1 . New Applications: Environmental Update EPA Method 8321B-Determination of Phenoxyacid Herbicides in Water by Solid Phase Extraction and LC-MS/MS Detection Determination of Pesticide Residues in Blueberries by AOAC QuEChERS Approach and Dispersive SPE Cleanup with a Novel Sorbent ChloroFiltr A Determination of Veterinary Drug Residues in Milk Using Polymeric SPE and UHPLC-MS/MS Analysis EPA Method 545: Determination of Cylindrospermopsin and Anatoxin-a in Drinking Water by Aqueous Direct Injection and LC/MS/MS Detection of Aflatoxins in Milk at Picogram Levels Using SPE and LC-MS/MS Forensic Simultaneous Determination of Prescription and Designer Benzodiazepines in Urine by SPE and LC-MS/MS SOFT-Determination of 35 Pesticides and 3 Cannabinoids in Marijuana Edibles Extraction of Synthetic and Naturally Occurring Cannabinoids in Urine Using SPE and LC-MS/MS LC-MS/MS Method for 89 Banned or Controlled Drugs in the Horse Racing Industry Updated- Quantitative Analysis of EtG and EtS in Urine Using Clean Screen ETG and LC-MS/MS Clinical Thyroid Hormones in Serum and Plasma Using SPE Extraction and UHPLC-MS/MS Analysis
2 Determination of Phenoxyacid Herbicides in Water by Solid Phase Extraction and LC-MS/MS Detection UCT Part Numbers: ECHLD156-P (Enviro Clean HL DVB 5mg/6mL, PE Frits) VMFSTFR12 (Sample Transfer Tubes) EPA Method 8321B* Procedure: 1. Sample Pretreatment a) Adjust sample ph to <1 with 1:1 sulfuric acid in water, low ph is critical to obtain high recoveries. 2. Cartridge Conditioning a) Attach sample transfer tubes (VMFSTFR12) to the top of the SPE cartridges (ECHLD156-P), and attach the SPE cartridges to an SPE manifold. b) Wash the SPE cartridges (with transfer tubes connected) using 1 ml methylene chloride, let solvent soak sorbent for 2 min before drawing to waste, leave full vacuum on for 1 min. c) Condition the SPE cartridges with 1 ml methanol, leave a thin layer above the frit. d) Equilibrate the SPE cartridges with 15 ml DI water, leave a thin layer above the frit. 3. Sample Loading a) Insert the stainless steel ends of the sample transfer tubes into sample bottles, adjust vacuum for a fast dropwise sample flow (about 2-25 ml/min). b) After all sample is passed through, dry the SPE cartridges under full vacuum for 1 min.
3 4. Analyte Elution a) Insert the collection vials to the manifold. b) Rinse the sample bottles with 5 ml acetonitrile, apply the rinse to the SPE cartridges. Let the elution solvent soak the sorbent for 1-2 min before drawing through slowly. c) Repeat the elution (step 4b) with 2 additional aliquots of 5 ml acetonitrile. Instrumental Analysis Analyze the eluate directly by LC-MS/MS, or concentrate to 1 ml and analyze by HPLC. Note: Use acid washed sodium sulfate and glassware if SPE eluates need be dried and concentrated. Results Compound LCS1 Recovery% LCS2 Recovery% RPD% 2,4-D MCPA Dichlorprop Mecoprop ,4,5-T Dichlorobenzoic acid ,4-DB Acifluorfen Silvex Bentazone *EPA Method 8321B SOLVENT-EXTRACTABLE NONVOLATILE COMPOUNDS BY HIGH-PERFORMANCE LIQUID CHROMATOGRAPHY/THERMOSPRAY/MASS SPECTROMETRY (HPLC/TS/MS) OR ULTRAVIOLET (UV) DETECTION UCT, LLC 2731 Bartram Road Bristol, PA UCT, LLC 215 All rights reserved
4 Determination of Pesticide Residues in Blueberries by AOAC QuEChERS Approach and Dispersive SPE Cleanup with a Novel Sorbent ChloroFiltr UCT Part Numbers: ECMSSA5CT-MP - Mylar pouch containing 6 g MgSO 4 and 1.5 g NaOAc with 5-mL centrifuge tubes included CUMPSGGC182CT - 2 ml centrifuge tube containing 15 mg MgSO 4, 5 mg PSA, 5 mg C18, and 5 mg ChloroFiltr SLAQ1ID21-3UM - Selectra Aqueous C18, 1 x 2.1 mm, 3 µm SLAQGDC2-3UM - Selectra Aqueous C18, Guard column, 1 x 2.1 mm, 3 µm SLGRDHLDR - Guard Cartridge Holder Summary: Blueberry has been ranked as one of the healthiest fruits for its high antioxidant content that helps combating free radicals, which could damage DNA and cellular structures . Application of pesticides during plant cultivation is common to increase product yield, therefore it is valuable developing effective analytical methods for the determination of pesticide residues in blueberries, which however is challenging due to matrix complexity as blueberries are rich in anthocyanins, sugars, polyphenols, vitamins, minerals, and other interfering components. This application outlines a simple, fast, and cost-effective method for the determination of multi-class pesticides, including one of the most problematic pesticides, pymetrozine in blueberries. The acetate buffered AOAC QuEChERS protocol demonstrated higher extraction efficiency for pymetrozine than the other 2 QuEChERS protocols (the EN citrate buffered or the original unbuffered), thus was selected for the extraction of pesticide residues in blueberries. 15 g of homogenized blueberries were extracted with 15 ml of acetonitrile (MeCN) containing 1% acetic acid (HAc). 6 g magnesium sulfate (MgSO 4 ) and 1.5 g sodium acetate (NaOAc) were employed to enhance phase separation and partition of pesticides into the MeCN layer. After shaking and centrifugation, 1 ml of the supernatant was transferred to a 2-mL dspe tube containing the optimized cleanup sorbents of 15 mg MgSO 4, 5 mg PSA, 5 mg C18, and 5 mg
5 ChloroFiltr. Residual water was absorbed by MgSO 4, anthocynins, polyphenols, sugars and organic acids were removed by PSA, lipids and other non-polar interferences were retained by C18, while chlorophylls were removed by ChloroFiltr, resulting in clean extract for LC/MS/MS analysis. UCT s aqueous C18 HPLC column was used for analyte retention and separation, which showed superior retention especially for several very polar pesticides, such as methamidophos and acephate. Procedure: 1. QuEChERS extraction a) Weigh 15 ±.3 g of homogenized blueberry sample into 5-mL centrifuge tubes. b) Add triphenyl phosphate (TPP) as internal standard (IS) (optional), and appropriate amounts of spiking solution to fortified samples. c) Add 15 ml of MeCN with 1% HAc. Cap and shake for 1 min at 1 strokes/min using a Spex 21 Geno-Grinder. d) Add salts (6 g MgSO 4 and 1.5 g NaOAc) from pouch (ECMSSA5CT- MP) to the 5-mL tube, and vortex for 1 sec to break up salt agglomerates. e) Shake for 1 min at 1 strokes/min using the Geno-Grinder. f) Centrifuge at 3 rcf for 5 min. 2. dspe cleanup a) Transfer 1 ml of the supernatant to a 2-mL dspe tube (CUMPSGGC182CT). b) Shake for 1 min at 1 strokes/min using the Spex 21 Geno-Grinder. c) Centrifuge at 3 rcf for 5 min. d) Transfer.2 ml of the cleaned extract into a 2-mL auto-sampler vial, add.2 ml of reagent water, and vortex for 3 sec. e) The samples are ready for LC-MS/MS analysis.
