ) Viscosity ( External source ) Equation
More informationNPTEL Quiz Hydraulics
Introduction NPTEL Quiz Hydraulics 1. An ideal fluid is a. One which obeys Newton s law of viscosity b. Frictionless and incompressible c. Very viscous d. Frictionless and compressible 2. The unit of kinematic
More informationLOSSES DUE TO PIPE FITTINGS
LOSSES DUE TO PIPE FITTINGS Aim: To determine the losses across the fittings in a pipe network Theory: The resistance to flow in a pipe network causes loss in the pressure head along the flow. The overall
More informationCHAPTER THREE FLUID MECHANICS
CHAPTER THREE FLUID MECHANICS 3.1. Measurement of Pressure Drop for Flow through Different Geometries 3.. Determination of Operating Characteristics of a Centrifugal Pump 3.3. Energy Losses in Pipes under
More informationLaboratory work No 2: Calibration of Orifice Flow Meter
Laboratory work No : Calibration of Orifice Flow Meter 1. Objective Calibrate the orifice flow meter and draw the graphs p = f 1 (Q) and C d = f (Re ).. Necessary equipment 1. Orifice flow meter. Measuring
More informationJawaharlal Nehru Engineering College
Jawaharlal Nehru Engineering College Laboratory Manual For MOMENTUM TRANSFER Second Year Students Chemical Engineering Author JNEC, Aurangabad FORWARD It is my great pleasure to present this laboratory
More informationCE MECHANICS OF FLUIDS
CE60  MECHANICS OF FLUIDS (FOR III SEMESTER) UNIT II FLUID STATICS & KINEMATICS PREPARED BY R.SURYA, M.E Assistant Professor DEPARTMENT OF CIVIL ENGINEERING DEPARTMENT OF CIVIL ENGINEERING SRI VIDYA COLLEGE
More informationFor example an empty bucket weighs 2.0kg. After 7 seconds of collecting water the bucket weighs 8.0kg, then:
Hydraulic Coefficient & Flow Measurements ELEMENTARY HYDRAULICS National Certificate in Technology (Civil Engineering) Chapter 3 1. Mass flow rate If we want to measure the rate at which water is flowing
More informationPROPERTIES OF FLUIDS
Unit  I Chapter  PROPERTIES OF FLUIDS Solutions of Examples for Practice Example.9 : Given data : u = y y, = 8 Poise = 0.8 Pas To find : Shear stress. Step  : Calculate the shear stress at various
More informationFlowmeter Discharge Coefficient Estimation
Bankston 1 Flowmeter Discharge Coefficient Estimation Elizabeth Bankston Team 1 Abstract An Edibon FME18 Flow Meter demonstration system was used to obtain experimental values for this experiment. The
More information2.The lines that are tangent to the velocity vectors throughout the flow field are called steady flow lines. True or False A. True B.
CHAPTER 03 1. Write Newton's second law of motion. YOUR ANSWER: F = ma 2.The lines that are tangent to the velocity vectors throughout the flow field are called steady flow lines. True or False 3.Streamwise
More informationABSTRACT I. INTRODUCTION
2016 IJSRSET Volume 2 Issue 4 Print ISSN : 23951990 Online ISSN : 23944099 Themed Section: Engineering and Technology Analysis of Compressible Effect in the Flow Metering By Orifice Plate Using Prasanna
More informationInstruction Manual. Equipment for Engineering Education
Equipment for Engineering Education Instruction Manual HM15007 Bernoulli s Principle Demonstrator GUNT Gerätebau GmbH PO Box 1125 D22881 Barsbüttel Germany Phone (040) 6708540 Fax (040) 67085442 Instruction
More informationFlow Measurement in Pipes and Ducts COURSE CONTENT
Flow Measurement in Pipes and Ducts Dr. Harlan H. Bengtson, P.E. COURSE CONTENT 1. Introduction This course is about measurement of the flow rate of a fluid flowing under pressure in a closed conduit.
More informationChapter 1 Fluid and their Properties
June 15 Jan  16 GTU Paper Analysis (New Syllabus) Chapter 1 Fluid and their Properties Sr. No. Questions Theory 2. Explain the following terms: Relative density 2. Kinematic viscosity 3. Cavitation 4.
