# Flow Measurement in Pipes and Ducts COURSE CONTENT

Size: px
Start display at page:

Transcription

1

2 Flow Measurement in Pipes and Ducts Dr. Harlan H. Bengtson, P.E. COURSE CONTENT 1. Introduction This course is about measurement of the flow rate of a fluid flowing under pressure in a closed conduit. The closed conduit is often circular, but also may be square or rectangular (such as a heating duct) or any other shape. The other major category of flow is open channel flow, which is the flow of a liquid with a free surface open to atmospheric pressure. Measurement of the flow rate of a fluid flowing under pressure, is carried out for a variety of purposes, such as billing for water supply to homes or businesses or, for monitoring or process control of a wide variety of industrial processes, which involve flowing fluids. Several categories of pipe flow measurement devices will be described and discussed, including some associated calculations. 1

3 2. Learning Objectives At the conclusion of this course, the student will Be able to calculate flow rate from measured pressure difference, fluid properties, and meter parameters, using the provided equations for venturi, orifice, and flow nozzle meters. Be able to determine which type of ISO standard pressure tap locations are being used for a given orifice meter. Be able to calculate the orifice coefficient, C o, for specified orifice and pipe diameters, pressure tap locations and fluid properties. Be able to estimate the density of a specified gas at specified temperature and pressure using the Ideal Gas Equation. Be able to calculate the velocity of a fluid for given pitot tube reading and fluid density. Know the general configuration and principle of operation of rotameters and positive displacement, electromagnetic, target, turbine, vortex, and ultrasonic meters. Know recommended applications for each of the type of flow meter discussed in this course. Be familiar with the general characteristics of the types of flow meters discussed in this course, as summarized in Table 2 in the course content. 2

4 3. Types of Pipe Flow Measurement Devices The types of pipe flow measuring devices to be discussed in this course are as follows: i) Differential pressure flow meters a) Venturi meter b) Orifice meter c) Flow nozzle meter ii) Velocity flow meters pitot / pitot-static tubes iii) Variable area flow meters - rotameters iv) Positive displacement flow meters v) Miscellaneous a) Electromagnetic flow meters b) Target flow meters d) Turbine flow meters e) Vortex flow meters f) Ultrasonic flow meters 3

5 4. Differential Pressure Flow meters Three types of commonly used differential pressure flow meters are the orifice meter, the venturi meter, and the flow nozzle meter. These three all function by introducing a reduced area through which the fluid must flow. The decrease in area causes an increase in velocity, which in turn results in a decrease of pressure. With these flow meters, the pressure difference between the point of maximum velocity (minimum pressure) and the undisturbed upstream flow is measured and can be correlated with flow rate. Using the principles of conservation of mass (the continuity equation) and the conservation of energy (the energy equation without friction or Bernoulli equation), the following equation can be derived for ideal flow between the upstream, undisturbed flow (subscript 1) and the downstream conditions where the flow area is constricted (subscript 2): Where: Q ideal = ideal flow rate (neglecting viscosity and other friction effects), cfs A 2 = constricted cross-sectional area normal to flow, ft 2 P 1 = upstream (undisturbed) pressure in pipe, lb/ft 2 P 2 = pressure in pipe where flow area is constricted to A 2, lb/ft 2 4

6 = D 2 /D 1 = (diam. at A 2 )/(pipe diam.) = fluid density, slugs/ft 3 A discharge coefficient, C, is typically put into equation (1) to account for friction and any other non-ideal factors, giving the following general equation for differential pressure meters: Where: Q = flow rate through the pipe and meter, cfs C = discharge coefficient, dimensionless All other parameters are as defined above Each of the three types of differential pressure flow meters will now be considered separately. Venturi Meter: Fluid enters a venturi meter through a converging cone of angle 15 o to 20 o. It then passes through the throat, which has the minimum crosssectional area, maximum velocity, and minimum pressure in the meter. The fluid then slows down through a diverging cone of angle 5 o to 7 o, for the transition back to the full pipe diameter. Figure 1 shows the shape of a typical venturi meter and the parameters defined above as applied to this type of meter. D 2 is the diameter of the throat and P 2 is the pressure at the throat. D 1 and P 1 are in the pipe before entering the converging portion of the meter. 5

7 Figure 1. Venturi Meter Parameters Due to the smooth transition to the throat and gradual transition back to full pipe diameter, the head loss through a venturi meter is quite low and the discharge coefficient is quite high. For a venturi meter the discharge coefficient is typically called the venturi coefficient, C v, giving the following equation for a venturi meter: The value of the venturi coefficient, C v, will typically range from 0.95 to nearly one. In ISO 5167 ( ISO :2003 see reference #2 for this course), C v is given as for cast iron or machined venturi meters and for welded sheet 6