6 LC-MS/MS Method: HPLC: Thermo Scientific Dionex UltiMate 3 LC System Column: UCT, Selectra, aq C18, 1 x 2.1 mm, 3 µm Guard column: UCT, Selectra, aq C18, 1 x 2. mm, 3 µm Column temperature: 4 C Column flow rate:.3 ml/min Auto-sampler temperature: 1 C Injection volume: 2 µl Gradient program: Time (min) A% (1 mm ammonium B% (.1% formic acid acetate in DI water) in MeOH) Divert mobile phase to waste from 1.5 and min to prevent ion source contamination. MS parameters Instrumentation Thermo Scientific TSQ Vantage tandem MS Polarity ESI + Spray voltage 35 V Vaporizer temperature 45 C Ion transfer capillary 35 C Sheath gas pressure 5 arbitrary units Auxiliary gas pressure 4 arbitrary units Q1 and Q3 peak width.4 and.7 Da Collision gas and pressure Ar at 1.5 mtorr Cycle time.5 sec Acquisition method EZ Method (scheduled SRM)
8 Results: Selection of dspe cleanup sorbents: Different sorbent mixtures (A - E) were packed in 2-mL dspe centrifuge tubes for blueberry extract cleanup: A. 15 mg MgSO 4 and 5 mg PSA B. 15 mg MgSO 4 and 15 mg PSA C. 15 mg MgSO 4, 5 mg PSA, and 5 mg C18 D. 15 mg MgSO 4, 5 mg PSA, 5 mg C18, and 7.5 mg GCB E. 15 mg MgSO 4, 5 mg PSA, 5 mg C18, and 5 mg ChloroFiltr Photographs, from left to right: crude blueberry extract, and extracts cleaned with sorbent mixture A, B, C, D, and E, respectively. Illustrated in the above picture, cleanup of blueberry extracts with PSA only (A and B) or PSA and C18 (C) is inefficient for complete pigment removal. With the addition of either GCB (D) or ChloroFiltr (E), colorless extracts were obtained; therefore sorbent mixtures D and E were selected for the recovery study. Recovery study: Blueberry samples were fortified with 1 ng/g and 5 ng/g of pesticides, and underwent the AOAC QuEChERS extraction and dspe cleanup with 15 mg MgSO4, 5 mg PSA, 5 mg C18, and 7.5 mg GCB (D) or 5 mg ChloroFiltr (E) as described above. The mean recoveries and RSD% of 6 replicated samples are listed in the table below.
9 Accuracy and Precision of Pesticides in Spiked Blueberries Compound dspe cleanup with PSA/C18/GCB dspe cleanup with PSA/C18/ChloroFiltr Spiked at 1 ng/g Spiked at 5 ng/g Spiked at 1 ng/g Spiked at 5 ng/g Recovery% RSD% Recovery% RSD% Recovery% RSD% Recovery% RSD% Methamidophos Acephate Aldicarb_sulfoxide Oxydemeton_methyl Dichrotophos Pymetrozine. na Dimethoate Triethylphosphorothioate Carbendazim Dichlorvos Fenamiphos_sulfone Fenamiphos_sulfoxide Simazine Carbaryl Tebuthiuron Thiabendazole Famphur Flutriafol Thionazin Atrazine DEET Malathion Triadimefon Pyrimethanil Acetochlor Sulfotep Tebuconazole Zoxamide Diazinon Cyprodinil Pyrazophos Ethion Profenofos Chlorpyrifos
10 The recoveries of several pesticides such as pymetrozine, carbendazim, thiabendazole, and cyprodinil, were found be adversely affected by GCB, but unaffected when ChloroFiltr was used; therefore, this sorbent combination was selected for blueberry extract cleanup in the final optimized procedure. The graphs below demonstrate the recovery comparison using GCB versus ChloroFiltr at 2 contrasting levels (1 and 5 ng/g) Recovery% GCB 2 ChloroFiltr Spiked at 1 ng/g Recovery% GCB ChloroFiltr Spiked at 5 ng/g
11 C:\Xcalibur\...\1ppb_aQC18_2uL_4C2 8/19/215 1:58:24 AM RT: NL: 3.14E5 RT: NL: 1.1E6 RT: NL: 1.77E6 RT: NL: 1.5E6 RT: NL: 2.44E ms [ , ] M S 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] M S 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C NL: 6.72E NL: 3.75E NL: 3.12E NL: 7.32E NL: 4.48E ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] M S 1ppb_aQC18_2uL_ 4C ms [ , ] M S 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C NL: 9.65E NL: 1.16 E NL: 4.93E NL: 7.18E NL: 1.42E ms [ , ] MS 1ppb_aQC18_2uL_ 4C2 NL: 2.78E ms [ , ] MS 1ppb_aQC18_2uL_ 4C2 NL: 2.72E ms [ , ] MS 1ppb_aQC18_2uL_ 4C2 NL: 1.32E ms [ , ] MS 1ppb_aQC18_2uL_ 4C2 NL: 5.94E ms [ , ] MS 1ppb_aQC18_2uL_ 4C NL: 4.31E ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] M S 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C NL: 1.43E NL: 1.5E NL: 6.38E NL: 1.62E NL: 1.43E ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C NL: 1.83E NL: 7.46E NL: 5.98E NL: 2.46E NL: 8.26E ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] M S 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C NL: 4.62E NL: 3.71E NL: 1.35E NL: 1.13E NL: 7.43E ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C ms [ , ] M S 1ppb_aQC18_2uL_ 4C ms [ , ] MS 1ppb_aQC18_2uL_ 4C Time (min) Time (min) Time (min) Time (min) Time (min) Chromatograms of 34 Pesticides and TPP (IS) in 1:1 MeCN:H 2 O (1 ppb) using UCT Aqueous C18 HPLC Column (Compound order can be found in the SRM table.)
12 .65 Cyprodinil Y = *X R^2 =.9998 W: 1/X Area Ratio ng/ml Matrix-matched Calibration Curve of Cyprodinil (R 2 =.9998) References:  UCT, LLC 2731 Bartram Road Bristol, PA UCT, LLC 215 All rights reserved
13 A Determination of Veterinary Drug Residues in Milk Using Polymeric SPE and UHPLC-MS/MS Analysis UCT Part Numbers: ECHLD126-P EnviroClean HLDVB, 2 mg/6ml SPE cartridge, PE Frit SLDA1ID21-18UM Selectra DA, mm, 1.8 µm HPLC column SLDAGDC2-18UM Selectra DA, 1 2. mm, 1.8 µm guard cartridge SLGRDHLDR Guard cartridge holder Summary: This application note outlines a multi-class, multi-reside method for the determination of 49 representative veterinary drugs in milk using a simple, solid-phase extraction (SPE) procedure and analysis by UHPLC-MS/MS. To achieve fast and simultaneous extraction of the various drug residues, a generic liquid extraction procedure using EDTA/acetic acid buffer is conducted prior to extraction on a polymeric SPE cartridge. UHPLC separation is carried out with a Selectra DA column, which exhibits alternative selectivity to a C18 phase and is capable of enhanced retention for the more polar drugs. The method was evaluated for each compound at three varying concentrations (1, 1 and 1 μg/kg). For most compounds, recoveries were between 7% and 12% and reproducibility was <2%. In addition, the majority of compounds could be accurately detected at a concentration of 1 μg/kg, demonstrating that the presented method is sufficient to monitor a wide range of veterinary drugs in milk. The drugs investigated belonged to several different classes, including β-agonists, macrolides, amphenicols, sulfonamides, tetracyclines and quinolones. Introduction: Veterinary drugs are frequently administered to food-producing animals, including dairy cows, to treat and prevent disease and/or increase growth rates. The inappropriate or illegal use of these drugs can result in the presence of their residues in food of animal origin which could pose a potential threat to human health. Milk is an important food commodity that is consumed by a large portion of the population, including infants. To ensure food safety and prevent the unnecessary exposure of consumers to veterinary drugs, it is vital to test milk for drug residues. The United
14 States, European Union (EU), CODEX and other international organizations have established maximum residue limits (MRLs) for veterinary drugs in a variety of biological matrices, including milk [1-3]. The MRLs for milk are typically lower than those set for other biological matrices (muscle, liver and kidney) and span a wide concentration range (low µg/kg to >1 µg/kg). In addition, a number of drugs are prohibited for use in food producing animals or are unauthorized for use in lactating animals and require very low detection limits ( 2 μg/kg). Milk is a complex matrix containing dissolved fats, carbohydrates, proteins and minerals (including calcium), which can complicate the development of a fast, easy and reliable analytical method for the identification and quantification of veterinary drug residues. Development of a multi-class, multi-residue (MMR) method can be challenging not only due to the inclusion of a large number of drugs with diverse physicochemical properties, but also on account of the complex sample matrix and the instability of certain drug classes (e.g. β-lactams, tetracyclines and macrolides). A MMR method should ideally be capable of extracting a wide range of drugs, reduce major matrix interferences, obtain good analyte recovery, be reproducible and achieve adequate limits of detection (LOD s). The use of a generic sample preparation procedure, such as SPE using a polymeric sorbent, is a suitable approach for achieving these goals. Ultra-high performance liquid chromatography coupled with tandem mass spectrometry (UHPLC-MS/MS) is the detection system of choice for veterinary drugs as it allows rapid detection of trace-level residues in complex matrices. However, the diverse physicochemical properties of the veterinary drugs still pose challenges and analytical conditions must be optimized to obtain adequate sensitivity of all the compounds as well as good retention and peak shape of problematic compounds.