More informationApplied Fluid Mechanics
Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and
More informationApplied Fluid Mechanics
Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and
More informationHead Discharge Relationship of Thin Plated Rectangular Lab Fabricated Sharp Crested Weirs
Journal of Applied Fluid Mechanics, Vol. 9, No. 3, pp. 12311235, 2016. Available online at www.jafmonline.net, ISSN 17353572, EISSN 17353645. DOI: 10.18869/acadpub.jafm.68.228.23128 Head Discharge Relationship
More informationCHAPTER 3 BASIC EQUATIONS IN FLUID MECHANICS NOOR ALIZA AHMAD
CHAPTER 3 BASIC EQUATIONS IN FLUID MECHANICS 1 INTRODUCTION Flow often referred as an ideal fluid. We presume that such a fluid has no viscosity. However, this is an idealized situation that does not exist.
More informationLaboratory exercise 1: Open channel flow measurement
Chapter 1 Laboratory exercise 1: Open channel flow measurement Laboratory exercise Open channel flow measurement is placed on the Faculty of Civil and Geodetic Engineering, on Department of Environmental
More information[Prasanna m a*et al., 5(6): July, 2016] ISSN: IC Value: 3.00 Impact Factor: 4.116
IJESRT INTERNATIONAL JOURNAL OF ENGINEERING SCIENCES & RESEARCH TECHNOLOGY NUMERICAL ANALYSIS OF COMPRESSIBLE EFFECT IN THE FLOW METERING BY CLASSICAL VENTURIMETER Prasanna M A *, Dr V Seshadri, Yogesh
More informationSUMMER 14 EXAMINATION
Important Instructions to examiners: 1) The answers should be examined by key words and not as wordtoword as given in the model answer scheme. 2) The model answer and the answer written by candidate
More informationESSEX COUNTY COLLEGE Engineering Technologies and Computer Sciences Division MET 215 Fluid Mechanics Course Outline
ESSEX COUNTY COLLEGE Engineering Technologies and Computer Sciences Division MET 215 Fluid Mechanics Course Outline Course Number & Name: MET 215 Fluid Mechanics Credit Hours: 3.0 Contact Hours: 4.5 Lecture:
More informationFLUID MECHANICS. Chapter 3 Elementary Fluid Dynamics  The Bernoulli Equation
FLUID MECHANICS Chapter 3 Elementary Fluid Dynamics  The Bernoulli Equation CHAP 3. ELEMENTARY FLUID DYNAMICS  THE BERNOULLI EQUATION CONTENTS 3. Newton s Second Law 3. F = ma along a Streamline 3.3
More informationChapter 3 Fluid Statics
Chapter 3 Fluid Statics 3.1 Pressure Pressure : The ratio of normal force to area at a point. Pressure often varies from point to point. Pressure is a scalar quantity; it has magnitude only It produces
More informationME332 FLUID MECHANICS LABORATORY (PART I)
ME332 FLUID MECHANICS LABORATORY (PART I) Mihir Sen Department of Aerospace and Mechanical Engineering University of Notre Dame Notre Dame, IN 46556 Version: January 14, 2002 Contents Unit 1: Hydrostatics
More informationDischarge measurements
Discharge measurements DISCHARGE MEASUREMENTS Suitability of methods and equipment for measurements where, under what conditions and for what reason discharge is measured determination of discharge in
More informationMechanical Measurements and Metrology Prof. S. P. Venkateshan Department of Mechanical Engineering Indian Institute of Technology, Madras
Mechanical Measurements and Metrology Prof. S. P. Venkateshan Department of Mechanical Engineering Indian Institute of Technology, Madras Module  3 Lecture  33 Measurement of Volume and Mass Flow Rate
More informationCompressible Gas Flow
Compressible Gas Flow by Elizabeth Adolph Submitted to Dr. C. Grant Willson CHE53M Department of Chemical Engineering The University of Texas at Austin Fall 008 Compressible Gas Flow Abstract In this lab,
More informationMeasurements using Bernoulli s equation
An Internet Book on Fluid Dynamics Measurements using Bernoulli s equation Many fluid measurement devices and techniques are based on Bernoulli s equation and we list them here with analysis and discussion.
