# EXPERIMENT NO. 4 CALIBRATION OF AN ORIFICE PLATE FLOWMETER MECHANICAL ENGINEERING DEPARTMENT KING SAUD UNIVERSITY RIYADH

Size: px
Start display at page:

Download "EXPERIMENT NO. 4 CALIBRATION OF AN ORIFICE PLATE FLOWMETER MECHANICAL ENGINEERING DEPARTMENT KING SAUD UNIVERSITY RIYADH"

Transcription

1 EXPERIMENT NO. 4 CALIBRATION OF AN ORIFICE PLATE FLOWMETER MECHANICAL ENGINEERING DEPARTMENT KING SAUD UNIVERSITY RIYADH Submitted By: ABDULLAH IBN ABDULRAHMAN ID: GROUP A EXPERIMENT PERFORMED ON 9/01/013 1

2 1 Introduction In this experiment an orifice plate flow-meter is calibrated and the calculated coefficient of discharge, C d, is compared to the given value in the lab report as well as other published data. The orifice plate flow-meter provides a simple and low cost method for measuring the flow rate in a pipe using the pressure drop measurement across the plate. The orifice plate is a circular plate with a sharp square edge hole in the center inserted in a pipe. When a fluid is allowed to flow inside the pipe, the orifice plate obstructs the flow which results in fluid pressure loss. A schematic diagram of an orifice plate installed in a pipe with flow from left to right is shown in figure 1. The pressure loss is dependent on the orifice diameter, pipe diameter and the flow rate. Hence the flow rate is less than the theoretical flow rate which would occur if there were no losses. Figure 1: Orifice plate schematic For ideal flow (no viscous effects) the flow rate can be calculated from Bernoulli s equation, as follows, [3], p 1 ρ + V 1 + gz 1 = p ρ + V + gz (1) Where, p, ρ, V, z and g are the fluid pressure, density, velocity, elevation and acceleration due to gravity, respectively. The subscripts 1 and denote the conditions upstream and downstream of the plate, respectively, as shown in figure Since the flow is incompressible we can write the flow velocity upstream of the orifice, V 1, in terms of the throat velocity, V, using the continuity equation given in equation, [3],

3 Figure : Orifice plate stations A 1 V 1 = A V () V 1 = A V = d V = β V (3) A 1 d 1 Where, β = d d 1 and d 1, d are the pipe and throat diameters, respectively. Substituting equation 3 into equation 1 assuming the flow is horizontal (z 1 = z = 0), we get, V V 1 = [ 1 ( ) ] 4 d d 1 V = (p 1 p ) ρ [ 1 β 4] V = (p 1 p ) ρ V = 1 1 β 4 (p 1 p ) ρ (4) (5) (6) Since the flow rate, Q, is given by Q = AV, where A and V are the cross-section area and average speed at any section, respectively, then we write the flow rate across the orifice plate as follows, Q = A V = A (p 1 p ) 1 β 4 ρ Equation 7, gives an expression for the maximum theoretical volume flow rate assuming ideal flow. However, since real flows are viscous, hence, the maximum flow rate will be less than the ideal value due to viscous losses. This is taken into account by multiplying the ideal Q by a factor, C d, this factor is the coefficient of discharge. Hence, we can define the coefficient of discharge as the ratio of actual to ideal volume flow rate. Q Actual = C d Q Ideal = C da (p 1 p ) 1 β 4 ρ The coefficient of discharge depends on the Reynolds number based on the pipe s diameter as well as the orifice diameter to pipe diameter ratio, β. The main objective of this experiment is to determine this coefficient. In the next section, the apparatus and techniques used to determine C d are discussed. In section 3 the results (7) (8) 3

4 and discussion are presented, then finally conclusions are drawn in section 4.1 Apparatus and Techniques Apparatus The orifice plate used is part of SOLTEQ flow-meter measurement apparatus, model FM 101, shown in Figure 3. Figure 3: SOLTEQ flow-meter measurement apparatus, FM101. The SOLTEQ FM 101 flow meter apparatus consists of three different types of volume flow-meters. It contains a Venturi type flow-meter, an orifice plate flow-meter as well as a Rotameter. All flow-meters are connected in series. The apparatus also includes a flow control valve that can be used to adjust the volume flow rate. Water is allowed into the apparatus through the water inlet and is dumped through the water outlet as shown in Figure 3. The apparatus is connected to a TQ H-10 Volumetric Bench, which contains a closed circuit water pump. The water exiting the FM 101 module is dumped in a collection tank and is pumped back into the water inlet. The SOLTEQ FM 101 also contains a multi-tube manometer to measure the difference in pressure across the Venturi and orifice plate flow-meters. The manometer has a range of 450 mm and the graduation is in 1 mm increments. The orifice plate flow-meter, shown in Figure 4(a), is fitted with two pressure taps located one tube diameter upstream of the orifice plate and half a tube diameter downstream the plate. The difference in these pressures is the required p in equation 8. The tube diameter, D, is equal to 6 mm while the orifice-meter throat diameter, DT, is equal to 16 mm. Therefore the ratio of diameters, β, is β = DDT =

