# Lecture 20. Measuring Pressure and Temperature (Chapter 9) Measuring Pressure Measuring Temperature MECH 373. Instrumentation and Measurements

Save this PDF as:

Size: px
Start display at page:

Download "Lecture 20. Measuring Pressure and Temperature (Chapter 9) Measuring Pressure Measuring Temperature MECH 373. Instrumentation and Measurements"

## Transcription

1 MECH 373 Instrumentation and Measurements Lecture 20 Measuring Pressure and Temperature (Chapter 9) Measuring Pressure Measuring Temperature 1

2 Measuring Acceleration and Vibration Accelerometers using Piezoelectric Sensing Elements An accelerometer using a piezoelectric material as the sensing element is shown below: It consists of a housing, a mass called the seismic mass, and a piezoelectric sensing element, which typically uses the longitudinal piezoelectric effect. An initial force between the mass and sensor is obtained with a preloading spring sleeve. 2

3 Measuring Acceleration and Vibration As the housing for the accelerometer is subject to an acceleration, the force exerted by the mass on the quartz crystal is altered. This generates a charge on the crystal, which can be sensed with a charge amplifier. Piezoelectric accelerometers are available in many ranges up to ±1000g, where g is the acceleration due to gravity. Quartz crystal accelerometers can have very high values of natural frequency up to 125 khz. This allows them to measure frequencies as high as 25 khz. 3

4 Measuring Acceleration and Vibration Vibrometer An instrument that is used to measure the ground motion in earthquakes and sometimes to measure vibration in machines is called the vibrometer. 4

5 Measuring Acceleration and Vibration Although the basic components are the same as the piezoelectric or strain-gage accelerometers, the mode of operation is different. In the vibrometer, the spring is quite soft and as the housing moves, the mass remains approximately stationary. The relative motion, y, is large and sensed with a potentiometer. These devices are used to measure vibrations with frequencies that are high relative to the natural frequency of the spring-mass system, which is often less than 1 Hz. The vibrometer effectively measures the displacement of the base rather than the acceleration. Thus, these devices are most sensitive to vibrations with moderate frequencies and fairly large displacement amplitudes. High frequency vibrations usually have small values of displacement amplitude and are better measured with accelerometers. 5

6 Load Cell Measuring Force Load cell is a term used to describe a transducer that generates a voltage signal as a result of an applied force, usually along a particular direction. Virtually any simple metal structure deforms when subjected to a force, and as long as the resulting stresses are below the material yield stress, the deflection (δ) and resulting strain (ε) are linear functions of the applied force: F = C δ 1 F = C2ε where C s are constants determined from analysis or calibration. The most common force-measuring devices are strain-gage load cells. They are often constructed of a metal and have a shape such that the range of forces to be measured results in a measurable output voltage over the desired operating range. Figure 8.34(a) shows a cantilever beam instrumented with four strain gages, two on the top and two on the bottom, to measure normal or bending stresses. These four gages form the Wheatstone bridge and offer effective temperature compensation. The output of the bridge is four times the output of an individual gage. 6

7 An unbalanced Wheatstone Bridge R x is the STRAIN GAUGE, generally V AB 0 V AB Unknown R (STRAIN GAUGE connected here) R x changes due to strain V AB changes 7

8 Measuring Force Figure 8.34(b) shows a hollow-cylinder load cell. It also uses four strain gages and is also temperature compensated. As the cylinder is compressed, it becomes slightly shorter, while the diameter becomes slightly larger. As a result, two gages measure the axial compression. The other two, mounted transversely, measure the tensile diametral strain. Since the transverse strain is only Poisson's ratio times the axial strain, the output is less than four times the output of a single axial strain gage. For a Poisson's ratio of 0.3, the output will be about 2.6 times the output of a single axial strain gage. 8