8 between 2 x 10 5 and Information on the venturi coefficient will typically be provided by venturi meter manufacturers. Example #1: Water at 50 o F is flowing through a venturi meter with a 2 inch 0.99 for this meter under these flow conditions. What is the flow rate through the meter if the pressure difference, P 1 P 2, is measured as 8 inches of Hg. Solution: The density of water in the temperature range from 32 o to 70 o F is 1.94 slugs/ft 3, to three significant figures, so that value will be used here. A 2 = D 2 2 /4 = ft 2. = 2/4 = 0.5. Converting the pressure difference to lb/ft 2 : P 1 P 2 = (8 in Hg)(70.73 lb/ft 2 /in Hg) = lb/ft 2. Substituting all of these values into equation (3): Orifice Meter: The orifice meter is the simplest of the three differential pressure flow meters. It consists of a circular plate with a hole in the middle, typically held in place between pipe flanges, as shown in figure 2. 7

### Flow Measurement in Pipes and Ducts COURSE CONTENT

Flow Measurement in Pipes and Ducts Dr. Harlan H. Bengtson, P.E. COURSE CONTENT 1. Introduction This course is about measurement of the flow rate of a fluid flowing under pressure in a closed conduit.

### Orifice and Venturi Pipe Flow Meters

Orifice and Venturi Pipe Flow Meters by Harlan H. Bengtson, PhD, P.E. 1. Introduction Your Course Title Here The flow rate of a fluid flowing in a pipe under pressure is measured for a variety of applications,

### Orifice and Venturi Pipe Flow Meters

Orifice and Venturi Pipe Flow Meters For Liquid and Gas Flow by Harlan H. Bengtson, PhD, P.E. 1. Introduction Orifice and Venturi Pipe Flow Meters The flow rate of a fluid flowing in a pipe under pressure

### Flow Measurement in Pipes and Ducts

Flow Measurement in Pipes and Ducts Course No: M04-040 Credit: 4 PDH Harlan H. Bengtson, Ph.D., P.E. Continuing Education and Development, Inc. 9 Greyridge Farm Court Stony Point, NY 10980 P: (877) 322-5800

### Experiment No.4: Flow through Venturi meter. Background and Theory

Experiment No.4: Flow through Venturi meter Background and Theory Introduction Flow meters are used in the industry to measure the volumetric flow rate of fluids. Differential pressure type flow meters

### CHAPTER (13) FLOW MEASUREMENTS

CHAPTER (13) FLOW MEASUREMENTS 09/12/2010 Dr. Munzer Ebaid 1 Instruments for the Measurements of Flow Rate 1. Direct Methods: Volume or weight measurements. 2. Indirect Methods: Venturi meters, Orifices

### Lecture23. Flowmeter Design.

Lecture23 Flowmeter Design. Contents of lecture Design of flowmeter Principles of flow measurement; i) Venturi and ii) Orifice meter and nozzle Relationship between flow rate and pressure drop Relation

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### 5 ENERGY EQUATION OF FLUID MOTION

5 ENERGY EQUATION OF FLUID MOTION 5.1 Introduction In order to develop the equations that describe a flow, it is assumed that fluids are subject to certain fundamental laws of physics. The pertinent laws

### Lecture 13 Flow Measurement in Pipes. I. Introduction

Lecture 13 Flow Measurement in Pipes I. Introduction There are a wide variety of methods for measuring discharge and velocity in pipes, or closed conduits Many of these methods can provide very accurate

### Experiment (4): Flow measurement

Experiment (4): Flow measurement Introduction: The flow measuring apparatus is used to familiarize the students with typical methods of flow measurement of an incompressible fluid and, at the same time

### FE Fluids Review March 23, 2012 Steve Burian (Civil & Environmental Engineering)

Topic: Fluid Properties 1. If 6 m 3 of oil weighs 47 kn, calculate its specific weight, density, and specific gravity. 2. 10.0 L of an incompressible liquid exert a force of 20 N at the earth s surface.

### Mass of fluid leaving per unit time

5 ENERGY EQUATION OF FLUID MOTION 5.1 Eulerian Approach & Control Volume In order to develop the equations that describe a flow, it is assumed that fluids are subject to certain fundamental laws of physics.