5 (a) Orifice plate flow-meter. (b) Rotameter. Figure 4: SOLTEQ FM-101. Procedures and Techniques In this experiment, the SOLTEQ FM-101 was connected to the TQ H-10 Volumetric Bench. The flow-rate was set at three different values 5, 10 and 15 L/min. This was done by adjusting the flow control valve. The volume flow-rate value was read off the Rotameter. Before any measurements were taken, the flow was allowed to reach a steady condition such that there are no air bubbles in the system and all pressure readings were stable. Once the pressure readings in the manometer were stable and all air bubbles were out, the pressure measurements were recorded for each volume flow-rate. Each time the valve position changed, the flow was allowed to stabilize before recording the new pressure readings. The measurements were repeated for the 10 L/min flow-rate case for uncertainty estimates purposes. It is worth noting that reading the water height in the manometer was difficult due to the fact that the water used was clear and the background was white. 3 Results and Discussions In this section the results will be presented and discussed. The atmospheric pressure and temperature during the experiment were measured to be 940 ± 1.5 mbar and ± 1 C, respectively. Since the working fluid is water, the water temperature was measured to be 0 ± 1 C. The water density, ρ, and viscosity, µ, at this temperature were taken as 998. kg/m3 and 10 3 P a s, respectively, []. The pressure measurements upstream, p1, and downstream, p, of the orifice plate are shown in Table 1. Also shown in Table 1, are the average velocity U and the Reynolds 5

6 number, Re U defined as follows, U = Q A = 4Q πd (9) Re U = ρud µ (10) Where, ρ and µ are the water density and viscosity, respectively. Finally, using equation 8 Table 1: Pressure Measurement Q [L/min] p 1 [mm H O] p [mm H O] U [m/s] Re U C d we can write the coefficient of discharge, C d, as follows, C d = Q A ρ (1 β 4 ) (p 1 p ) (11) Equation 11 is used to calculate the values of C d shown in Table 1. The measured discharge coefficient values are close to the supplied value in the lab manual of While recording the pressure measurements it was noticed that there are some fluctuations in the pressure values. This might be due to the fact that the pipe length between the Venturi-meter,and the orifice flow-meter is too short, less than three pipe diameters. This was amplified when the flow rate was set to 5L/min, which corresponds to a Re = This value is in the transition region, hence the flow is not fully turbulent. To get a better reading there should be longer pipe connecting the venturi-meter and the orifice plate or the Reynolds number should be increased. In the next section the uncertainty in measured quantities is presented. 3.1 Uncertainty Analysis Uncertainty analysis is performed based on two readings of the pressure difference at 10 L/min flow-rate. The values are listed in Table. Table : Pressure Measurement for uncertainty estimates at Q = 10 L/min Run 1 p S p p 1 [mm H O] p [mm H O] Where, p and S p are the mean and standard deviation of the measured pressures, respectively. The uncertainty in the measured pressure is given by U p = t p S α/,ν n, Where t is the t-distribution and α = 0.05 is the level of significance, and ν = n 1, where n = is the number of observations. Hence, the uncertainties in the upstream and downstream pressures 6

7 are given as follows, S p1 U p1 = t 0.05,1 n = = 6.35 mm H O (1) S p U p = t 0.05,1 n = = 1.71 mm H O (13) Using, equation 11 we can estimate the uncertainty in C d, U Cd, as follows, [4], U Cd = ( Cd Q U Q ) ( Cd + p 1 U p1 (14) ) ( ) Cd + U p (15) p The uncertianty in the measured flow-rate is taken to be 4% F ull Scale, such that δq = 0.88 L/min. Using equation 11 and the calculated pressure and flow-rate uncertainties we can evaluate the uncertainty in C d as U Cd = ± Since in general the discharge coefficient is a function of the Reynolds number, Re, a plot of the measured C d versus the Reynolds number is shown in Figure 5. Also shown in the figure is a curve fit for experimental data reported in the literature versus Re as given by equation 16, [1]. The length L 1 and L are the pressure taps locations upstream and downstream, respectively. In this case, L 1 = d 1 and L = d. Figure 5: Discharge coefficient versus Reynolds number. 7

8 C d = β β β.5 ( ( ) [ L1 β 4 ] D (1 β ) Re ( L D ) 0.75 ) β 3 (16) The measured coefficient of discharge is C d = 0.67 ± while the value supplied in the lab manual is This is about 6% difference. However, the measured values are within 4.5% from the empirical relation given by equation 16 for Q = 10, 15 L/min while the difference is 8% for the case Q = 5 L/min. This is probably due to the low Reynolds number which is in the transition region. To get higher accuracy we need to make more measurements with more accurate pressure transducers. 4 Conclusions In this work the discharge coefficient of an orifice plate flow-meter was estimated at three values of flow rate, Q = 5, 10 and 15 L/min. The measured values were compared with the values obtained from the lab manual and from a curve fit of empirical data. The following conclusions are made, The mean coefficient of discharge was estimated to be C d = 0.67 ± The difference between current measured C d and values in the literature are within experimental uncertainty Main source of uncertainty is the unsteadiness in pressure measurement. This is due to the fact that the Reynolds number is in the transition region for low values of flow rates The variation in C d versus the Reynolds number follows the curve fit of empirical data in the literature The results can be improved if a more accurate manometer is used as well as more data points are taken Using a very light dye in the water would improve will eliminate human reading errors References [1] Emperical fit for c d : [] Water properties: [3] Robert W. Fox Alan T. McDonald Philip J. Pritchard. Fluid Mechanics. John Wiley & Sons, eighth edition, 01. [4] Thomas G. Beckwith Roy D. Marangoni John H. Lienhard V. Mechanical Measurements. Addison Wesley, fifth edition,