15 Sample Preparation Procedure: A. Sample extraction 1. Weigh 5 g of milk into a 15 ml polypropylene centrifuge tube. 2. Add 5 ml of.1m EDTA-Na 2 + 2% acetic acid. 3. Vortex for 5 minutes to de-proteinize the milk. 4. Centrifuge for 5 minutes at 35 g. Note: A larger volume of extraction solvent or a second extraction of the milk sample (5mL buffer) can be carried out if deemed necessary. B. SPE extraction 1. Condition SPE cartridge with: a) 1 3 ml methanol b) 1 3 ml ultrapure water 2. Apply the supernatant to the SPE cartridge, taking care to avoid any transfer of the lipid layer. If required, use a low vacuum to draw the sample through ( 5 ml/min). C. Wash cartridge ml ultrapure water ml 1% methanol. 3. Dry cartridge under vacuum ( 1 inhg) for 5-1 minutes to remove residual water ml hexane. 5. Dry cartridge under vacuum ( 1 inhg) for 1 minute to remove residual hexane. D. Elution 1. Elute with 3 ml acetone. 2. Evaporate the sample to dryness at 35-4 C under a gentle stream of nitrogen. 3. Reconstitute in 1 ml of methanol:water (5:5, v/v). 4. Filter extract with a.22 µm nylon (or other suitable membrane) syringe filter into an autosampler vial.
16 LC-MS/MS Conditions: HPLC Conditions Instrumentation Thermo Scientific TM Dionex TM Ultimate TM 3 HPLC column UCT Selectra DA, mm, 1.8 µm (p/n: SLDA1ID21-18UM) Guard column UCT Selectra DA, 1 2. mm, 1.8 µm (p/n: SLDAGDC2-18UM) Guard column p/n: SLGRDHLDR holder Column temp. 6 C Flow rate 4 µl/min Injection volume 5 µl Autosampler temp. 1 C Wash solvent Methanol Divert valve Divert to waste at -1.5 and min to reduce ion source contamination Time Mobile phase A Mobile phase B (min) Water +.1% formic Methanol +.1% acid formic acid. 95% 5%.5 7% 3% 4. 7% 3% 5. 4% 6% 8. 4% 6% 8.5 % 1% 12. % 1% % 5% % 5% MS Conditions Instrumentation Thermo Scientific TM TSQ Ionization mode Vantage ESI + & ESI TM (QqQ) Spray voltage 4 V Vaporizer 45 C temperature Capillary 35 C temperature Sheath gas 55 arbitrary units pressure Auxiliary gas 45 arbitrary units pressure Ion sweep gas arbitrary units Declustering V potential Q1 and Q3 peak.7 Da width Collision gas Argon Collision gas 1.7 mtorr pressure Acquisition EZ method (scheduled SRM) Cycle time.5 sec Software Xcalibur TM version 2.2
17 SRM Transitions Analyte t R (min) Precursor ion Product ion 1 CE 1 (V) Product ion 2 CE 2 (V) S-lens (V) Sulfanilamide Albuterol Albuterol-D 3 (IS) Lincomycin Ampicillin Trimethoprim Trimethoprim- 13 C 3 (IS) Thiamphenicol(ESI - ) Sulfadiazine Sulfathiazole Norfloxacin Ormetoprim Thiabendazole Thiabendazole-D 6 (IS) Oxytetracycline Cefalexin Ofloxacin Ciprofloxacin Ciprofloxacin- 15 N- 13 C 3 (IS) Tetracycline Sulfamethoxazole Sulfamethoxazole- 13 C 6 (IS) Sulfamerazine Lomefloxacin Sulfamethizole Chloramphenicol (ESI - ) Cefotaxime
18 Enrofloxacin Demeclocycline Sulfachloropyridazine Sulfamethazine Sulfamethazine- 13 C 6 (IS) Azithromycin Sarafloxacin Clindamycin Chlortetracycline Cefazolin Doxycycline Diphenhydramine Carbadox Sulfadimethoxine Erythromycin Erythromycin- 13 C 2 (IS) Cephalothin Penicillin G Anhydroeythromycin Clarithromycin Ceftiofur Penicillin V Tylosin Roxithromycin Oxolinic acid Oxacillin Cloxacillin Flumequine Virginiamycin CE = collision energy
19 Results and Discussion: Summary of the recovery, reproducibility and method performance data generated (n=12 each). 1 µg/kg 1 µg/kg 1 µg/kg Analyte Recovery (%) RSD (%) Recovery (%) RSD (%) Recovery (%) RSD (%) LCL (µg/kg) Linearity (R 2 ) Sulfanilamide 87.1 a Albuterol Lincomycin Ampicillin Trimethoprim Thiamphenicol Sulfadiazine Sulfathiazole Norfloxacin Ormetoprim Thiabendazole Oxytetracycline Cefalexin 69.3 a Ofloxacin 11.2 a Ciprofloxacin Tetracycline Sulfamethoxazole Sulfamerazine Lomefloxacin Sulfamethizole Enrofloxacin Chloramphenicol Cefotaxime Demeclocycline Sulfachloropyridazine Sulfamethazine
20 Azithromycin Sarafloxacin Clindamycin Chlortetracycline Cefazolin Doxycycline Diphenhydramine Carbadox Sulfadimethoxine Erythromycin Cephalothin Penicillin G Anhydroeythromycin Clarithromycin Ceftiofur Penicillin V Tylosin Roxithromycin Oxolinic acid Oxacillin Cloxacillin Flumequine Virginiamycin a n=6.
21 Chromatographic separation The unique chemistry of the Selectra DA column, which contains a polyaromatic stationary phase, provides orthogonal selectivity to a traditional C18 column and offers a high degree of retention and selectivity for aromatic compounds. The stationary phase is capable of retaining analytes through hydrophobic (dispersive) interactions as well as through pi-pi (π- π) interactions which exhibit a substantial increase in retention for dipolar, unsaturated or conjugated analytes. The Selectra DA column is ideally suited for the analysis of veterinary drug residues, as most compounds (and metabolites) possess aromatic functionality. In the final UHPLC-MS/MS method, methanol was chosen as the organic mobile phase solvent, as it was found to give better overall peak shape than acetonitrile, particularly for the tetracycline and fluoroquinolone antibiotics. A hold was included in the gradient to improve chromatographic separation and all compounds were successfully eluted in <12 min. The enhanced retention of the Selectra DA column ensured that the most polar compound included in the method, sulfanilamide, didn t elute until >2 minutes (3% methanol). Although it was possible to start the gradient at a higher percentage of organic solvent (2%) and reduce the overall run time, this required the use of a smaller injection volume (2 μl) which negatively affected the method sensitivity. Ultimately, the best sensitivity was obtained by starting the gradient at 5% methanol and using a 5 μl injection volume. RT: SM: 5G NL: 1.64E6 TIC MS 1ppb_sa mple_ Time (min) Figure 1. TIC chromatogram of an extracted milk sample (1 μg/kg) containing the 49 veterinary drugs and 7 internal standards.
22 Sample preparation procedure: Prior to instrumental analysis, a sample pre-treatment step is required to concentrate the analytes of interest and eliminate non-desirable matrix components. This is particularly important for the analysis of veterinary drugs in milk because of their low regulatory limits and the larger sample size required to obtain necessary method sensitivity. One of the biggest difficulties in milk analysis is the high fat, protein, and calcium content that can often interfere with instrumental analysis. The instability of certain drug classes, namely β-lactams, tetracyclines and macrolides, causes additional complications by limiting the conditions that can be used for sample extraction and cleanup. Therefore, the sample preparation procedure was optimized to remove as much co-extracted matrix components as possible while minimizing any loss of the veterinary drug residues. A simple deproteinization procedure using an EDTA/acetic acid buffer (sample ph should be 4-4.5) followed by centrifugation to separate the proteins and lipids was carried out prior to SPE extraction and cleanup. The inclusion of EDTA in the extraction buffer prevents the complexation of drugs with metal ions (e.g. calcium), particularly the tetracyclines and fluoroquinolones. After application of the sample supernatant to the SPE cartridge, the sorbent was washed with 1% methanol to remove polar matrix components and hexane to remove lipophilic compounds. Acetone was used as the SPE elution solvent as it was found to be more effective than methanol, particularly for hydrophobic compounds that contain multiple aromatic functional groups and are strongly retained on the DVB sorbent. Furthermore, acetone is a volatile organic solvent that is readily removed by evaporation under mild conditions (35-4 C). Filtration of the sample extract prior to LC-MS/MS analysis and the use of isotopically labeled internal standards and matrix-matched calibration curves are recommended in order to obtain the best possible results. For most compounds, the recovery was between 7% and 12% and the reproducibility <2%. Only a a small number of compounds gave results outside of the acceptable limits, which was due to analyte instability (cefalexin and ceftiofur) or inadequate sensitivity at the lowest concentration level (sulfanilamide and thiamphenicol). In addition, all compounds could be accurately detected at a concentration of 1 μg/kg and the vast majority of compounds at 1 μg/kg, demonstrating that the presented method is suitable for monitoring a wide range of veterinary drug residues in milk.