### FLOW MEASUREMENT IN PIPES EXPERIMENT

University of Leicester Engineering Department FLOW MEASUREMENT IN PIPES EXPERIMENT Page 1 FORMAL LABORATORY REPORT Name of the experiment: FLOW MEASUREMENT IN PIPES Author: Apollin nana chaazou Partner

### 04/01/1998 Developments in DP Flowmeters By Jesse Yoder

04/01/1998 Developments in DP Flowmeters By Jesse Yoder Developments in DP Flowmeters Improvements in Primary Elements Are Keeping Differential Pressure Flowmeters the First Choice for Many Process Applications

### If a stream of uniform velocity flows into a blunt body, the stream lines take a pattern similar to this: Streamlines around a blunt body

Venturimeter & Orificemeter ELEMENTARY HYDRAULICS National Certificate in Technology (Civil Engineering) Chapter 5 Applications of the Bernoulli Equation The Bernoulli equation can be applied to a great

### Predictionof discharge coefficient of Venturimeter at low Reynolds numbers by analytical and CFD Method

International Journal of Engineering and Technical Research (IJETR) ISSN: 2321-0869, Volume-3, Issue-5, May 2015 Predictionof discharge coefficient of Venturimeter at low Reynolds numbers by analytical

### Measurements using Bernoulli s equation

An Internet Book on Fluid Dynamics Measurements using Bernoulli s equation Many fluid measurement devices and techniques are based on Bernoulli s equation and we list them here with analysis and discussion.

### COURSE CODE : 3072 COURSE CATEGORY : B PERIODS/ WEEK : 5 PERIODS/ SEMESTER : 75 CREDIT : 5 TIME SCHEDULE

COURSE TITLE : FLUID MECHANICS COURSE CODE : 307 COURSE CATEGORY : B PERIODS/ WEEK : 5 PERIODS/ SEMESTER : 75 CREDIT : 5 TIME SCHEDULE MODULE TOPIC PERIOD 1 Properties of Fluids 0 Fluid Friction and Flow

### Lecture 24. Design of flow meters

Lecture 24 Design of flow meters Contents Exercise 1 Exercise 2 Exercise 3 Key Words: Fluid flow, Macroscopic Balance, Frictional Losses, Turbulent Flow, Venturimeter, Orifice Meter, Pitot Tube Exercise

### SELECCIÓN DE CAUDALIMETROS. Tablas de Fabricantes

SELECCIÓN DE CAUDALIMETROS Tablas de Fabricantes Flow Meter Selection Table http://www.instrumart.com/content.aspxcontentid=219 Página 1 de 1 17/09/2007 KEY Best for this application OK with some exceptions

### CHAPTER 3 BASIC EQUATIONS IN FLUID MECHANICS NOOR ALIZA AHMAD

CHAPTER 3 BASIC EQUATIONS IN FLUID MECHANICS 1 INTRODUCTION Flow often referred as an ideal fluid. We presume that such a fluid has no viscosity. However, this is an idealized situation that does not exist.

### vector H. If O is the point about which moments are desired, the angular moment about O is given:

The angular momentum A control volume analysis can be applied to the angular momentum, by letting B equal to angularmomentum vector H. If O is the point about which moments are desired, the angular moment

### Rate of Flow Quantity of fluid passing through any section (area) per unit time

Kinematics of Fluid Flow Kinematics is the science which deals with study of motion of liquids without considering the forces causing the motion. Rate of Flow Quantity of fluid passing through any section

### Flow rate and mass flow rate

EEN-E1040 Measurement and control of energy systems Flow measurements / 14 Sep 2017 WELCOME! v. 01 / T. Paloposki Flow rate and mass flow rate Consider the system shown here 1 Volume flow rate through

### The Expansibility Factor Equations in ISO and ISO : Do They Deliver What They Promise?

The Expansibility Factor Equations in ISO 567-2 and ISO 567-4: Do They Deliver What They Promise? Michael Reader-Harris, NEL INTRODUCTION The expansibility factor equations in ISO 567-2:2003 [] and ISO

### Mechanical Measurements and Metrology Prof. S. P. Venkateshan Department of Mechanical Engineering Indian Institute of Technology, Madras

Mechanical Measurements and Metrology Prof. S. P. Venkateshan Department of Mechanical Engineering Indian Institute of Technology, Madras Module - 3 Lecture - 33 Measurement of Volume and Mass Flow Rate