### Experiment (4): Flow measurement

Experiment (4): Flow measurement Introduction: The flow measuring apparatus is used to familiarize the students with typical methods of flow measurement of an incompressible fluid and, at the same time

### Compressible Gas Flow

Compressible Gas Flow by Elizabeth Adolph Submitted to Dr. C. Grant Willson CHE53M Department of Chemical Engineering The University of Texas at Austin Fall 008 Compressible Gas Flow Abstract In this lab,

### FLOW MEASUREMENT IN PIPES EXPERIMENT

University of Leicester Engineering Department FLOW MEASUREMENT IN PIPES EXPERIMENT Page 1 FORMAL LABORATORY REPORT Name of the experiment: FLOW MEASUREMENT IN PIPES Author: Apollin nana chaazou Partner

### ME332 FLUID MECHANICS LABORATORY (PART I)

ME332 FLUID MECHANICS LABORATORY (PART I) Mihir Sen Department of Aerospace and Mechanical Engineering University of Notre Dame Notre Dame, IN 46556 Version: January 14, 2002 Contents Unit 1: Hydrostatics

### Flowmeter Discharge Coefficient Estimation

Bankston 1 Flowmeter Discharge Coefficient Estimation Elizabeth Bankston Team 1 Abstract An Edibon FME18 Flow Meter demonstration system was used to obtain experimental values for this experiment. The

### Major and Minor Losses

Abstract Major and Minor Losses Caitlyn Collazo, Team 2 (1:00 pm) A Technovate fluid circuit system was used to determine the pressure drop across a pipe section and across an orifice. These pressure drops

### Chapter 4 DYNAMICS OF FLUID FLOW

Faculty Of Engineering at Shobra nd Year Civil - 016 Chapter 4 DYNAMICS OF FLUID FLOW 4-1 Types of Energy 4- Euler s Equation 4-3 Bernoulli s Equation 4-4 Total Energy Line (TEL) and Hydraulic Grade Line

### CHAPTER THREE FLUID MECHANICS

CHAPTER THREE FLUID MECHANICS 3.1. Measurement of Pressure Drop for Flow through Different Geometries 3.. Determination of Operating Characteristics of a Centrifugal Pump 3.3. Energy Losses in Pipes under

### Mass of fluid leaving per unit time

5 ENERGY EQUATION OF FLUID MOTION 5.1 Eulerian Approach & Control Volume In order to develop the equations that describe a flow, it is assumed that fluids are subject to certain fundamental laws of physics.

### ME332 FLUID MECHANICS LABORATORY (PART II)

ME332 FLUID MECHANICS LABORATORY (PART II) Mihir Sen Department of Aerospace and Mechanical Engineering University of Notre Dame Notre Dame, IN 46556 Version: April 2, 2002 Contents Unit 5: Momentum transfer

### Measurements using Bernoulli s equation

An Internet Book on Fluid Dynamics Measurements using Bernoulli s equation Many fluid measurement devices and techniques are based on Bernoulli s equation and we list them here with analysis and discussion.

### EXPERIMENT No.1 FLOW MEASUREMENT BY ORIFICEMETER

EXPERIMENT No.1 FLOW MEASUREMENT BY ORIFICEMETER 1.1 AIM: To determine the co-efficient of discharge of the orifice meter 1.2 EQUIPMENTS REQUIRED: Orifice meter test rig, Stopwatch 1.3 PREPARATION 1.3.1

### 5 ENERGY EQUATION OF FLUID MOTION

5 ENERGY EQUATION OF FLUID MOTION 5.1 Introduction In order to develop the equations that describe a flow, it is assumed that fluids are subject to certain fundamental laws of physics. The pertinent laws

### LECTURE 6- ENERGY LOSSES IN HYDRAULIC SYSTEMS SELF EVALUATION QUESTIONS AND ANSWERS

LECTURE 6- ENERGY LOSSES IN HYDRAULIC SYSTEMS SELF EVALUATION QUESTIONS AND ANSWERS 1. What is the head loss ( in units of bars) across a 30mm wide open gate valve when oil ( SG=0.9) flow through at a

### Lecture 24. Design of flow meters

Lecture 24 Design of flow meters Contents Exercise 1 Exercise 2 Exercise 3 Key Words: Fluid flow, Macroscopic Balance, Frictional Losses, Turbulent Flow, Venturimeter, Orifice Meter, Pitot Tube Exercise

### Lab Section Date. ME4751 Air Flow Rate Measurement

Name Lab Section Date ME4751 Air Flow Rate Measurement Objective The objective of this experiment is to determine the volumetric flow rate of air flowing through a pipe using a Pitot-static tube and a