23 Figure 2. Example of an eight-point matrix-matched calibration curve (.5-2 µg/kg, equivalent to ng/ml in final extract) UCT, LLC 2731 Bartram Road Bristol, PA UCT, LLC 214 All rights reserved
24 EPA Method 545: Determination of Cylindrospermopsin and Anatoxin-a in Drinking Water by Aqueous Direct Injection and LC/MS/MS UCT Part Numbers: SLAQ1ID21-3UM - Selectra Aqueous C18, 1 x 2.1mm, 3µm SLAQGDC2-3UM - Selectra Aqueous C18, Guard column, 1 x 2.mm, 3µm SLGRDHLDR - Guard Cartridge Holder Summary: Cyanotoxins are toxins that are produced by some species of photosynthetic cyanobacteria (also known as blue-green algae) blooming under certain conditions , such as stagnant or slow moving warm water with high-level nutrients like phosphates and nitrogen. Cyanotoxins are dangerous to humans and wildlife, affecting their livers (hepatotoxic), nervous systems (neurotoxic), and skin (acutely dermatotoxic). Human exposure to cyanotoxins can occur through ingestion of either the contaminated drinking water or exposed fish and shellfish, in addition to inhalation or dermal contact with the contaminated recreational water . This application note describes a direct aqueous injection and liquid chromatography tandem mass spectrometry (DAI-LC/MS/MS) method for the determination of cylindrospermopsin (hepatotoxin) and anatoxin-a (neurotoxin) in drinking water under EPA Method 545 . These two target analytes are also included in the UCMR4 screening compound list that will be monitored by public drinking water systems soon. UCT s Aqueous C18 HPLC column was utilized for analyte retention and separation, which had demonstrated excellent consistency in peak area and retention times. 7-point calibration curves were constructed for analyte quantification. The responses were linear (R ) over the analytical range from.1 to 1 µg/l. Excellent accuracy ( %) and precision (RSD% < 1%, n=7) were achieved in fortified reagent water and tap water samples.
25 Procedure: 1. Preserve water samples with 1 g/l of sodium bisulfate (antimicrobial) and.1 g/l of ascorbic acid (dechlorination). 2. Add 1 μl of.5-2 ng/μl internal standard mixture to 2-mL vials, and appropriate amounts of spiking solutions for fortified samples, and bring the final volume to 1 ml with the preserved water samples. 3. Vortex the samples for 3 sec and analyze by LC-MS/MS equipped with an Aqueous C18 HPLC column. LC-MS/MS method: HPLC: Thermo Scientific Dionex UltiMate 3 LC System Column: UCT, Selectra, aq C18, 1 x 2.1 mm, 3 µm Guard column: UCT, Selectra, aq C18, 1 x 2. mm, 3 µm Column temperature: 3 C Column flow rate:.3 ml/min Auto-sampler temperature: 1 C Injection volume: 5 µl Gradient program: Time (min) A% (5 mm acetic acid in DI water) B% (MeOH) Divert mobile phase to waste from 1.8 and min to prevent ion source contamination.
26 MS parameters Instrumentation Thermo Scientific TSQ Vantage tandem MS Polarity ESI + Spray voltage 4 V Vaporizer temperature 4 C Ion transfer capillary temperature 35 C Sheath gas pressure 5 arbitrary units Auxiliary gas pressure 25 arbitrary units Q1 and Q3 peak width (FWHM).2 and.7 Da Collision gas and pressure Ar at 1.5 mtorr Cycle time.6 sec Acquisition method EZ Method (scheduled SRM) Retention Times and SRM Transitions Compound Rt (min) Precursor Product 1 CE 1 Product 2 CE 2 S-Lens RF Uracil-d L-phenylalanine-d Cylindrospermopsin Anatoxin-a Results: Relative Standard Deviation (RSD) of Peak Area and Retention Times by aq C18 HPLC Column Compound Peak Area RSD% (n=44) Retention Time Uracil-d L-phenylalanine-d Cylindrospermopsin ND.1 Anatoxin-a ND.3 ND: not determined.
27 Area Ratio Analytical Range and Linearity Data Compound Analytical range Linearity (R 2 ) (µg/l) Reagent water Tap water Cylindrospermopsin Anatoxin-a Calibration Curve of Anatoxin-a in Reagent Water.65 Anatoxin-a Y = *X R^2 =.999 W: 1/X ppb Accuracy and Precision in Fortified Reagent Water (n=7) Compound Spiked at.1 µg/l Spiked at 2.5 µg/l Recovery% RSD% Recovery% RSD% Cylindrospermopsin Anatoxin-a Accuracy and Precision in Fortified Tap Water (n=7) Compound Spiked at.1 µg/l Spiked at 2.5 µg/l Recovery% RSD% Recovery% RSD% Cylindrospermopsin Anatoxin-a
28 References:    UCT, LLC 2731 Bartram Road Bristol, PA UCT, LLC 215 All rights reserved
29 Detection of Aflatoxins in Milk at Picogram Levels Using SPE and LC-MS/MS UCT Part Numbers: ECHLD126-P EnviroClean HL DVB, 2 mg/6 ml SPE cartridge SLC-181ID21-3UM Selectra C18, mm, 3 µm HPLC column SLC-18GDC2-3UM Selectra C18, 1 2. mm, 3 µm guard cartridge SLGRDHLDR Guard cartridge holder August 215 Summary: Aflatoxins are naturally occurring mycotoxins that are produced by several species of fungi (Aspergillus flavus and Aspergillus parasiticus). They are classified by the International Agency for Research on Cancer as group 1 carcinogens (compounds known to be carcinogenic in humans) . Aflatoxins can occur in food products as a result of fungal contamination of crops (prior to harvest or during storage). There are approximately 2 related aflatoxin metabolites, although only B1, B2, G1 and G2 are normally found in food . Of these, aflatoxin B1 is the most biologically active and most commonly encountered . Aflatoxins can occur in milk as a result of dairy animals consuming contaminated feed. The main residue of concern in milk is aflatoxin M1, the major metabolite of B1. The intake of contaminated milk, even at low concentrations, is a significant threat to human health, especially to children who are a major consumer of dairy products. Therefore, the US Food and Drug Administration has established a tolerance of.5 μg/kg for aflatoxin M1 in milk , while the European Union has imposed more stringent limits -.5 μg/kg in raw milk and.25 μg/kg in infant formula . No limits have been established for aflatoxin B1, B2, G1 and G2 in milk. This application note outlines a method for the low level determination of aflatoxins in milk using a polymeric solid-phase extraction (SPE) cartridge. Analysis is performed by LC-MS/MS using a Selectra DA HPLC column. The method was optimized to allow the detection of aflatoxins at the low regulatory concentrations required. Recovery studies were carried out by spiking whole milk at two concentration levels (.25 and.5 µg/kg). Matrix-matched calibration curves, ranging from.1-2 µg/kg, were used for quantitation. The mean recovery was found to be in the range of 84 to 1%, and repeatability was 7%.
30 Procedure: Aflatoxins are relatively unstable in light and air, particularly in polar solvents or when exposed to oxidizing agents, ultraviolet light, or solutions with a ph below 3 or above 1. They should be protected from ultraviolet light as much as possible. 1. Sample extraction 1 (aqueous extraction) a) Weigh 2 g of milk into a 5 ml polypropylene centrifuge tube. b) Add 2 µl of glacial acetic acid. c) Vortex for 5 minutes to deproteinize the milk. d) Centrifuge for 5 minutes at 4 g. 2. SPE extraction a) Condition SPE cartridge with: ml methanol ml ultrapure water b) Apply the supernatant to the SPE cartridge, taking care to avoid any transfer of the lipid layer. If required, use a low vacuum to draw the sample through ( 5 ml/min). 3. Sample extraction 2 (solvent extraction) a) Add 1 ml acetone to any residual milk solids from the aqueous extraction. b) Vortex for 2 minutes to extract the aflatoxins. c) Centrifuge for 5 minutes at 4 g. d) Transfer the supernatant to a clean polypropylene tube and evaporate to.5 ml at 5 C under a gentle stream of nitrogen. e) Add 1 ml ultrapure water and vortex briefly. f) Apply sample to the SPE cartridge (same cartridge as step B). 4. Wash cartridge a) 1 3 ml ultrapure water. b) 1 3 ml 5% methanol. c) Dry cartridge under vacuum ( 1 inhg) for 5-1 minutes to remove residual water. d) 1 3 ml hexane. e) Dry cartridge under vacuum ( 1 inhg) for 1 minute to remove residual hexane. 5. Elution a) Elute the aflatoxins with 4 ml acetone. b) Evaporate the sample to dryness at 5 C under a gentle stream of nitrogen. c) Reconstitute in 1 ml of methanol:water (5:5, v/v). d) Filter extract with a.22 µm nylon (or other suitable membrane) syringe filter into an autosampler vial.