9 Measuring Force Due to their simple design, any range can be readily manufactured. Commercial load cells are available with ranges from ounces up to several hundred thousand pounds. Unlike accelerometers, it is not useful to specify the frequency response of commercial load cells because the mass and flexibility of the instrumented system control the dynamic response. Furthermore, an installed load cell will add flexibility to the system and also affect the dynamic response. If the flexibility of strain-gage load cells is too high, load cells using piezoelectric sensors, which are much stiffer, are commercially available. 9

10 Measurement Systems 10

11 Measuring Pressure Pressure is measured in three different forms: absolute pressure, gage pressure, and differential pressure. Absolute pressure is that used in thermodynamics to determine the state of a substance. Gage pressure is the pressure relative to the local ambient air pressure. Differential pressure is simply the difference in pressure at two points in a system. The relationship p between = p different + p pressure forms is abs gage ambient Pressure Transducers A very common and inexpensive device used to measure fluid pressure is the diaphragm strain-gage pressure transducer, shown in next slide. 11

12 Measuring Pressure The test pressure is applied to one side of the diaphragm, a reference to the other side. A deflection of the diaphragm is sensed with strain gages. The reference pressure can be atmospheric, so the transducer measures gage pressure. The reference side could be sealed and evacuated so that the transducer measures absolute pressure. Both sides could be connected to different test pressures so that the measurement is of differential pressure. 12

13 Measuring Pressure In the past, the diaphragm was made of metal and foil strain gages were used. Recently, it has become common to make the diaphragm of a semiconductor material (usually silicon) with semiconductor strain gages formed into the diaphragm. This is a less expensive construction technique, and since semiconductor gages have high gage factors, the sensitivity is improved. Normally, the Wheatstone-bridge signal conditioner is built into the transducer (all branches of the bridge are active gages), and the strain gages are connected to give temperature compensation. Most strain-gage pressure transducers produce a DC output in the millivolt range, but some include internal amplifiers and have outputs in the range 0 to 5 or 0 to 10 V. The higher voltage output units are less susceptible to environmental electrical noise. Pressures can also be sensed with LVDT (linear variable differential transformer) devices like the one shown below: 13

14 Measuring Pressure This diagram shows an arrangement with a flexible chamber called a capsule and an LVDT to sense the displacement. This design is more expensive than those using strain-gage sensors but may be more durable in an application requiring a long lifetime. The output is a DC voltage with a range on the order of 0-5 or 0-10 V. In the process industries, the voltage output usually be converted to a 4 to 20 ma current for signal transmission. 14

15 Measuring Pressure Capacitive sensors are sometimes used in pressure transducers. When one or more fixed metal plates are placed directly above or below a metallic diaphragm, a capacitor is created that forms an effective secondary element. Displacement of the diagram changes the average gap separation, which varies the capacitance developed between the two plates. Capacitive pressure transducers, like those shown below, are particularly useful for very low pressures (as low as 0.1 Pa) since captive sensors can detect extremely small deflections. 15

16 Measuring Pressure Transducers used for high-frequency pressure measurements usually use a piezoelectric sensing element. A piezoelectric transducer is shown below: These transducers generally use transverse-effect piezoelectric sensing elements. The piezoelectric material is very stiff, and the transducers have a high natural frequency in many applications. If the diaphragm (or other displacing element) is very flexible, the natural frequency will be low and the transducer output will be misleading for high-frequency pressure measurements. 16

17 Measuring Pressure The diaphragm is of the flush-mounted type. When the transducer is installed, it comes into direct contact with the fluid in the pipe or chamber. There are two reasons for this: - If a cavity were included as in the other transducers, it might significantly alter the measurand due to loading. - The natural frequency would be reduced and the ability to respond to transients would be impaired. Piezoelectric pressure transducers can have natural frequencies up to 150 khz and are usable up to about 30 khz. 17

18 Measuring Pressure (Summary) Strain gage types Capacitive very low pressures Piezoelectric high-frequency pressure LVDT 18

### 10 Measurement of Acceleration, Vibration and Shock Transducers

Chapter 10: Acceleration, Vibration and Shock Measurement Dr. Lufti Al-Sharif (Revision 1.0, 25/5/2008) 1. Introduction This chapter examines the measurement of acceleration, vibration and shock. It starts