### Lecture23. Flowmeter Design.

Lecture23 Flowmeter Design. Contents of lecture Design of flowmeter Principles of flow measurement; i) Venturi and ii) Orifice meter and nozzle Relationship between flow rate and pressure drop Relation

### DETERMINATION OF DISCHARGE AND HEAD LOSS USING A FLOW-MEASURING APPARATUS

DETERMINATION OF DISCHARGE AND HEAD LOSS USING A FLOW-MEASURING APPARATUS 1. INTRODUCTION Through use of the Flow-Measuring Apparatus, this experiment is designed to accustom students to typical methods

### 2 Internal Fluid Flow

Internal Fluid Flow.1 Definitions Fluid Dynamics The study of fluids in motion. Static Pressure The pressure at a given point exerted by the static head of the fluid present directly above that point.

### MCE380: Measurements and Instrumentation Lab

MCE380: Measurements and Instrumentation Lab Chapter 8: Flow Measurements Topics: Basic Flow Equations Flow Obstruction Meters Positive Displacement Flowmeters Other Methods Holman, Ch. 7 Cleveland State

### Experiment No.4: Flow through Venturi meter. Background and Theory

Experiment No.4: Flow through Venturi meter Background and Theory Introduction Flow meters are used in the industry to measure the volumetric flow rate of fluids. Differential pressure type flow meters

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### Lecture 13 Flow Measurement in Pipes. I. Introduction

Lecture 13 Flow Measurement in Pipes I. Introduction There are a wide variety of methods for measuring discharge and velocity in pipes, or closed conduits Many of these methods can provide very accurate

### Chapter (4) Motion of Fluid Particles and Streams

Chapter (4) Motion of Fluid Particles and Streams Read all Theoretical subjects from (slides Dr.K.AlASTAL) Patterns of Flow Reynolds Number (R e ): A dimensionless number used to identify the type of flow.

### FLUID MECHANICS D203 SAE SOLUTIONS TUTORIAL 2 APPLICATIONS OF BERNOULLI SELF ASSESSMENT EXERCISE 1

FLUID MECHANICS D203 SAE SOLUTIONS TUTORIAL 2 APPLICATIONS OF BERNOULLI SELF ASSESSMENT EXERCISE 1 1. A pipe 100 mm bore diameter carries oil of density 900 kg/m3 at a rate of 4 kg/s. The pipe reduces

### EXPERIMENT NO: F5. Losses in Piping Systems

SJSU ME115 - THERMAL ENGINEERING LAB EXPERIMENT NO: F5 Losses in Piping Systems Objective One of the most common problems in fluid mechanics is the estimation of pressure loss. It is the objective of this

### Laboratory work No 2: Calibration of Orifice Flow Meter

Laboratory work No : Calibration of Orifice Flow Meter 1. Objective Calibrate the orifice flow meter and draw the graphs p = f 1 (Q) and C d = f (Re ).. Necessary equipment 1. Orifice flow meter. Measuring

### FACULTY OF CHEMICAL & ENERGY ENGINEERING FLUID MECHANICS LABORATORY TITLE OF EXPERIMENT: MINOR LOSSES IN PIPE (E4)

FACULTY OF CHEMICAL & ENERGY ENGINEERING FLUID MECHANICS LABORATORY TITLE OF EXPERIMENT: MINOR LOSSES IN PIPE (E4) 1 1.0 Objectives The objective of this experiment is to calculate loss coefficient (K

### Flow Measurement in Pipes and Ducts COURSE CONTENT

Flow Measurement in Pipes and Ducts Dr. Harlan H. Bengtson, P.E. COURSE CONTENT 1. Introduction This course is about measurement of the flow rate of a fluid flowing under pressure in a closed conduit.

### Fluid Mechanics. du dy

FLUID MECHANICS Technical English - I 1 th week Fluid Mechanics FLUID STATICS FLUID DYNAMICS Fluid Statics or Hydrostatics is the study of fluids at rest. The main equation required for this is Newton's

### VENTURIMETER EXPERIMENT

ENTURIMETER EXERIMENT. OBJECTİE The main objectives of this experiment is to obtain the coefficient of discharge from experimental data by utilizing venturi meter and, also the relationship between Reynolds

### Rate of Flow Quantity of fluid passing through any section (area) per unit time

Kinematics of Fluid Flow Kinematics is the science which deals with study of motion of liquids without considering the forces causing the motion. Rate of Flow Quantity of fluid passing through any section

### Lecture 3 The energy equation

Lecture 3 The energy equation Dr Tim Gough: t.gough@bradford.ac.uk General information Lab groups now assigned Timetable up to week 6 published Is there anyone not yet on the list? Week 3 Week 4 Week 5

### I. To find the coefficient of discharge for vcnturi meter. 2. To find the coefficient of discharge for ori rice meter.