31 LC-MS/MS Conditions: HPLC Conditions Instrumentation Thermo Scientific TM Dionex TM Ultimate TM 3 HPLC column Guard column Guard column holder UCT Selectra C18, mm, 3 µm (p/n: SLC-181ID21-3UM) UCT Selectra C18, 1 2. mm, 3 µm (p/n: SLC-18GDC2-3UM) p/n: SLGRDHLDR Column temp. 4 C Mobile phase A Water +.1% formic acid Mobile phase B Acetonitrile +.1% formic acid Flow rate 3 µl/min Gradient min (5% B), 5-6 min (hold 1% B), min (equilibrate 5% B) Injection volume 2 µl Autosampler temp. 1 C Wash solvent Divert valve Methanol Divert to waste at -4 and 6-11 min to reduce ion source contamination MS Conditions Instrumentation Thermo Scientific TM TSQ Ionization mode Vantage ESI + TM Spray voltage 35 V Vaporizer 4 C temperature Capillary 35 C temperature Sheath gas pressure 55 arbitrary units Auxiliary gas 45 arbitrary units pressure Ion sweep gas arbitrary units Declustering V potential Q1 and Q3 peak.2 and.7 Da width Collision gas Argon Collision gas 2.2 mtorr pressure Acquisition method EZ method (scheduled SRM) Cycle time.6 sec Software Xcalibur TM version 2.2 Analyte t R (min) Precursor ion SRM Transitions Product ion 1 CE 1 Product ion 2 CE 2 S-lens (V) Aflatoxin M Aflatoxin G Aflatoxin G Aflatoxin B Aflatoxin B
32 Area Results and Discussion: Accuracy & Precision Data for Whole Milk.25 µg/kg (n=5).5 µg/kg (n=5) Mean Recovery RSD Mean Recovery RSD (%) (%) (%) (%) Aflatoxin M Aflatoxin B Aflatoxin B Aflatoxin G Aflatoxin G Aflatoxin_B1 Y = *X R^2 =.9999 W: 1/X ng/ml Figure 1. Example of a seven point matrix-matched calibration curve (.2,.5, 1, 2, 1, 2 and 4 ng/ml; equivalent to.1,.25,.5,.1,.5, 1 and 2 µg/kg in milk).
33 Chromatograms RT: SM: 7G NL: 7.76E4 ms [ , ] MS High_ NL: 5.6E4 8 ms [ , ] MS High_ NL: 1.88E5 8 ms [ , ] MS High_ NL: 1.54E5 ms [ , ] MS High_ NL: 2.53E5 ms [ , ] MS High_ Time (min) Figure 2. Chromatogram of an extracted milk sample fortified at.5 µg/kg. RT: SM: 7G NL: 4.42E2 ms [ , ] MS Std_ NL: 3.7E2 8 ms [ , ] MS Std_ NL: 1.E3 8 ms [ , ] MS Std_ NL: 1.7E3 ms [ , ] MS Std_ NL: 1.32E3 ms [ , ] MS Std_ Time (min) Figure 3. Chromatogram of the lowest matrix-matched calibration point (.1 µg/kg).
34 Prior to instrumental analysis, a sample pre-treatment step is required to concentrate the analyte(s) of interest and eliminate non-desirable matrix components. This is particularly important for the analysis of aflatoxins in milk because of the low regulatory limits established for aflatoxin M1 and the larger sample size required to obtain the necessary method sensitivity. SPE is ideally suited to achieving these objectives. One of the biggest difficulties in milk analysis is the high fat and protein content that can often interfere with instrumental analysis. The sample preparation procedure was therefore optimized to remove as much co-extracted matrix components as possible. Initially, a simple deproteinization step using acetic acid followed by centrifugation was executed to separate the proteins and lipids prior to SPE extraction. However, the recoveries were found to be low (<4%) using this approach, which is most likely caused by the adsorption of the aflatoxins onto proteins or lipids in the milk. It was determined that a solvent extraction step was necessary to adequately extract the aflatoxin residues from milk prior to SPE cleanup. Due to the high water content of milk, direct extraction with an organic solvent would result in a large volume of supernatant that could not be directly applied to the SPE cartridge (organic content too high) or require a time consuming evaporation step to remove the solvent. As a result, a twostep extraction procedure incorporating an initial aqueous extraction step was included in the final method. This simple step removes most of the water from the sample prior to a second extraction with acetone, a volatile organic solvent that is readily removed by evaporation. This extract is then reconstituted in water prior to application to the SPE cartridge. The SPE sorbent was washed with 5% methanol to remove medium to highly polar matrix components and hexane to remove lipophilic compounds. Acetone was used as the SPE elution solvent as it was found to be more effective than methanol and very easy to remove by evaporation. Filtration of the final sample extract prior to LC-MS/MS analysis and the use of matrix-matched calibration curves and/or isotopically labeled internal standards are recommended to obtain optimal results.
35 References: 1. IARC Monographs on the Evaluation of Carcinogenic Risks to Humans, International Agency for Research on Cancer (IARC) website, (accessed June 215). 2. European Mycotoxins Awareness Network website, (accessed June 215). 3. European Food Safety Authority website, (accessed June 215) 4. FDA Compliance Policy Guide, Sec Whole Milk, Lowfat Milk, Skim Milk - Aflatoxin M1. 5. Commission Regulation (EU) No 165/21. Amending Regulation (EC) No 1881/26 Setting Maximum Levels for Certain Contaminants in Foodstuffs as Regards Aflatoxins. Official Journal of the European Union, Feb 26, 21, pp L 5/8 L 5/ UCT, LLC 2731 Bartram Road Bristol, PA UCT, LLC 211 All rights reserved
36 Simultaneous Determination of Prescription and Designer Benzodiazepines in Urine by SPE and LC-MS/MS UCT Part Numbers: CSXCE16 Clean Screen XCEL, 13mg/6mL SPHACE51-5- Select ph Buffer, 1 mm Acetate ph 5. BETA-GLUC-5-5mL β - glucuronidase enzyme - liquid form SLDA1ID21-3UM- Selectra DA Column, 1 x 2.1mm, 3µm SLDAGDC21-3UM- Selectra DA Guard Column, 1 x 2.1mm, 3µm SLGRDHLDR - Guard Cartridge Holder SUMMARY: Benzodiazepines, frequently referred to as Benzos, are prescribed for the treatment of anxiet y, insomnia, muscle spasms, alcohol w ithdraw al and seizureprevention on account of their ability to depress the central nervous system. Generally, these drugs are deemed safe and highly effective w hen used properly and for short durations of time. How ever, long term use can lead to both physical and psychological dependence consequentially triggering abuse 1. Benzodiazepines are also recurrently utilized as illegal recreational drugs. In this case, they may be ground to a pow der, mixed w ith w ater and injected, as w ell as being sw allow ed as pills. Their administration is often accompanied by the use of other drugs, such as alcohol and opioids for an enhanced overall effect. Similar to other commonly abused compounds, such as cannabinoids or amphetamines, legal alternatives have been developed for Benzos as w ell in an attempt to bypass the controlled substances act. These new designer drugs are structural or functional analogs of the controlled substance designed to not only mimic the pharmacological effects of the original drug, but also avoid illegal classification and/or detection in a standard drug test. 2 Keeping up w ith the ever changing designer drug market has proven to be a real challenge for laboratories across the country. Given that these compounds are derived from template structures, it w ill prove valuable for labs to have a method that can not only target current metabolites of interest, but also the latest ones being formulated.