### APPLICATIONS OF VIBRATION TRANSDUCERS

APPLICATIONS OF VIBRATION TRANSDUCERS 1) Measurements on Structures or Machinery Casings: Accelerometers and Velocity Sensors Used in gas turbines, axial compressors, small and mid-size pumps. These sensors

### Force and Displacement Measurement

Force and Displacement Measurement Prof. R.G. Longoria Updated Fall 20 Simple ways to measure a force http://scienceblogs.com/dotphysics/200/02/diy_force_probe.php Example: Key Force/Deflection measure

### Sensors and Transducers. mywbut.com

Sensors and Transducers 1 Objectives At the end of this chapter, the students should be able to: describe the principle of operation of various sensors and transducers; namely.. Resistive Position Transducers.

### Unit 3 Transducers. Lecture_3.1 Introduction to Transducers

Unit 3 Transducers Lecture_3.1 Introduction to Transducers Introduction to transducers A transducer is a device that converts one form of energy to other form. It converts the measurand to a usable electrical

### MAS.836 PROBLEM SET THREE

MAS.836 PROBLEM SET THREE FSR, Strain Gauge, and Piezo Circuits: The purpose of this problem set is to familiarize yourself with the most common forms of pressure and force measurement. The circuits you

### Midterm 2 PROBLEM POINTS MAX

Midterm 2 PROBLEM POINTS MAX 1 30 2 24 3 15 4 45 5 36 1 Personally, I liked the University; they gave us money and facilities, we didn't have to produce anything. You've never been out of college. You

### Overview. Sensors? Commonly Detectable Phenomenon Physical Principles How Sensors Work? Need for Sensors Choosing a Sensor Examples

Intro to Sensors Overview Sensors? Commonly Detectable Phenomenon Physical Principles How Sensors Work? Need for Sensors Choosing a Sensor Examples Sensors? American National Standards Institute A device

### Slide 1. Temperatures Light (Optoelectronics) Magnetic Fields Strain Pressure Displacement and Rotation Acceleration Electronic Sensors

Slide 1 Electronic Sensors Electronic sensors can be designed to detect a variety of quantitative aspects of a given physical system. Such quantities include: Temperatures Light (Optoelectronics) Magnetic

### Force Sensors. What is a force sensor?

orce Sensors What is a force sensor? In physics, the definition of force is any agent that causes a mass to move. When you push an object, say a toy wagon, you re applying a force to make the wagon roll.

### INTRODUCTION TO PIEZO TRANSDUCERS

PIEZO SYSTEMS, INC. 65 Tower Office Park Woburn, MA 01801 USA Tel: 781 933 4850 Fax: 781 933 4743 email: sales@piezo.com Find Search for a product or category HOME PRODUCTS CUSTOM OEM CATALOG TECHNICAL

### Sensors and transducers

Sensors and transducers Measurement is an important subsystem of a mechatronics system. Its main function is to collect the information on system status and to feed it to the micro-processor(s) for controlling

### Finite Element Analysis of Piezoelectric Cantilever

Finite Element Analysis of Piezoelectric Cantilever Nitin N More Department of Mechanical Engineering K.L.E S College of Engineering and Technology, Belgaum, Karnataka, India. Abstract- Energy (or power)

### Prof. S.K. Saha. Sensors 1. Lecture 5 June 11, Prof. S.K. Saha. Purpose Classification Internal Sensors. External Sensors.