Flow Measurement by Venturi and Orifice Meters Objectives: I. To find the coefficient of discharge for vcnturi meter. 2. To find the coefficient of discharge for ori rice meter. Venturi and Orificemeters

### CHAPTER 3 BASIC EQUATIONS IN FLUID MECHANICS NOOR ALIZA AHMAD

CHAPTER 3 BASIC EQUATIONS IN FLUID MECHANICS 1 INTRODUCTION Flow often referred as an ideal fluid. We presume that such a fluid has no viscosity. However, this is an idealized situation that does not exist.

### Calibration of Orifice Flow Meter and Venturi Flow Meter

Calibration of Orifice Flow Meter and Venturi Flow Meter D. Till Abstract Orifice and venturi flow meters decrease the pressure of a fluid b increasing its velocit as it flows through them. This is done

### Flow Measurement in Pipes and Ducts COURSE CONTENT

Flow Measurement in Pipes and Ducts Dr. Harlan H. Bengtson, P.E. COURSE CONTENT 1. Introduction This course is about measurement of the flow rate of a fluid flowing under pressure in a closed conduit.

### Q1 Give answers to all of the following questions (5 marks each):

FLUID MECHANICS First Year Exam Solutions 03 Q Give answers to all of the following questions (5 marks each): (a) A cylinder of m in diameter is made with material of relative density 0.5. It is moored

### MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, FLUID MECHANICS LABORATORY MANUAL

MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, AURANGABAD. (M.S.) DEPARTMENT OF CIVIL ENGINEERING FLUID MECHANICS LABORATORY MANUAL Prepared By Mr. L. K. Kokate Lab Incharge Approved By

### Mechanical Measurements and Metrology Prof. S. P. Venkateshan Department of Mechanical Engineering Indian Institute of Technology, Madras

Mechanical Measurements and Metrology Prof. S. P. Venkateshan Department of Mechanical Engineering Indian Institute of Technology, Madras Module - 3 Lecture - 33 Measurement of Volume and Mass Flow Rate

### The Mechatronics Design for Measuring Fluid Friction Losses in Pipe Flows Rıza Gurbuz

Solid State Phenomena Vol. 113 (2006) pp 603-608 Online available since 2006/Jun/15 at www.scientific.net (2006) Trans Tech Publications, Switzerland doi:10.4028/www.scientific.net/ssp.113.603 The Mechatronics

### ABSTRACT I. INTRODUCTION

2016 IJSRSET Volume 2 Issue 4 Print ISSN : 2395-1990 Online ISSN : 2394-4099 Themed Section: Engineering and Technology Analysis of Compressible Effect in the Flow Metering By Orifice Plate Using Prasanna

### Experiment- To determine the coefficient of impact for vanes. Experiment To determine the coefficient of discharge of an orifice meter.

SUBJECT: FLUID MECHANICS VIVA QUESTIONS (M.E 4 th SEM) Experiment- To determine the coefficient of impact for vanes. Q1. Explain impulse momentum principal. Ans1. Momentum equation is based on Newton s

### ME 4600:483 Lab Notes Revised 11/16/2015. Flow Measurement

Table of Contents Flow Measurement Flow Measurement... 1 I. Objective... 1 II. Apparatus... 1 III. Principles and Background... 1 Pitot-Static Tubes... 2 Orifice Plates and Unrecoverable Losses... 4 Flow

### BRCM COLLEGE OF ENGINEERING & TECHNOLOGY Practical Experiment Instructions Sheet

Exp. Title FLUID MECHANICS- I LAB Syllabus FM-I Semester-4 th Page No. 1 of 1 Internal Marks: 25 L T P External Marks: 25 0 0 2 Total Marks: 50 1. To determine the met centric height of a floating body

### [Prasanna m a*et al., 5(6): July, 2016] ISSN: IC Value: 3.00 Impact Factor: 4.116

IJESRT INTERNATIONAL JOURNAL OF ENGINEERING SCIENCES & RESEARCH TECHNOLOGY NUMERICAL ANALYSIS OF COMPRESSIBLE EFFECT IN THE FLOW METERING BY CLASSICAL VENTURIMETER Prasanna M A *, Dr V Seshadri, Yogesh

### V/ t = 0 p/ t = 0 ρ/ t = 0. V/ s = 0 p/ s = 0 ρ/ s = 0

UNIT III FLOW THROUGH PIPES 1. List the types of fluid flow. Steady and unsteady flow Uniform and non-uniform flow Laminar and Turbulent flow Compressible and incompressible flow Rotational and ir-rotational

### Chapter 3 Bernoulli Equation

1 Bernoulli Equation 3.1 Flow Patterns: Streamlines, Pathlines, Streaklines 1) A streamline, is a line that is everywhere tangent to the velocity vector at a given instant. Examples of streamlines around

### Calibrate Rotameter and Orifice Meter and Explore Reynolds #

CM3215 Fundamentals of Chemical Engineering Laboratory Calibrate Rotameter and Orifice Meter and Explore Reynolds # Extra features! Professor Faith Department of Chemical Engineering Michigan Technological

### CHAPTER (13) FLOW MEASUREMENTS

CHAPTER (13) FLOW MEASUREMENTS 09/12/2010 Dr. Munzer Ebaid 1 Instruments for the Measurements of Flow Rate 1. Direct Methods: Volume or weight measurements. 2. Indirect Methods: Venturi meters, Orifices