37 PROCEDURE: Sample Pretreatment To 1 ml of urine sample, add 1 ml of 1 mm Acetate Buffer (ph= 5) and 25-5 μl of concentrated Selectrazyme β-glucuronidase (BETA-GLUC-5). Vortex and heat for 1-2 hours at 65 C; Allow sample to cool Do not adjust ph~ sample is ready to be added to the extraction column. SPE Method: 1. Attach SPE cartridges (CSXCE16) to a glass block manifold or positive pressure manifold. 2. Load the pretreated sample, adjust vacuum or pressure for a slow dropw ise sample flow (1-2mL/min). 3. Wash the SPE cartridges w ith 3 ml of Acetate Buffer ph 5 (SPHACE51-5). Dry the SPE cartridges under full vacuum or pressure for 5 minutes. 4. Repeat the w ash w ith 3 ml Methylene Chloride. Dry the SPE cartridges under full vacuum or pressure for 1 min. 5. Insert collection rack w ith test tubes to the manifold, and elute the retained analytes w ith 3 ml of Ethyl Acetate:NH 4 OH (98:2). 6. Evaporate the eluate to dryness at 45 C under a gentle stream of nitrogen, and reconstitute w ith 1 μl of 5% MeOH in DI w ater. 7. Vortex the extract for 3 sec and transfer to 2-μL inserts held in 2-mL vials.
38 HPLC Conditions Instrumentation Agilent 12 Binary Pump SL HPLC column UCT Selectra DA, mm, 3 µm (p/n: SLDA1ID21-3UM) Guard column UCT Selectra DA, 1 2. mm, 3 µm (p/n: SLDAGDC21-3UM) Guard column holder p/n: SLGRDHLDR Column temp. 4 C Mobile phase A Water +.1% formic acid Mobile phase B Methanol +.1% formic acid Flow rate 3 µl/min Gradient min,7%b; 1-6 min,1%b; 6-9 min,1%b; min, 7%B Injection volume 1 µl Autosampler temp. 1 C Wash solvent Methanol MS Conditions Instrumentation API 4 QTRAP MS/MS Ionization mode ESI + Spray voltage 42 V Vaporizer temperature 65 C Capillary temperature 35 C Sheath gas pressure 4 arbitrary units Auxiliary gas pressure 5 arbitrary units Ion sweep gas arbitrary units MRM Transitions (ESI positive, dwell time: 5 ms) Compound Retention Time MRM Transitions (min.) Q1 Q3 ion 1 Q3 ion amino Clonazepam Midazolam Lorazepam Oxazepam Clonazepam Flubromazepam Alpha-Hydroxy-Alprazolam Nordiazepam Phenazepam Pyrazolam Temazepam Flubromazolam Alprazolam Diclazepam Diazepam Etizolam
40 Matrix Matched Calibration Curve: Prescription Benzodiazepine (Temazepam) Chromatogram of 1 ng/ml solvent standard
41 Discussion: Traditional benzodiazepines primarily function as neutral analytes, how ever, there are some that have w eakly basic functional groups allow ing for ionization at a fixed ph. The Clean Screen XCEL I column w as chosen for this application due to its capability of simultaneously extracting neutral and basic compounds, w hile eliminating the need for time-consuming column conditioning and extensive solvent usage for sample cleanup. The urine samples w ere adjusted to a ph of 5 for optimal enzyme hydrolysis. Follow ing incubation, they w ere loaded directly onto the Clean Screen XCEL I extraction columns. Although the majority of the analytes are primarily retained via hydrophobic interactions, use of this ph allow s for any potential ionizable groups to be fully charged and retained via ion exchange. The percentage of organic in the w ash w as optimized in order to provide sufficient cleanup for the benzodiazepines w ithout compromising on overall analyte recovery. After trying several various solvent combinations of Hexane, Methylene Chloride and buffer containing a percentage of Acetonitrile, it w as determined that a buffer w ash sequentially follow ed by a Methylene Chloride w ash gave the cleanest final extract w ith the best overall recovery. An elution solvent of Ethyl Acetate w ith 2% Ammonium Hydroxide w as chosen on account of its ability to not only displace any hydrophobic interactions, but also simultaneously disrupt ionic retention factors. Conclusion: 1. By utilizing UCT s Clean Screen XCEL I extraction columns in conjunction w ith the Selectra DA HPLC column, prescription and synthetic Benzodiazepine levels can be monitored simultaneously reducing both sample preparation and instrumental analysis time. 2. The universal nature of this extraction method makes it applicable to existing Benzodiazepines along w ith other new ly emerging analogs. 3. It is strongly recommended to use matrix-matched calibration curves, w hich include isotopically labeled internal standards to compensate for any remaining matrix that is not removed via the extraction procedure. References: 1. Benzodiazepines Drug Class Information on RxList.com RxList. N.p., 2 Apr Web 28 Sept w w.ncbi.nlm.nih.gov/pubmed/ UCT, LLC 2731 Bartram Road Bristol, PA UCT, LLC 215 All rights reserved
42 Determination of 35 Pesticides and 3 Cannabinoids in Marijuana Edibles UCT Part Numbers: ECQUUS95CT-MP QuEChERS salts for THC Potency and Pesticide Testing- 5 ml Centrifuge Tubes included ECQUUS142CT- Dispersive SPE sorbent blend for Pesticide Testing in Edibles- 2 ml Centrifuge Tubes included SLAQ1ID21-3UM- Selectra Aqueous C18 HPLC Column 1 x 2.1 mm, 3µm SLAQGDC2-3UM- Selectra Aqueous C18 Guard Column, 1 x 2.1mm, 3µm SLGRDHLDR - Guard Cartridge Holder Introduction: As of July 215, 23 states and Washington D.C. in USA have legalized the medical use of marijuana, w hile 4 states and Washington D.C. have legalized the recreational use of marijuana. As a result, many forensic toxicology labs are looking for fast, reliable, and cost-effective methods to determine cannabis potency and pesticides in edibles. This application utilizes the advantages of QuEChERS (Quick, Easy, Cheap, Effective, Rugged, and Safe) to extract 35 pesticides and 3 cannabinoids including tetrahydrocannabinol (THC), cannabidiol (CBD), and cannabinol (CBN) in edibles, follow ed by either serial dilutions for cannabis potency analysis, or a dispersive solid phase extraction (dspe) cleanup for pesticide residue analysis. This hybrid method allow s the QuEChERs technique, w hich is extensively used in the food testing industry, to be utilized in a forensic setting. Procedure: (a) Sample pre-treatment 1. For hard candies and chocolate, grind to a fine pow der using a SPEX 677 freezer mill. Figure 1: Hard candy before (left) and after (right) freezer mill grinding
43 2. For gummy samples, cut into slim pieces. Although freezer mill can grind gummies to pow der at low temperature w ith the use of liquid nitrogen, it returns to gel state w hen temperature goes up to room temperature, thus gummy samples should be cut instead of ground. 3. For sodas, degas for 3 min by sonication. Figure 2: Degassing of Reef cola (left) and Orange kush (right) (b) QuEChERS extraction 1. Weigh 1 g of the pre-treated samples (hard candies, gummies, brow nies, chocolate, and oil) into 5-mL centrifuge tubes, add internal standard (optional) and 1 ml of reagent w ater, and hydrate for 1 hr using a horizontal shaker. For sodas, add 1 ml of the degassed sample and internal standard (optional) to 5-mL centrifuges. 2. Add 1 ml of acetonitrile (MeCN) w ith 1% acetic acid. 3. Add QuEChERS extraction salts from pouches (ECQUUS95CT-MP), and vortex for 1 sec to break up salt agglomerates.
44 4. Shake for 1 min at 1 stroke/min using a SPEX Geno/Grinder. For gummy samples, add 2 metal balls and shake for 1 min at 1 stroke/min. 5. Centrifuge at 3 rcf for 5 min. Figure 3: Samples after QuEChERS extraction (from left to right: hard candies, gummies, soda, and chocolate) (c) dspe cleanup for pesticide residue analysis 1. Transfer 1 ml of the supernatants to 2-mL dspe tube (ECQUUS142CT). 2. Shake for 1 min at 1 stroke/min using the SPEX Geno/Grinder. 3. Centrifuge at 3 rcf for 5 min. 4. Transfer 2 µl extract to the 2-mL auto-sampler vials, add 2 µl of DI w ater, and vortex for 3 sec. Figure 4: Comparison of QuEChERS extracts before and after dspe cleanup (from left to right: hard candies and gummies) (d) Make serial dilutions for cannabinoid analysis 1. Perform serial dilutions (2 to 2, times depending on the cannabinoid concentration in different samples) of the QuEChERS extracts to 1 to 2 ng/ml. 2. Spike the diluted samples w ith 5 and 15% of the target cannabinoids, w hich are used to quantify t he cannabinoid concentration according to the standard addition method.