### MCE380: Measurements and Instrumentation Lab

MCE380: Measurements and Instrumentation Lab Chapter 8: Flow Measurements Topics: Basic Flow Equations Flow Obstruction Meters Positive Displacement Flowmeters Other Methods Holman, Ch. 7 Cleveland State

### Chapter 4 DYNAMICS OF FLUID FLOW

Faculty Of Engineering at Shobra nd Year Civil - 016 Chapter 4 DYNAMICS OF FLUID FLOW 4-1 Types of Energy 4- Euler s Equation 4-3 Bernoulli s Equation 4-4 Total Energy Line (TEL) and Hydraulic Grade Line

### Chapter 3 Bernoulli Equation

1 Bernoulli Equation 3.1 Flow Patterns: Streamlines, Pathlines, Streaklines 1) A streamline, is a line that is everywhere tangent to the velocity vector at a given instant. Examples of streamlines around

### ME411 Engineering Measurement & Instrumentation. Winter 2017 Lecture 11

ME411 Engineering Measurement & Instrumentation Winter 2017 Lecture 11 1 Flow Measurement Identify an effect that depends on flow rate Process control requires accurate measurement of flow control Mixing

### EXPERIMENT No.1 FLOW MEASUREMENT BY ORIFICEMETER

EXPERIMENT No.1 FLOW MEASUREMENT BY ORIFICEMETER 1.1 AIM: To determine the co-efficient of discharge of the orifice meter 1.2 EQUIPMENTS REQUIRED: Orifice meter test rig, Stopwatch 1.3 PREPARATION 1.3.1

### Engineers Edge, LLC PDH & Professional Training

510 N. Crosslane Rd. Monroe, Georgia 30656 (770) 266-6915 fax (678) 643-1758 Engineers Edge, LLC PDH & Professional Training Copyright, All Rights Reserved Engineers Edge, LLC Pipe Flow-Friction Factor

### AEROSPACE ENGINEERING DEPARTMENT. Second Year - Second Term ( ) Fluid Mechanics & Gas Dynamics

AEROSPACE ENGINEERING DEPARTMENT Second Year - Second Term (2008-2009) Fluid Mechanics & Gas Dynamics Similitude,Dimensional Analysis &Modeling (1) [7.2R*] Some common variables in fluid mechanics include:

### ABSTRACT I. INTRODUCTION

2016 IJSRSET Volume 2 Issue 4 Print ISSN : 2395-1990 Online ISSN : 2394-4099 Themed Section: Engineering and Technology Analysis of Compressible Effect in the Flow Metering By Orifice Plate Using Prasanna

### New Website: Mr. Peterson s Address:

Brad Peterson, P.E. New Website: http://njut2009fall.weebly.com Mr. Peterson s Email Address: bradpeterson@engineer.com Lesson 1, Properties of Fluids, 2009 Sept 04, Rev Sept 18 Lesson 2, Fluid Statics,

### ISO 9906 INTERNATIONAL STANDARD. Rotodynamic pumps Hydraulic performance acceptance tests Grades 1 and 2

INTERNATIONAL STANDARD ISO 9906 First edition 1999-1-15 Rotodynamic pumps Hydraulic performance acceptance tests Grades 1 and Pompes rotodynamiques Essais de fonctionnement hydraulique pour la réception

### Flow Measurement: Physical principle employed in various types of flow meters. PART 4 PREPARED BY: ESO OLUWATOBILOBA

Flow Measurement: Physical principle employed in various types of flow meters. PART 4 PREPARED BY: ESO OLUWATOBILOBA Pressure Differential Flow Meters In a differential pressure drop device the flow is

### ENGINEERING FLUID MECHANICS. CHAPTER 1 Properties of Fluids

CHAPTER 1 Properties of Fluids ENGINEERING FLUID MECHANICS 1.1 Introduction 1.2 Development of Fluid Mechanics 1.3 Units of Measurement (SI units) 1.4 Mass, Density, Specific Weight, Specific Volume, Specific

### Fundamentals of Flow Metering

Technical Data Sheet 00816-0100-3031 Fundamentals of Flow Metering A reliable flow indication is dependent upon the correct measurement of A and V. If, for example, air bubbles are present in the fluid,

### POWER UNDERSTANDING MEASUREMENT UNCERTAINTY IN DP FLOW DEVICES

Proceedings of the ASME 2014 Power Conference POWER2014 July 28-31, 2014, Baltimore, Maryland, USA POWER2014-32205 UNDERSTANDING MEASUREMENT UNCERTAINTY IN DP FLOW DEVICES Michael S. Hering Rosemount DP