### vector H. If O is the point about which moments are desired, the angular moment about O is given:

The angular momentum A control volume analysis can be applied to the angular momentum, by letting B equal to angularmomentum vector H. If O is the point about which moments are desired, the angular moment

### 04/01/1998 Developments in DP Flowmeters By Jesse Yoder

04/01/1998 Developments in DP Flowmeters By Jesse Yoder Developments in DP Flowmeters Improvements in Primary Elements Are Keeping Differential Pressure Flowmeters the First Choice for Many Process Applications

### Flow Measurement in Pipes and Ducts

Flow Measurement in Pipes and Ducts Course No: M04-040 Credit: 4 PDH Harlan H. Bengtson, Ph.D., P.E. Continuing Education and Development, Inc. 9 Greyridge Farm Court Stony Point, NY 10980 P: (877) 322-5800

### MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, AURANGABAD. (M.S.)

MAHATMA GANDHI MISSION S JAWAHARLAL NEHRU ENGINEERING COLLEGE, AURANGABAD. (M.S.) DEPARTMENT OF CIVIL ENGINEERING FLUID MECHANICS I LAB MANUAL Prepared By Prof. L. K. Kokate Lab Incharge Approved By Dr.

### FE Exam Fluids Review October 23, Important Concepts

FE Exam Fluids Review October 3, 013 mportant Concepts Density, specific volume, specific weight, specific gravity (Water 1000 kg/m^3, Air 1. kg/m^3) Meaning & Symbols? Stress, Pressure, Viscosity; Meaning

### Exam #2: Fluid Kinematics and Conservation Laws April 13, 2016, 7:00 p.m. 8:40 p.m. in CE 118

CVEN 311-501 (Socolofsky) Fluid Dynamics Exam #2: Fluid Kinematics and Conservation Laws April 13, 2016, 7:00 p.m. 8:40 p.m. in CE 118 Name: : UIN: : Instructions: Fill in your name and UIN in the space

### Figure 3: Problem 7. (a) 0.9 m (b) 1.8 m (c) 2.7 m (d) 3.6 m

1. For the manometer shown in figure 1, if the absolute pressure at point A is 1.013 10 5 Pa, the absolute pressure at point B is (ρ water =10 3 kg/m 3, ρ Hg =13.56 10 3 kg/m 3, ρ oil = 800kg/m 3 ): (a)

### CEE 3310 Control Volume Analysis, Oct. 10, = dt. sys

CEE 3310 Control Volume Analysis, Oct. 10, 2018 77 3.16 Review First Law of Thermodynamics ( ) de = dt Q Ẇ sys Sign convention: Work done by the surroundings on the system < 0, example, a pump! Work done

### CEE 3310 Control Volume Analysis, Oct. 7, D Steady State Head Form of the Energy Equation P. P 2g + z h f + h p h s.

CEE 3310 Control Volume Analysis, Oct. 7, 2015 81 3.21 Review 1-D Steady State Head Form of the Energy Equation ( ) ( ) 2g + z = 2g + z h f + h p h s out where h f is the friction head loss (which combines

### Physics 3 Summer 1990 Lab 7 - Hydrodynamics

Physics 3 Summer 1990 Lab 7 - Hydrodynamics Theory Consider an ideal liquid, one which is incompressible and which has no internal friction, flowing through pipe of varying cross section as shown in figure

### Head loss coefficient through sharp-edged orifices

Head loss coefficient through sharp-edged orifices Nicolas J. Adam, Giovanni De Cesare and Anton J. Schleiss Laboratory of Hydraulic Constructions, Ecole Polytechnique fédérale de Lausanne, Lausanne, Switzerland

### Study fluid dynamics. Understanding Bernoulli s Equation.

Chapter Objectives Study fluid dynamics. Understanding Bernoulli s Equation. Chapter Outline 1. Fluid Flow. Bernoulli s Equation 3. Viscosity and Turbulence 1. Fluid Flow An ideal fluid is a fluid that

### Visualization of flow pattern over or around immersed objects in open channel flow.

EXPERIMENT SEVEN: FLOW VISUALIZATION AND ANALYSIS I OBJECTIVE OF THE EXPERIMENT: Visualization of flow pattern over or around immersed objects in open channel flow. II THEORY AND EQUATION: Open channel:

### Benha University College of Engineering at Benha Questions For Corrective Final Examination Subject: Fluid Mechanics M 201 May 24/ 2016

Benha University College of Engineering at Benha Questions For Corrective Final Examination Subject: Fluid Mechanics M 01 May 4/ 016 Second year Mech. Time :180 min. Examiner:Dr.Mohamed Elsharnoby Attempt

### UNIT I FLUID PROPERTIES AND STATICS

SIDDHARTH GROUP OF INSTITUTIONS :: PUTTUR Siddharth Nagar, Narayanavanam Road 517583 QUESTION BANK (DESCRIPTIVE) Subject with Code : Fluid Mechanics (16CE106) Year & Sem: II-B.Tech & I-Sem Course & Branch:

### Orifice and Venturi Pipe Flow Meters

Orifice and Venturi Pipe Flow Meters by Harlan H. Bengtson, PhD, P.E. 1. Introduction Your Course Title Here The flow rate of a fluid flowing in a pipe under pressure is measured for a variety of applications,

### Lesson 6 Review of fundamentals: Fluid flow

Lesson 6 Review of fundamentals: Fluid flow The specific objective of this lesson is to conduct a brief review of the fundamentals of fluid flow and present: A general equation for conservation of mass

### FE Fluids Review March 23, 2012 Steve Burian (Civil & Environmental Engineering)

Topic: Fluid Properties 1. If 6 m 3 of oil weighs 47 kn, calculate its specific weight, density, and specific gravity. 2. 10.0 L of an incompressible liquid exert a force of 20 N at the earth s surface.