Lecture 5 June 11, 2009 Sensors Prof. S.K. Saha Dept. of Mech. Eng. IIT Delhi Announcement Outlines of slides in Lectures 1-4 on May 15, 18, 21, June 01, 2009, respectively, are available from: http://web.iitd.ac.in/~saha/

### DEVELOPMENT OF DROP WEIGHT IMPACT TEST MACHINE

CHAPTER-8 DEVELOPMENT OF DROP WEIGHT IMPACT TEST MACHINE 8.1 Introduction The behavior of materials is different when they are subjected to dynamic loading [9]. The testing of materials under dynamic conditions

### MET 301 EXPERIMENT # 2 APPLICATION OF BONDED STRAIN GAGES

MET 301 EPERIMENT # 2 APPLICATION OF BONDED STRAIN GAGES 1. Objective To understand the working principle of bonded strain gauge and to study the stress and strain in a hollow cylindrical shaft under bending,

### 1.103 CIVIL ENGINEERING MATERIALS LABORATORY (1-2-3) Dr. J.T. Germaine Spring 2004 LABORATORY ASSIGNMENT NUMBER 6

1.103 CIVIL ENGINEERING MATERIALS LABORATORY (1-2-3) Dr. J.T. Germaine MIT Spring 2004 LABORATORY ASSIGNMENT NUMBER 6 COMPRESSION TESTING AND ANISOTROPY OF WOOD Purpose: Reading: During this laboratory

### Robotics Intelligent sensors (part 2)

Robotics Intelligent sensors (part ) Tullio Facchinetti Tuesday 6 th December, 06 http://robot.unipv.it/toolleeo Pressure measurement static pressure is a force applied to

### EE C245 ME C218 Introduction to MEMS Design

EE C245 ME C218 Introduction to MEMS Design Fall 2007 Prof. Clark T.-C. Nguyen Dept. of Electrical Engineering & Computer Sciences University of California at Berkeley Berkeley, CA 94720 Lecture 16: Energy

### Because the third wire carries practically no current (due to the voltmeter's extremely high internal resistance), its resistance will not drop any

Strain gauges If a strip of conductive metal is stretched, it will become skinnier and longer, both changes resulting in an increase of electrical resistance end-to-end. Conversely, if a strip of conductive

### SCB10H Series Pressure Elements PRODUCT FAMILY SPEFICIFATION. Doc. No B

PRODUCT FAMILY SPEFICIFATION SCB10H Series Pressure Elements SCB10H Series Pressure Elements Doc. No. 82 1250 00 B Table of Contents 1 General Description... 3 1.1 Introduction... 3 1.2 General Description...

### AOE 3054 Measuring static displacement and deformation in structures

AOE 3054 Measuring static displacement and deformation in structures Dr. Eric Johnson Preparation for experiments 2 and 5. elevant reading: Lab manual, experiment 2. Notes on Maxwell s eciprocal Theorem

### Modeling and Experimentation: Mass-Spring-Damper System Dynamics

Modeling and Experimentation: Mass-Spring-Damper System Dynamics Prof. R.G. Longoria Department of Mechanical Engineering The University of Texas at Austin July 20, 2014 Overview 1 This lab is meant to

### Model tests and FE-modelling of dynamic soil-structure interaction

Shock and Vibration 19 (2012) 1061 1069 1061 DOI 10.3233/SAV-2012-0712 IOS Press Model tests and FE-modelling of dynamic soil-structure interaction N. Kodama a, * and K. Komiya b a Waseda Institute for

### CHAPTER 9: TEMPERATURE, PRESSURE, STRAIN AND MOTION MEASUREMENTS

CHAPTER 9: TEMPERATURE, PRESSURE, STRAIN AND MOTION MEASUREMENTS I. MEASUREMENT OF TEMPERATURE 1. Introduction In industrial-process control, temperature is the most frequently controlled and measured

### Modelling of Different MEMS Pressure Sensors using COMSOL Multiphysics

International Journal of Current Engineering and Technology E-ISSN 2277 4106, P-ISSN 2347 5161 2017 INPRESSCO, All Rights Reserved Available at http://inpressco.com/category/ijcet Research Article Modelling

### Meteorology 432. Barometry Spring 2013

Meteorology 432 Barometry Spring 2013 Basics Revisited Objective: Measure the static pressure exerted by the atmosphere. Static Pressure: Force per unit area in the absence of air motion. In this case,