### 405 Compact Orifice Series and 1595 Conditioning Orifice Plate Flow Test Data Book and Flow Handbook

405 Compact Orifice Series and 1595 Conditioning Orifice Plate Flow Test Book and Flow Handbook www.rosemount.com 405 and 1595 405 Compact Orifice Series and 1595 Conditioning Orifice Plate Flow Test

### MEDIDAS EN FLUIDOS. INEL 5205 Instrumentación. Monday, November 19, 12

MEDIDAS EN FLUIDOS INEL 5205 Instrumentación SENSORES DE PRESIÓN Unidades de Presión 1 psi! 1 lbf/in 2 = 6,894.76! Pa = 6,894.76!newton/m 2 = 68.948 10 3!bar (1 bar = 10 5 Pa) = 68.046 10 3!atm = 51.715

### Signature: (Note that unsigned exams will be given a score of zero.)

Neatly print your name: Signature: (Note that unsigned exams will be given a score of zero.) Circle your lecture section (-1 point if not circled, or circled incorrectly): Prof. Dabiri Prof. Wassgren Prof.

### Lecture 3 The energy equation

Lecture 3 The energy equation Dr Tim Gough: t.gough@bradford.ac.uk General information Lab groups now assigned Timetable up to week 6 published Is there anyone not yet on the list? Week 3 Week 4 Week 5

### Volume and Mass Flow Rate Measurement Author: John M. Cimbala, Penn State University Latest revision: 07 December 2007

Volume and Mass Flow Rate Measurement Author: John M. Cimbala, Penn State University Latest revision: 07 ecember 2007 Introduction and notation In many engineering applications, either mass flow rate or

### Alden Test Results On AMITY Insert Venturi

Alden Test Results On AMITY Insert Venturi Richard W. Miller PhD, PE A laboratory test was performed on an AMITY designed insert Venturi at the Alden on October 27, 2009 to obtain the coefficient of discharge

### Flow Monitoring Technologies in Oil and Gas. Don Ford, Technical Specialist Spartan Controls Ltd.

Flow Monitoring Technologies in Oil and Gas Don Ford, Technical Specialist Spartan Controls Ltd. Agenda Flow technologies at a glance Oil and Gas at a glance Select Flow Technologies in Depth Differential

### ME332 FLUID MECHANICS LABORATORY (PART I)

ME332 FLUID MECHANICS LABORATORY (PART I) Mihir Sen Department of Aerospace and Mechanical Engineering University of Notre Dame Notre Dame, IN 46556 Version: January 14, 2002 Contents Unit 1: Hydrostatics

### FLOW MEASUREMENT. INC 102 Fundamental of Instrumentation and Process Control 2/2560

FLOW MEASUREMENT INC 102 Fundamental of Instrumentation and Process Control 2/2560 TABLE OF CONTENTS A. INTRODUCTION B. LOCAL FLOW MEASUREMENT B.1 Particle Image Velocimetry (PIV) B.2 Laser doppler anemometry

### 2.The lines that are tangent to the velocity vectors throughout the flow field are called steady flow lines. True or False A. True B.

CHAPTER 03 1. Write Newton's second law of motion. YOUR ANSWER: F = ma 2.The lines that are tangent to the velocity vectors throughout the flow field are called steady flow lines. True or False 3.Streamwise

### VENTURIMETER EXPERIMENT

ENTURIMETER EXERIMENT. OBJECTİE The main objectives of this experiment is to obtain the coefficient of discharge from experimental data by utilizing venturi meter and, also the relationship between Reynolds

### Computation of pressure loss for differential producing flowmeters

XXVI. ASR '00 Seminar, Instruments and Control, Ostrava, April 6-7, 00 Paper 5 Computation of pressure loss for differential producing flowmeters ORLÍKOVÁ, Soňa Ing., Ústav automatizace a měřicí techniky,

### 405 Compact Orifice Series and 1595 Conditioning Orifice Plate Flow Test Data Book and Flow Handbook

Reference Manual 405 Compact Orifice Series and 1595 Conditioning Orifice Plate Flow Test Book and Flow Handbook www.rosemount.com Reference Manual 405 and 1595 405 Compact Orifice Series and 1595 Conditioning

### An Essential Requirement in CV Based Industrial Appliances.

Measurement of Flow P M V Subbarao Professor Mechanical Engineering Department An Essential Requirement in CV Based Industrial Appliances. Mathematics of Flow Rate The Scalar Product of two vectors, namely

### INSTITUTE OF AERONAUTICAL ENGINEERING Dundigal, Hyderabad AERONAUTICAL ENGINEERING QUESTION BANK : AERONAUTICAL ENGINEERING.