### Orifice and Venturi Pipe Flow Meters

Orifice and Venturi Pipe Flow Meters For Liquid and Gas Flow by Harlan H. Bengtson, PhD, P.E. 1. Introduction Orifice and Venturi Pipe Flow Meters The flow rate of a fluid flowing in a pipe under pressure

### Instruction Manual. Equipment for Engineering Education

Equipment for Engineering Education Instruction Manual HM15007 Bernoulli s Principle Demonstrator GUNT Gerätebau GmbH PO Box 1125 D-22881 Barsbüttel Germany Phone (040) 670854-0 Fax (040) 670854-42 Instruction

### 53:071 Principles of Hydraulics Laboratory Experiment #3 ANALYSIS OF OPEN-CHANNEL FLOW TRANSITIONS USING THE SPECIFIC ENERGY DIAGRAM

53:071 Principles of Hydraulics Laboratory Experiment #3 ANALYSIS OF OPEN-CHANNEL FLOW TRANSITIONS USING THE SPECIFIC ENERGY DIAGRAM Principle Adaptation of the Bernoulli equation to open-channel flows

### Predictionof discharge coefficient of Venturimeter at low Reynolds numbers by analytical and CFD Method

International Journal of Engineering and Technical Research (IJETR) ISSN: 2321-0869, Volume-3, Issue-5, May 2015 Predictionof discharge coefficient of Venturimeter at low Reynolds numbers by analytical

### Water Circuit Lab. The pressure drop along a straight pipe segment can be calculated using the following set of equations:

Water Circuit Lab When a fluid flows in a conduit, there is friction between the flowing fluid and the pipe walls. The result of this friction is a net loss of energy in the flowing fluid. The fluid pressure

### Estimation of Flow Meter Losses

1 Estimation of Flow Meter Losses T. Littleton Abstract. In this article, experimental data concerning flow rates, discharge coefficients, pressure losses, and energy losses of water flowing through flow

### Experimental Study on Port to Channel Flow Distribution of Plate Heat Exchangers

Proceedings of Fifth International Conference on Enhanced, Compact and Ultra-Compact Heat Exchangers: Science, Engineering and Technology, Eds. R.K. Shah, M. Ishizuka, T.M. Rudy, and V.V. Wadekar, Engineering

### In which of the following scenarios is applying the following form of Bernoulli s equation: steady, inviscid, uniform stream of water. Ma = 0.

bernoulli_11 In which of the following scenarios is applying the following form of Bernoulli s equation: p V z constant! g + g + = from point 1 to point valid? a. 1 stagnant column of water steady, inviscid,

### ISO 9906 INTERNATIONAL STANDARD. Rotodynamic pumps Hydraulic performance acceptance tests Grades 1 and 2

INTERNATIONAL STANDARD ISO 9906 First edition 1999-1-15 Rotodynamic pumps Hydraulic performance acceptance tests Grades 1 and Pompes rotodynamiques Essais de fonctionnement hydraulique pour la réception

### Chapter 5 Control Volume Approach and Continuity Equation

Chapter 5 Control Volume Approach and Continuity Equation Lagrangian and Eulerian Approach To evaluate the pressure and velocities at arbitrary locations in a flow field. The flow into a sudden contraction,

### Experimental Investigations on the Local Distribution of wall static pressure coefficient Due To an Impinging Slot Air Jet on a Confined Rough Surface

Experimental Investigations on the Local Distribution of wall static pressure coefficient Due To an Impinging Slot Air Jet on a Confined Rough Surface 1 Adimurthy. M 1 BLDEA s VP DR. P G Halakatti college

### 3.25 Pressure form of Bernoulli Equation

CEE 3310 Control Volume Analysis, Oct 3, 2012 83 3.24 Review The Energy Equation Q Ẇshaft = d dt CV ) (û + v2 2 + gz ρ d + (û + v2 CS 2 + gz + ) ρ( v n) da ρ where Q is the heat energy transfer rate, Ẇ

### CALCULATION OF COEFFICIENT OF DISCHARGE OF ORIFICE PLATE AND FLOW NOZZLE

CALCULATION OF COEFFICIENT OF DISCHARGE OF ORIFICE PLATE AND FLOW NOZZLE A.NAGABHUSHANA RAO, M.PADMANABHAM ABSTRACT An orifice plate is a device used for measuring flow rate. Either a volumetric or mass