### Strain and Force San José State University A. Mysore Spring 2009

Strain and Force Strain Gage Measures strain as a change in length L, observed by change in resistance R, for a given resistivity ρ and cross-sectional area A. For elastic materials that follow Hooke s

### Resonant Column and Torsional Cyclic Shear System

Resonant Column and Torsional Cyclic Shear System Combined Resonant Column (RC) & Torsional Cyclic Shear (TCS) Test apparatus to determinate with saturated soil : Shear Modulus Damping Modulus versus Shear

### TEMPERATURE AND HUMIDITY DEPENDENCE ON STABILITY OF TORQUE MEASURING DEVICES Tassanai Sanponpute and Nittaya Arksonnarong

IMEKO 22 nd TC3, 15 th TC5 and 3 rd TC22 International Conferences 3 to 5 February, 2014, Cape Town, Republic of South Africa TEMPERATURE AND HUMIDITY DEPENDENCE ON STABILITY OF TORQUE MEASURING DEVICES

### Nanomotion Precision Piezo Ceramic Motors

Nanomotion Precision Piezo Ceramic Motors The Company Nanomotion was founded in 1992 Developed enabling technology for ultrasonic piezo-ceramic motors Entered the market in 1996, selling products to leading

### Piezoelectric Actuation in a High Bandwidth Valve

Ferroelectrics, 408:32 40, 2010 Copyright Taylor & Francis Group, LLC ISSN: 0015-0193 print / 1563-5112 online DOI: 10.1080/00150193.2010.484994 Piezoelectric Actuation in a High Bandwidth Valve D. T.

### Load Cell Design Using COMSOL Multiphysics

Load Cell Design Using COMSOL Multiphysics Andrei Marchidan, Tarah N. Sullivan and Joseph L. Palladino Department of Engineering, Trinity College, Hartford, CT 06106, USA joseph.palladino@trincoll.edu

### Introduction. Energy Generation with the Piezo Effect

Introduction The term Energy Harvesting is popularly used when electricity is generated from sources such as ambient temperature, vibrations or air flows. Since there are now electronic circuits whose

### CEE575 - Homework 1. Resistive Sensing: Due Monday, January 29

CEE575 - Homework 1 Resistive Sensing: Due Monday, January 29 Problem 1: Planes A metallic wire embedded in a strain gage is 4 cm long with a diameter of 0.1 mm. The gage is mounted on the upper surface

### Characteristics of instruments

Characteristics of instruments Accuracy Accuracy is a property of a complete measurement rather than a single element. Accuracy is quantified using measurement error: E = measured value true value = system

### Physical Properties Testing Technical Bulletin

Technical Bulletin MANUFACTURER Raven Lining Systems 13105 E. 61 st Street, Suite A Broken Arrow, OK 74012 (918) 615-0020 TENSILE TESTING OF PLASTICS ASTM D638, ISO 527 Tensile tests measure the force

### Design of a MEMS Capacitive Comb-drive Accelerometer

Design of a MEMS Capacitive Comb-drive Accelerometer Tolga Kaya* 1, Behrouz Shiari 2, Kevin Petsch 1 and David Yates 2 1 Central Michigan University, 2 University of Michigan * kaya2t@cmich.edu Abstract:

### Experimental Investigation of Transient Hydrodynamic Forces of a Single-Blade Centrifugal Pump

Proceedings of the 4th WSEAS International Conference on luid Mechanics and Aerodynamics, Elounda, Greece, August 1-, 006 (pp1-6) Experimental Investigation of Transient Hydrodynamic orces of a Single-Blade

### Use of reciprocity calibrated accelerometer standards for performing routine laboratory comparison calibrations. Technical paper 226

Use of reciprocity calibrated accelerometer standards for performing routine laboratory comparison calibrations Technical paper 6 Use of reciprocity calibrated accelerometer standards for performing routine