Course Name Course Code Class Branch INSTITUTE OF AERONAUTICAL ENGINEERING Dundigal, Hyderabad - 00 0 AERONAUTICAL ENGINEERING : Mechanics of Fluids : A00 : II-I- B. Tech Year : 0 0 Course Coordinator

### Hydraulics and hydrology

Hydraulics and hydrology - project exercises - Class 4 and 5 Pipe flow Discharge (Q) (called also as the volume flow rate) is the volume of fluid that passes through an area per unit time. The discharge

### PREVIEW COPY. Flow Measurement. Table of Contents. Properties of Fluid Flow...3. Primary Measuring Devices Secondary Measuring Devices...

Flow Measurement Table of Contents Lesson One Lesson Two Lesson Three Lesson Four Properties of Fluid Flow...3 Primary Measuring Devices...19 Secondary Measuring Devices...35 Variable-Area Instruments...51

### Pipe Flow/Friction Factor Calculations using Excel Spreadsheets

Pipe Flow/Friction Factor Calculations using Excel Spreadsheets Harlan H. Bengtson, PE, PhD Emeritus Professor of Civil Engineering Southern Illinois University Edwardsville Table of Contents Introduction

### Instrumentation & Data Acquisition Systems

Instrumentation & Data Acquisition Systems Section 5 -Flow Robert W. Harrison, PE Bob@TheHarrisonHouse.com Made in USA 1 The Perfect Flowmeter Infinite rangeability / turndown with negligible uncertainty

### PART II. Fluid Mechanics Pressure. Fluid Mechanics Pressure. Fluid Mechanics Specific Gravity. Some applications of fluid mechanics

ART II Some applications of fluid mechanics Fluid Mechanics ressure ressure = F/A Units: Newton's per square meter, Nm -, kgm - s - The same unit is also known as a ascal, a, i.e. a = Nm - ) English units:

### Detailed Outline, M E 320 Fluid Flow, Spring Semester 2015

Detailed Outline, M E 320 Fluid Flow, Spring Semester 2015 I. Introduction (Chapters 1 and 2) A. What is Fluid Mechanics? 1. What is a fluid? 2. What is mechanics? B. Classification of Fluid Flows 1. Viscous

### EXPERIMENT NO. 4 CALIBRATION OF AN ORIFICE PLATE FLOWMETER MECHANICAL ENGINEERING DEPARTMENT KING SAUD UNIVERSITY RIYADH

EXPERIMENT NO. 4 CALIBRATION OF AN ORIFICE PLATE FLOWMETER MECHANICAL ENGINEERING DEPARTMENT KING SAUD UNIVERSITY RIYADH Submitted By: ABDULLAH IBN ABDULRAHMAN ID: 13456789 GROUP A EXPERIMENT PERFORMED

### Chapter Four fluid flow mass, energy, Bernoulli and momentum

4-1Conservation of Mass Principle Consider a control volume of arbitrary shape, as shown in Fig (4-1). Figure (4-1): the differential control volume and differential control volume (Total mass entering

### FINAL DRAFT: October 25, 2010 API TECHNICAL REPORT FUEL GAS MEASUREMENT. FINAL DRAFT: October 25, 2010 Page 0

FINAL DRAFT: October 25, 2010 API TECHNICAL REPORT FUEL GAS MEASUREMENT FINAL DRAFT: October 25, 2010 Page 0 Foreword Add API boiler plate FINAL DRAFT: October 25, 2010 Page 1 TABLE OF CONTENTS Introduction...

### 1-Reynold s Experiment

Lect.No.8 2 nd Semester Flow Dynamics in Closed Conduit (Pipe Flow) 1 of 21 The flow in closed conduit ( flow in pipe ) is differ from this occur in open channel where the flow in pipe is at a pressure

### 405 Compact Orifice Series and 1595 Conditioning Orifice Plate Flow Test Data Book and Flow Handbook

405 Compact Orifice Series and 1595 Conditioning Orifice Plate Flow Test Book and Flow Handbook www.rosemount.com 405 Compact Orifice Series and 1595 Conditioning Orifice Plate Flow Test Book NOTICE Read

### Flow Introduction. 1.1 Introduction Objectives History of DP Flow Pressure DP Flow

1 DP Flow Introduction TOPIC PAGE 1.1 Introduction... 10 1.2 Objectives... 10 1.3 History of DP Flow... 10 1.4 Pressure... 11 1.5 DP Flow 101... 12 1.6 DP Flow Measurement Applications... 15 1.7 Flowmeter