EPA Method 1694: Analysis of Pharmaceuticals and Personal Care Products in Water UCT Part Numbers ECHLD156-P ENVIRO-CLEAN HL DVB 500 mg, 6 ml cartridge VMF016GL 16 position glass block manifold VMFSTFR12
E N V I R O Selectively removing Chlorophyll without Compromising your Planar Pesticides INNOVATION THROUGH CHEMISTRY ChloroFiltr is a polymeric sorbent available exclusively from. It is designed to replace
Analysis of Tobacco Alkaloids UCT Part Number: EUBCX1HL2Z (200 mg benzenesulfonic acid high load, 10 ml) October 2012 Tobacco alkaloids are extracted with a strong cation exchange sorbent using an acidic
Macrolides in Honey Using Agilent Bond Elut Plexa SPE, Poroshell 120, and LC/MS/MS Application Note Food Testing and Agriculture Author Chen-Hao (Andy) Zhai and Rong-jie Fu Agilent Technologies (Shanghai)
UCT SELECTRA AQUEOUS C18 HPLC COLUMNS I N N O VAT I O N T H R O U G H C H E M I S T R Y SELECTRA AQUEOUS C18 HPLC COLUMNS Product Benefits Similar non-polar retention to traditional C18 Some selectivity
Rapid Screening and Confirmation of Melamine Residues in Milk and Its Products by Liquid Chromatography Tandem Mass Spectrometry Application Note Food Authors Jianqiu Mi, Zhengxiang Zhang, Zhixu Zhang,
Determination of Pesticide Residues in Oats by Automated QuEChERS and LC/QQQ Application Note Abstract The QuEChERS (Quick-Easy-Cheap-Effective-Rugged-Safe) sample extraction method was developed for the
EPA Method 549.2 Revision 1.0 Determination of Diquat and Paraquat in Drinking Water by Liquid-Solid Extraction and High Performance Liquid Chromatography with Ultraviolet Detection* UCT Products: ENVIRO-CLEAN
Application Note: 56 Analysis of Illegal Dyes in Food Matrices using Automated Online Sample Preparation with LC/MS Yang Shi, Catherine Lafontaine, Matthew Berube, John Fink, François Espourteille Thermo
Application Note AN785 Extraction of Aflatoxins and chratoxin From Dried Chili Using ISLUTE Myco prior to LC-MS/MS Analysis Page Extraction of Aflatoxins and chratoxin from Dried Chili Using ISLUTE Myco
LC-MS/MS Method for the Determination of Diclofenac in Human Plasma J. Jones, Thermo Fisher Scientific, Runcorn, Cheshire, UK Application Note 20569 Key Words SPE, SOLA, Accucore RP-MS, diclofenac, Core
1 Supporting Information 2 3 Toxicity, Teratogenic and Estrogenic Effects of Bisphenol A and its Alternative Replacements Bisphenol S, Bisphenol F and Bisphenol AF in Zebrafish. 4 5 John Moreman, Okhyun
Aminoglycosides in Milk Using Agilent Bond Elut Plexa SPE, Agilent Poroshell 120, and LC/Tandem MS Application ote Food Testing & Agriculture Author Andy Zhai Agilent Technologies, Inc. Shanghai Co. Ltd.
HPLC Winter Webinars Part 2: Sample Preparation for HPLC Jon Bardsley, Application Chemist Thermo Fisher Scientific, Runcorn/UK The world leader in serving science What am I Going to Talk About? What do
Thermo Scientific HyperSep Dispersive SPE Products Efficient sample preparation and clean-up using the QuEChERS Method The QuEChERS method overcomes the problems associated with time consuming, expensive
Ultrasensitive EPA Method 1694 with the Agilent 6460 LC/MS/MS with Jet Stream Technology for Pharmaceuticals and Personal Care Products in Water Application Note Environmental Authors Imma Ferrer, E. Michael
Simultaneous Determination of Paraquat and Diquat in Environmental Water Samples by HPLC-MS/MS Richard Jack, Xiaodong Liu, Leo Wang, and Chris Pohl OT70806_E 08/13S 1 The world leader in serving science
Sensitive Femtogram Determination of Aflatoxins B 1, B 2, G 1 and G 2 in Food Matrices using Triple Quadrupole LC/MS Application Note Food Safety Authors Yang Chen and Jack Cappozzo National Center for
LC/MS/MS Analysis of Pesticide Residues in Apples Using Agilent Chem Elut Cartridges Application ote Food Testing & Agriculture Author Andy Zhai Agilent Technologies Shanghai Co. Ltd. Abstract This application
APPLICATION NOTE 72686 Determination of underivatized aflatoxins,, G2, and in ground hazelnuts by immunoaffinity solid-phase extraction with HPLC-FLD detection Authors Sylvia Grosse, Mauro De Pra, Frank
Maximizing Triple Quadrupole Mass Spectrometry Productivity with the Agilent StreamSelect LC/MS System Application Note Authors Kevin McCann, Sameer Nene, Doug McIntyre, Edmond Neo, Dennis Nagtalon, and
POSTER NOTE 64922 LC-MS/MS Analysis of Phytocannabinoids and their LC-MS/MS Analysis of Phytocannabinoid Metabolites in Urine, Oral Fluid and Blood Rory M Doyle, Sherry Gregory*, Thermo Fisher Scientific,
Application Note EPA Method 535 EPA Method 535: Detection of Degradates of Chloroacetanilides and other Acetamide Herbicides in Water by LC/MS/MS API 3200 LC/MS/MS System Overview Described here is the
Quantification of growth promoters olaquindox and carbadox in animal feedstuff with the Agilent 126 Infinity Binary LC system with UV detection Application Note Food Author Srividya Kailasam Agilent Technologies,
Analysis of fumonisin, FB1 and FB2 mycotoxins in corn food and feed samples using the Agilent 1120 Compact LC System coupled to the Agilent 6140 Single Quadrupole LC/MS System Application Note Food Safety
Identification and Quantitation of Pesticides in Chamomile and Ginger Extracts Using an Agilent 646 Triple Quadrupole LC/MS system with Triggered MRM Application Note Authors Thomas Glauner Agilent Technologies,
Agilent Bond Elut QuEChERS Food Safety Applications Notebook: Volume 2 INNOVATIVE APPROACHES FOR TODAY S FOOD ANALYSIS CHALLENGES Table of Contents QuEChERS Page Food Safety Overview...3 What is QuEChERS?...4
LC/MS/MS of Fungicides and Metabolites in Orange Juice with Agilent Bond Elut Plexa and Poroshell 10 Application Note Food Testing & Agriculture Author Irina Dioumaeva Agilent Technologies, Inc. Abstract
Variable Data-Independent Acquisition (vdia) Delivers High Selectivity and Sensitivity in Combined Targeted and Untargeted Analyses for Small Molecules Olaf Scheibner 1, Markus Kellmann 1, Charles Yang
Chromatography Technical Note No AS90 Fully automated QuEChERS clean-up and LC/MS-QQQ analysis of pesticides in fruits and vegetables. Paul H. Roberts, Anatune Ltd. Hardwick, Cambridgeshire, UK. Introduction
Using UHPLC-Triple Quadrupole MS/MS to Detect the Presence of Bark Extract and Adulteration in Dietary Supplements and Botanicals Application Note Food Testing & Agriculture Authors James Neal-Kababick
24-03-2017 Bilag 3 Appendix B: Function tests Purchase of two (2) UPLC/UHPLC-MS/MS systems 2017 B1 B2 B3 B4 Function test for Nitrofuran metabolites in muscle Function test for Antibiotics in milk Function
Ultrafast Analysis of Buprenorphine and Norbuprenorphine in Urine Using the Agilent RapidFire High-Throughput Mass Spectrometry System Application Note Authors Mohamed Youssef and Vaughn P. Miller Agilent
Oasis HLB DISKS I. Introduction Oasis HLB is a Hydrophilic-Lipophilic-Balanced, water-wettable, reversed-phase sorbent for all your SPE needs. It is made from a specific ratio of two monomers, the hydrophilic
LC/MS/MS of Fungicides and Metabolites in Apple Juice with Agilent Bond Elut Plexa and Poroshell 120 Application Note Food Testing & Agriculture Author Irina Dioumaeva Agilent Technologies, Inc. Abstract
Solid Phase Extraction Method Development Tips and Tricks Carol Haney Ball, Ph.D. Application Scientist Agilent Technologies, Inc. February 3, 2009 Sources of Error Generated and Time Spent During a Typical
Introduction The enrichment of hydrophilic analytes from polar matrices is a challenge because they often cannot be retained by C 18 / C 8 -modified silica or hydrophobic polymer adsorbents. To overcome
Analysis of Synthetic Cannabinoids and Metabolites: Adding New Compounds to an Existing LC-MS/MS Method By Sharon Lupo and Frances Carroll Abstract The analysis of synthetic cannabinoids and their metabolites
Determination of ormones in Drinking Water by LC/MS/MS Using an Agilent InfinityLab Poroshell P Column (EPA 539) Application Note Environmental Authors Rong-jie Fu and Chen-ao (Andy) Zhai Agilent Technologies
Determination of pesticides in baby food by UHPLC/MS/MS using the Agilent 1290 Infinity LC system and the Agilent 6460 triple quadrupole LC/MS Application Note Food Authors Gerd Vanhoenacker, Frank David,
Determination of Carbonyl Compounds In Water by Dinitrophenylhydrazine Derivatization and HPLC/UV* EPA Method 8315A UCT Part Number: EUC1812M15 (Unendcapped C18-2000 mg/15 ml cartridge) March 2013 Method
Trace-Level Analysis of Melamine in Milk Products on Agilent 789A/5975C GC/MSD Using a ew Agilent J&W DB-5ms Ultra Inert Column and SampliQ SCX Cartridges Application ote Gas Chromatography/Mass Spectrometry/Food
APPLICATION NOTE Speciation of Bromine Compounds in Ozonated Drinking Water using Ion Chromatography and Inductively Coupled Plasma Mass Spectrometry AN43227 Antonella Guzzonato 1, Shona McSheehy Ducos
application Note Gas Chromatography/ Mass Spectrometry Author Timothy D. Ruppel PerkinElmer, Inc. Shelton, CT 06484 USA Benzoylecgonine in Urine by SAMHSA GC/MS Introduction The United States Department
LC/MS/MS qua ntitation of β-estradiol 17-acetate using an Agilent 6460 Triple Quadrupole LC/MS working in ESI negative ion mode Application Note Authors Siji Joseph Agilent Technologies India Pvt. Ltd.