More informationThe Bernoulli Equation
The Bernoulli Equation The most used and the most abused equation in fluid mechanics. Newton s Second Law: F = ma In general, most real flows are 3D, unsteady (x, y, z, t; r,θ, z, t; etc) Let consider
More informationExam #2: Fluid Kinematics and Conservation Laws April 13, 2016, 7:00 p.m. 8:40 p.m. in CE 118
CVEN 311501 (Socolofsky) Fluid Dynamics Exam #2: Fluid Kinematics and Conservation Laws April 13, 2016, 7:00 p.m. 8:40 p.m. in CE 118 Name: : UIN: : Instructions: Fill in your name and UIN in the space
More informationCHAPTER (13) FLOW MEASUREMENTS
CHAPTER (13) FLOW MEASUREMENTS 09/12/2010 Dr. Munzer Ebaid 1 Instruments for the Measurements of Flow Rate 1. Direct Methods: Volume or weight measurements. 2. Indirect Methods: Venturi meters, Orifices
More informationExperimental Investigation on the Influence of Density of Fluid. On Efficiency of V Notch
International Journal of Advances in Scientific Research and Engineering (ijasre) EISSN : 24548006 DOI: http://dx.doi.org/10.7324/ijasre.2017.32515 Vol.3 (9) Oct  2017 Experimental Investigation on
More informationEXPERIMENT NO: F5. Losses in Piping Systems
SJSU ME115  THERMAL ENGINEERING LAB EXPERIMENT NO: F5 Losses in Piping Systems Objective One of the most common problems in fluid mechanics is the estimation of pressure loss. It is the objective of this
More informationANNA UNIVERSITY OF TECHNOLOGY  COIMBATORE
ANNA UNIVERSITY OF TECHNOLOGY  COIMBATORE P.A COLLEGE OF ENGINEERING AND TECHNOLOGY, POLLACHI  02. DEPARTMENT OF MECHANICAL ENGINEERING YEAR / SEMESTER  II / III ME 2208  FLUID MECHANICS AND MACHINERY
More informationChapter 1 INTRODUCTION
Chapter 1 INTRODUCTION 11 The Fluid. 12 Dimensions. 13 Units. 14 Fluid Properties. 1 11 The Fluid: It is the substance that deforms continuously when subjected to a shear stress. Matter Solid Fluid
More informationSLIDING FRICTION & CONSERVATION OF ENERGY
SLIDING FRICTION & CONSERVATION OF ENERGY Saddleback College Physics Department (adapted from PASCO Scientific) Purpose: Part I To experimentally determine the coefficient of kinetic (sliding) friction,
More informationQ1 Give answers to all of the following questions (5 marks each):
FLUID MECHANICS First Year Exam Solutions 03 Q Give answers to all of the following questions (5 marks each): (a) A cylinder of m in diameter is made with material of relative density 0.5. It is moored
More informationVisualization of flow pattern over or around immersed objects in open channel flow.
EXPERIMENT SEVEN: FLOW VISUALIZATION AND ANALYSIS I OBJECTIVE OF THE EXPERIMENT: Visualization of flow pattern over or around immersed objects in open channel flow. II THEORY AND EQUATION: Open channel:
More information53:071 Principles of Hydraulics Laboratory Experiment #3 ANALYSIS OF OPENCHANNEL FLOW TRANSITIONS USING THE SPECIFIC ENERGY DIAGRAM
53:071 Principles of Hydraulics Laboratory Experiment #3 ANALYSIS OF OPENCHANNEL FLOW TRANSITIONS USING THE SPECIFIC ENERGY DIAGRAM Principle Adaptation of the Bernoulli equation to openchannel flows
More informationNew Website: Mr. Peterson s Address:
Brad Peterson, P.E. New Website: http://njut2009fall.weebly.com Mr. Peterson s Email Address: bradpeterson@engineer.com Lesson 1, Properties of Fluids, 2009 Sept 04, Rev Sept 18 Lesson 2, Fluid Statics,
More informationDEPARTMENT OF CHEMICAL ENGINEERING University of Engineering & Technology, Lahore. Fluid Mechanics Lab
DEPARTMENT OF CHEMICAL ENGINEERING University of Engineering & Technology, Lahore Fluid Mechanics Lab Introduction Fluid Mechanics laboratory provides a hands on environment that is crucial for developing
More informationOrifice and Venturi Pipe Flow Meters
Orifice and Venturi Pipe Flow Meters by Harlan H. Bengtson, PhD, P.E. 1. Introduction Your Course Title Here The flow rate of a fluid flowing in a pipe under pressure is measured for a variety of applications,
More informationCEE 3310 Control Volume Analysis, Oct. 7, D Steady State Head Form of the Energy Equation P. P 2g + z h f + h p h s.