### Discharge measurements

Discharge measurements DISCHARGE MEASUREMENTS Suitability of methods and equipment for measurements where, under what conditions and for what reason discharge is measured determination of discharge in

### Atmospheric pressure. 9 ft. 6 ft

Name CEE 4 Final Exam, Aut 00; Answer all questions; 145 points total. Some information that might be helpful is provided below. A Moody diagram is printed on the last page. For water at 0 o C (68 o F):

### Lesson 6 Review of fundamentals: Fluid flow

Lesson 6 Review of fundamentals: Fluid flow The specific objective of this lesson is to conduct a brief review of the fundamentals of fluid flow and present: A general equation for conservation of mass

### Roll No. :... Invigilator s Signature :.. CS/B.Tech (EIE)/SEM-5/EI-501/ INDUSTRIAL INSTRUMENTATION II

Name : Roll No. :.... Invigilator s Signature :.. CS/B.Tech (EIE)/SEM-5/EI-501/2011-12 2011 INDUSTRIAL INSTRUMENTATION II Time Allotted : 3 Hours Full Marks : 70 The figures in the margin indicate full

### Laboratory exercise 1: Open channel flow measurement

Chapter 1 Laboratory exercise 1: Open channel flow measurement Laboratory exercise Open channel flow measurement is placed on the Faculty of Civil and Geodetic Engineering, on Department of Environmental

### 9.1 Introduction. 9.2 Fluid Flow. CHAPTER 9 Flow

CHAPTER 9 9.1 Introduction The accurate measurement of fluid flow is very important in many industrial applications. Optimum performance of many processes requires specific flow rates. The cost of many

### Objectives. Conservation of mass principle: Mass Equation The Bernoulli equation Conservation of energy principle: Energy equation

Objectives Conservation of mass principle: Mass Equation The Bernoulli equation Conservation of energy principle: Energy equation Conservation of Mass Conservation of Mass Mass, like energy, is a conserved

### [Prasanna m a*et al., 5(6): July, 2016] ISSN: IC Value: 3.00 Impact Factor: 4.116

IJESRT INTERNATIONAL JOURNAL OF ENGINEERING SCIENCES & RESEARCH TECHNOLOGY NUMERICAL ANALYSIS OF COMPRESSIBLE EFFECT IN THE FLOW METERING BY CLASSICAL VENTURIMETER Prasanna M A *, Dr V Seshadri, Yogesh

### 2 Internal Fluid Flow

Internal Fluid Flow.1 Definitions Fluid Dynamics The study of fluids in motion. Static Pressure The pressure at a given point exerted by the static head of the fluid present directly above that point.

### Final 1. (25) 2. (10) 3. (10) 4. (10) 5. (10) 6. (10) TOTAL = HW = % MIDTERM = % FINAL = % COURSE GRADE =

MAE101B: Advanced Fluid Mechanics Winter Quarter 2017 http://web.eng.ucsd.edu/~sgls/mae101b_2017/ Name: Final This is a three hour open-book exam. Please put your name on the top sheet of the exam. Answer

### DETERMINATION OF DISCHARGE AND HEAD LOSS USING A FLOW-MEASURING APPARATUS

DETERMINATION OF DISCHARGE AND HEAD LOSS USING A FLOW-MEASURING APPARATUS 1. INTRODUCTION Through use of the Flow-Measuring Apparatus, this experiment is designed to accustom students to typical methods

### F L U I D S Y S T E M D Y N A M I C S

F L U I D S Y S T E M D Y N A M I C S T he proper design, construction, operation, and maintenance of fluid systems requires understanding of the principles which govern them. These principles include

### Closed duct flows are full of fluid, have no free surface within, and are driven by a pressure gradient along the duct axis.