16 17 Leucomalachite green hydrochloride Leucomalachite green N N N + Cl - N N-[4-[[4-(dimethylamino)-phenyl] phenylmethylene]-2,5-cyclohexadien-1-ylidene ]-N-methylmethanaminium chloride C 23 H 25 ClN
APPLICATION NOTE 72482 Determination of urea in ultrapure water by IC-MS/MS Authors Soon Fatt Lee, 1 Fiona Teh Hui Boon, 1 Chris Cheah Hun Teong, 1 and Jeff Rohrer 2 ¹Thermo Fisher Scientific, Singapore
Benzene Contamination in BabyFood and Beverages by New Generation of Static Headspace Autosampler coupled to Fast GC-TOFMS Authors: Daniela Cavagnino DANI Instruments SpA viale Brianza, 87 Cologno Monzese
Technical Note 101 Method Development in Solid Phase Extraction using Non-Polar ISOLUTE SPE Columns for the Extraction of Aqueous Samples This technical note includes by specific information on the extraction
Determination of Pharmaceuticals in Environmental Samples BACKGROUND The full effects of pharmaceutical substances in the environment are largely unknown however the risk is significant enough that many
Evaluation of Several Columns and Solvents for Post-Extraction Gel Permeation Chromatography (GPC) Clean-up of Fish Tissue Prior to PCB Analysis Keywords: Automated GX-271 GPC Clean-up System, Contaminants,
DETERMINATION OF PESTICIDES IN FOOD USING UPLC WITH POLARITY SWITCHING TANDEM QUADRUPOLE LC/MS/MS Peter Hancock 1, Cristiana C. Leandro 2,3, Brendan J. Keely 2 and Richard J. Fussell 3 1 Waters Corporation,
25-OH VITAMIN D 3 IN PLASMA BY UV FAST CODE Z19610 BIOCHEMISTRY Quantification of 25-hydroxyvitamin D 3 (25-hydroxycholecalciferol) in serum is the best established indicator of vitamin D status. Vitamin
- A7/1 - Appendix 7 Format for the Compilation of Tier II Summaries - Active substance PART Section Analytical methods The example of a summary and assessment of data which follows is intended to illustrate
Detection of Trace Level Pharmaceuticals in Drinking Water by nline SPE Enrichment with the Agilent Infinity Series nline-spe Solution Application Note Environmental Author Edgar Naegele Agilent Technologies,
Use of An Improved Version of C8+SCX Mixed-Mode Solid Phase Extraction Material for Clean Extraction and Recovery of Basic, Zwitterionic, Neutral and Acidic Compounds from Biological Fluids Yuhui Yang,
The Effect of Contact Angle and Wetting on Performance in Solid Phase Extraction Edouard S. P. Bouvier, Randy E. Meirowitz and Uwe D. Neue Waters Corporation, 34 Maple Street, Milford, MA 01757 USA Presented
A Comprehensive Multi-Class Veterinary Medicines Method Using A New Best-In-Class Triple Quadrupole Mass Spectrometer Ed George - Senior Application Scientist - Thermo Fisher Scientific Danny Chan, George
25-OH VITAMIN D 3 AND 25-OH VITAMIN D 2 IN PLASMA BY UV / VIS FAST CODE Z19710 BIOCHEMISTRY Quantification of 25-hydroxyvitamin D 3 (25-hydroxycholecalciferol) in serum is the best established indicator
Development of analytical methods for Catecholamines, Metanephrines and related metabolites in urine and plasma matrices From sample preparation to LC/MS/MS Linda Côté Christophe Deckers Agilent Technologies
Application Note AN89.V.1 Extraction of Methylmalonic Acid from Serum Using ISOLUTE SAX Page 1 Extraction of Methylmalonic Acid from Serum Using ISOLUTE SAX Prior to LC-MS/MS Analysis Sample Preparation
application Note Gas Chromatography/ Mass Spectrometry Author Timothy D. Ruppel PerkinElmer, Inc. Shelton, CT 06484 USA Opiates in Urine by SAMHSA GC/MS Introduction The United States Department of Health
Quantitative analysis of small molecules in biological samples 100 Jeevan Prasain, Ph.D. Department of Pharmacology & Toxicology, UAB % 0 300 500 700 900 1100 1300 1500 1700 m/z Class Overview Introduction
Definitive / LC- Analysis: A Rugged 4-Min Method for High-Throughput Labs By Justin Steimling and Frances Carroll Abstract Methods for monitoring alcohol consumption biomarkers and are generally limited
Rapid, Reliable Metabolite Ratio Evaluation for MIST Assessments in Drug Discovery and Preclinical Studies Jonathan Danaceau, Erin Chambers, and Kenneth J. Fountain Waters Corporation, Milford, MA, USA
TN143 EVOLUTE WCX Columns for Solid Phase Extraction of Quaternary Amines and Strongly Basic Compounds from Aqueous Samples This Chemistry Data Sheet provides guidelines for the extraction of quaternary
A2. Bioanalytical method development 1. Optimization of chromatographic conditions Method development and optimization of chromatographic parameters is of utmost important for validating a method in biological
Plasma Metanephrines and 3-Methoxytyramine by LC/MS/MS Using Agilent SimpliQ WCX SPE, 129 Infi nity LC, and 646 Triple Quadrupole LC/MS Application Note Clinical Research Authors Linda Côté and Christophe
Application Note AutoMate-Q40 Automated QuEChERS Extraction for Pesticide Residues in Botanicals Tyler Trent, Applications Sales Specialist, Teledyne Tekmar P a g e 1 Abstract QuEChERS is a Quick-Easy-Cheap-
High Sensitivity HPLC Analysis of Perchlorate in Tap Water Using an Agilent 66 Triple Quadrupole LC/MS System Application Note Environmental Authors Don Noot, Matthew Noestheden, and Ralph Hindle Vogon
HPLC Columns Shodex TM ODP2 HP series columns Better retention of highly polar substances Technical notebook No. 6 Contents 1. Introduction 1-1. Specifications 1-2. Eluent Compatibility of ODP2 HP Series
38 Current Trends in Mass Spectrometry November 6 Assay Robustness Improvement for Drug Urinalysis Using FAIMS and H-SRM on a Triple- Quadrupole Mass Spectrometer This article demonstrates the improved
Pesticides Analysis Using the Agilent 5977A Series GC/MSD Application Note Food Testing and Agriculture Authors Jia-jia Wu and Wen-wen Wang Agilent Technologies Co. Ltd (China) Beijing, People s Republic
225-37872 Sep. 2018 Small Capacity Pretreatment Kit Micro Volume QuEChERS kit Instruction Manual Read this manual thoroughly before you use the product. Keep this manual for future reference. This page
Determination of Beta-Blockers in Urine Using Supercritical Fluid Chromatography and Mass Spectrometry Application Note Doping Control Authors Prof. Maria Kristina Parr Freie Universität Berlin Institute
Page 1 of 16 Method DAS-AM-01-058 Sulfotepp impurities in Chlorpyrifos EC formulations A. ABSTRACT Method DAS-AM-01-058 has been validated for the analysis of the relevant impurity sulfotepp in chlorpyrifos