CEE 3310 Control Volume Analysis, Oct. 7, 2015 81 3.21 Review 1D Steady State Head Form of the Energy Equation ( ) ( ) 2g + z = 2g + z h f + h p h s out where h f is the friction head loss (which combines
More informationPredictionof discharge coefficient of Venturimeter at low Reynolds numbers by analytical and CFD Method
International Journal of Engineering and Technical Research (IJETR) ISSN: 23210869, Volume3, Issue5, May 2015 Predictionof discharge coefficient of Venturimeter at low Reynolds numbers by analytical
More informationFluid Mechanics c) Orificemeter a) Viscous force, Turbulence force, Compressible force a) Turbulence force c) Integration d) The flow is rotational
Fluid Mechanics 1. Which is the cheapest device for measuring flow / discharge rate. a) Venturimeter b) Pitot tube c) Orificemeter d) None of the mentioned 2. Which forces are neglected to obtain Euler
More informationFluid Mechanics Prof. S. K. Som Department of Mechanical Engineering Indian Institute of Technology, Kharagpur. Lecture  9 Fluid Statics Part VI
Fluid Mechanics Prof. S. K. Som Department of Mechanical Engineering Indian Institute of Technology, Kharagpur Lecture  9 Fluid Statics Part VI Good morning, I welcome you all to this session of Fluid
More informationBenha University College of Engineering at Benha Questions For Corrective Final Examination Subject: Fluid Mechanics M 201 May 24/ 2016
Benha University College of Engineering at Benha Questions For Corrective Final Examination Subject: Fluid Mechanics M 01 May 4/ 016 Second year Mech. Time :180 min. Examiner:Dr.Mohamed Elsharnoby Attempt
More informationChapter Four fluid flow mass, energy, Bernoulli and momentum
41Conservation of Mass Principle Consider a control volume of arbitrary shape, as shown in Fig (41). Figure (41): the differential control volume and differential control volume (Total mass entering
More informationClosed duct flows are full of fluid, have no free surface within, and are driven by a pressure gradient along the duct axis.
OPEN CHANNEL FLOW Open channel flow is a flow of liquid, basically water in a conduit with a free surface. The open channel flows are driven by gravity alone, and the pressure gradient at the atmospheric
More informationChapter 7 The Energy Equation
Chapter 7 The Energy Equation 7.1 Energy, Work, and Power When matter has energy, the matter can be used to do work. A fluid can have several forms of energy. For example a fluid jet has kinetic energy,
More informationExperiment (3): Impact of jet
Experiment (3): Impact of jet Introduction: Impact of jets apparatus enables experiments to be carried out on the reaction force produced on vanes when a jet of water impacts on to the vane. The study
More informationObjectives. Conservation of mass principle: Mass Equation The Bernoulli equation Conservation of energy principle: Energy equation
Objectives Conservation of mass principle: Mass Equation The Bernoulli equation Conservation of energy principle: Energy equation Conservation of Mass Conservation of Mass Mass, like energy, is a conserved
More information2 Internal Fluid Flow
Internal Fluid Flow.1 Definitions Fluid Dynamics The study of fluids in motion. Static Pressure The pressure at a given point exerted by the static head of the fluid present directly above that point.
More informationChapter (6) Energy Equation and Its Applications
Chapter (6) Energy Equation and Its Applications Bernoulli Equation Bernoulli equation is one of the most useful equations in fluid mechanics and hydraulics. And it s a statement of the principle of conservation
More information