### FLUID MECHANICS. Chapter 3 Elementary Fluid Dynamics - The Bernoulli Equation

FLUID MECHANICS Chapter 3 Elementary Fluid Dynamics - The Bernoulli Equation CHAP 3. ELEMENTARY FLUID DYNAMICS - THE BERNOULLI EQUATION CONTENTS 3. Newton s Second Law 3. F = ma along a Streamline 3.3

### An Experimental Investigation to Control the Flow Emerging From a Wide Angle Diffuser

IOSR Journal of Engineering (IOSRJEN) ISSN: 5-3 ISBN: 878-879 PP 7-3 National Symposium on engineering and Research An Experimental Investigation to Control the Flow Emerging From a Wide Angle Diffuser

### Chapter (3) Water Flow in Pipes

Chapter (3) Water Flow in Pipes Water Flow in Pipes Bernoulli Equation Recall fluid mechanics course, the Bernoulli equation is: P 1 ρg + v 1 g + z 1 = P ρg + v g + z h P + h T + h L Here, we want to study

### 57:020 Mechanics of Fluids and Transfer Processes CONSERVATION OF MASS, LINEAR MOMENTUM, AND ENERGY IN A SLUICE GATE FLOW. dt dt. d ( momentum.

57: Mechani of Fluids and Transfer Processes CONSERVATION OF MASS, LINEAR MOMENTUM, AND ENERGY IN A SLUICE GATE FLOW Purpose To measure the total piezometric pressure at various locations along a vertical

### Given Find Water Properties

Venturi Example Given: A venturi is to be used to measure a 50 gpm flow of 70 F water in a 4-in ID pipe. Find: Select a throat diameter that provides Re d > 00,000 in the throat, and determine what differential

### MECHANICAL PROPERTIES OF FLUIDS:

Important Definitions: MECHANICAL PROPERTIES OF FLUIDS: Fluid: A substance that can flow is called Fluid Both liquids and gases are fluids Pressure: The normal force acting per unit area of a surface is

### If a stream of uniform velocity flows into a blunt body, the stream lines take a pattern similar to this: Streamlines around a blunt body

Venturimeter & Orificemeter ELEMENTARY HYDRAULICS National Certificate in Technology (Civil Engineering) Chapter 5 Applications of the Bernoulli Equation The Bernoulli equation can be applied to a great

### Chapter Four fluid flow mass, energy, Bernoulli and momentum

4-1Conservation of Mass Principle Consider a control volume of arbitrary shape, as shown in Fig (4-1). Figure (4-1): the differential control volume and differential control volume (Total mass entering

### ME411 Engineering Measurement & Instrumentation. Winter 2017 Lecture 11

ME411 Engineering Measurement & Instrumentation Winter 2017 Lecture 11 1 Flow Measurement Identify an effect that depends on flow rate Process control requires accurate measurement of flow control Mixing

### An Essential Requirement in CV Based Industrial Appliances.

Measurement of Flow P M V Subbarao Professor Mechanical Engineering Department An Essential Requirement in CV Based Industrial Appliances. Mathematics of Flow Rate The Scalar Product of two vectors, namely

### Prediction of Performance Characteristics of Orifice Plate Assembly for Non-Standard Conditions Using CFD

International Journal of Engineering and Technical Research (IJETR) ISSN: 2321-0869, Volume-3, Issue-5, May 2015 Prediction of Performance Characteristics of Orifice Plate Assembly for Non-Standard Conditions

### In steady flow the velocity of the fluid particles at any point is constant as time passes.

Chapter 10 Fluids Fluids in Motion In steady flow the velocity of the fluid particles at any point is constant as time passes. Unsteady flow exists whenever the velocity of the fluid particles at a point

### FLOW MEASUREMENT. INC 102 Fundamental of Instrumentation and Process Control 2/2560

FLOW MEASUREMENT INC 102 Fundamental of Instrumentation and Process Control 2/2560 TABLE OF CONTENTS A. INTRODUCTION B. LOCAL FLOW MEASUREMENT B.1 Particle Image Velocimetry (PIV) B.2 Laser doppler anemometry

### Evaluation of the Effect of Relative Humidity of Air on the Coefficients of Critical Flow Venturi Nozzles

Evaluation of the Effect of Relative Humidity of Air on the Coefficients of Critical Flow Venturi Nozzles K. Chahine and M. Ballico National Measurement Institute, Australia P O Box 264, Lindfield, NSW

### COURSE CODE : 3072 COURSE CATEGORY : B PERIODS/ WEEK : 5 PERIODS/ SEMESTER : 75 CREDIT : 5 TIME SCHEDULE

COURSE TITLE : FLUID MECHANICS COURSE CODE : 307 COURSE CATEGORY : B PERIODS/ WEEK : 5 PERIODS/ SEMESTER : 75 CREDIT : 5 TIME SCHEDULE MODULE TOPIC PERIOD 1 Properties of Fluids 0 Fluid Friction and Flow

### Fluid Dynamics Exercises and questions for the course

Fluid Dynamics Exercises and questions for the course January 15, 2014 A two dimensional flow field characterised by the following velocity components in polar coordinates is called a free vortex: u r