OPEN CHANNEL FLOW Open channel flow is a flow of liquid, basically water in a conduit with a free surface. The open channel flows are driven by gravity alone, and the pressure gradient at the atmospheric

### SCHOOL OF CHEMICAL ENGINEERING FACULTY OF ENGINEERING AND TECHNOLOGY SRM UNIVERSITY COURSE PLAN

SCHOOL OF CHEMICAL ENGINEERING FACULTY OF ENGINEERING AND TECHNOLOGY SRM UNIVERSITY COURSE PLAN Course code : CH0317 Course Title : Momentum Transfer Semester : V Course Time : July Nov 2011 Required Text

### 6.1 Momentum Equation for Frictionless Flow: Euler s Equation The equations of motion for frictionless flow, called Euler s

Chapter 6 INCOMPRESSIBLE INVISCID FLOW All real fluids possess viscosity. However in many flow cases it is reasonable to neglect the effects of viscosity. It is useful to investigate the dynamics of an

### Given Find Water Properties

Venturi Example Given: A venturi is to be used to measure a 50 gpm flow of 70 F water in a 4-in ID pipe. Find: Select a throat diameter that provides Re d > 00,000 in the throat, and determine what differential

### Industrial Instrumentation Prof. A. Barua Department of Electrical Engineering Indian Institute of Technology Kharagpur. Lecture - 14 Flowmeter ΙΙΙ

Industrial Instrumentation Prof. A. Barua Department of Electrical Engineering Indian Institute of Technology Kharagpur Lecture - 14 Flowmeter ΙΙΙ Welcome to lesson 14 of Industrial Instrumentation. In

### 1.060 Engineering Mechanics II Spring Problem Set 4

1.060 Engineering Mechanics II Spring 2006 Due on Monday, March 20th Problem Set 4 Important note: Please start a new sheet of paper for each problem in the problem set. Write the names of the group members

### Angular momentum equation

Angular momentum equation For angular momentum equation, B =H O the angular momentum vector about point O which moments are desired. Where β is The Reynolds transport equation can be written as follows:

### V-Cone Flowmeter DATASHEET

V-Cone Flowmeter ATASHEET JUNHO 013 IntraCone Cone Flow Meter Type: ICM etail: Cone-Element Technical Information 05/011 FLOW THE SPECIALIST IN LEVEL AN FLOW Cone Flow Meter Technical Information Intra-Automation

### HYDRAULICS 1 (HYDRODYNAMICS) SPRING 2005

HYDRAULICS (HYDRODYNAMICS) SPRING 005 Part. Fluid-Flow Principles. Introduction. Definitions. Notation and fluid properties.3 Hydrostatics.4 Fluid dynamics.5 Control volumes.6 Visualising fluid flow.7

### B.E/B.Tech/M.E/M.Tech : Chemical Engineering Regulation: 2016 PG Specialisation : NA Sub. Code / Sub. Name : CH16304 FLUID MECHANICS Unit : I

Department of Chemical Engineering B.E/B.Tech/M.E/M.Tech : Chemical Engineering Regulation: 2016 PG Specialisation : NA Sub. Code / Sub. Name : CH16304 FLUID MECHANICS Unit : I LP: CH 16304 Rev. No: 00

### Process Gas Flow Measurement and Performance Evaluation API TECHNICAL REPORT 2571 SECOND EDITION DRAFT, FEBRUARY 2018

Process Gas Flow Measurement and Performance Evaluation API TECHNICAL REPORT 2571 SECOND EDITION DRAFT, FEBRUARY 2018 (Note: modify document as needed once final proposal on name change is agreed, i.e.

### Q1 Give answers to all of the following questions (5 marks each):

FLUID MECHANICS First Year Exam Solutions 03 Q Give answers to all of the following questions (5 marks each): (a) A cylinder of m in diameter is made with material of relative density 0.5. It is moored

### HVAC Clinic. Duct Design

HVAC Clinic Duct Design Table Of Contents Introduction... 3 Fundamentals Of Duct Design... 3 Pressure Changes In A System... 8 Example 1... 13 Duct Design Methods... 15 Example 2... 15 Introduction The

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### Open Channel Hydraulics III - Sharpcrested

PDHonline Course H140 (2 PDH) Open Channel Hydraulics III - Sharpcrested Weirs Instructor: Harlan H. Bengtson, Ph.D., PE 2012 PDH Online PDH Center 5272 Meadow Estates Drive Fairfax, VA 22030-6658 Phone

### Fluid Mechanics c) Orificemeter a) Viscous force, Turbulence force, Compressible force a) Turbulence force c) Integration d) The flow is rotational

Fluid Mechanics 1. Which is the cheapest device for measuring flow / discharge rate. a) Venturimeter b) Pitot tube c) Orificemeter d) None of the mentioned 2. Which forces are neglected to obtain Euler

### Prediction of Performance Characteristics of Orifice Plate Assembly for Non-Standard Conditions Using CFD

International Journal of Engineering and Technical Research (IJETR) ISSN: 2321-0869, Volume-3, Issue-5, May 2015 Prediction of Performance Characteristics of Orifice Plate Assembly for Non-Standard Conditions