### THICK-FILM THERMALLY EXCITED RESONATOR FOR MASS FLOW MEASUREMENT. D. Crescini, D. Marioli and A. Taroni

THICK-FILM THERMALLY EXCITED RESONATOR FOR MASS FLOW MEASUREMENT D. Crescini, D. Marioli and A. Taroni Department of Electronics for Automation, Faculty of Engineering, University of Brescia, Via Branze

### Development of Laser Thickness Gauge in Steel Plate Shearing Line

JFE TECHNICAL REPORT No. 21 (Mar. 2016) Development of Laser Thickness Gauge in Steel Plate Shearing Line TEZUKA Koichi *1 Abstract: JFE Steel has developed a laser thickness gauge for the purpose of guaranteeing

### Piezo Theory: Chapter 1 - Physics & Design

Piezoelectric effect inverse piezoelectric effect The result of external forces to a piezoelectric material is positive and negative electrical charges at the surface of the material. If electrodes are

### Biomedical Instrumentation

Biomedical Instrumentation Winter 393 Bonab University Transducer,, and Actuator Transducer: a device that converts energy from one form to another : converts a physical parameter to an electrical output

### Francisco Paulo Lépore Neto. Marcelo Braga dos Santos. Introduction 1. Nomenclature. Experimental Apparatus and Formulation

Francisco Paulo Lépore Neto and Marcelo Braga dos Santos Francisco Paulo Lépore Neto fplepore@mecanica.ufu.br Federal University of Uberlandia School of Mechanical Engineering 38408-902 Uberlandia, MG,

### Module 4 : Deflection of Structures Lecture 4 : Strain Energy Method

Module 4 : Deflection of Structures Lecture 4 : Strain Energy Method Objectives In this course you will learn the following Deflection by strain energy method. Evaluation of strain energy in member under

### Influence of residual stresses in the structural behavior of. tubular columns and arches. Nuno Rocha Cima Gomes

October 2014 Influence of residual stresses in the structural behavior of Abstract tubular columns and arches Nuno Rocha Cima Gomes Instituto Superior Técnico, Universidade de Lisboa, Portugal Contact:

### The 5rd International Conference on. COMEC OCTOBER 2013, Brasov, Romania WORK SAFETY

The 5rd International Conference on Computational Mechanics and Virtual Engineering COMEC 2013 24 25 OCTOBER 2013, Brasov, Romania THEORETICAL STUDIES AND EXPERIMENTAL DETERMINATIONS FOCUSED ON THE INCREASE

### Electromagnetics in COMSOL Multiphysics is extended by add-on Modules

AC/DC Module Electromagnetics in COMSOL Multiphysics is extended by add-on Modules 1) Start Here 2) Add Modules based upon your needs 3) Additional Modules extend the physics you can address 4) Interface

### MECHANICS OF STRUCTURES SCI 1105 COURSE MATERIAL UNIT - I

MECHANICS OF STRUCTURES SCI 1105 COURSE MATERIAL UNIT - I Engineering Mechanics Branch of science which deals with the behavior of a body with the state of rest or motion, subjected to the action of forces.

### 2.3 Concept of Active Acoustic Metamaterials

Active Acoustic Metamaterials 23 2.3 Concept of Active Acoustic Metamaterials In order to understand the concept and the structure of an active acoustic metamaterial, consider the structure of the external

### COEFFICIENT OF DYNAMIC HORIZONTAL SUBGRADE REACTION OF PILE FOUNDATIONS ON PROBLEMATIC GROUND IN HOKKAIDO Hirofumi Fukushima 1

COEFFICIENT OF DYNAMIC HORIZONTAL SUBGRADE REACTION OF PILE FOUNDATIONS ON PROBLEMATIC GROUND IN HOKKAIDO Hirofumi Fukushima 1 Abstract In this study, static loading tests and dynamic shaking tests of

### MEMS Mechanical Fundamentals

ROCHESTER INSTITUTE OF TECHNOLOGY MICROELECTRONIC ENGINEERING MEMS Mechanical Fundamentals Dr. Lynn Fuller webpage: http://people.rit.edu/lffeee Electrical and Microelectronic Engineering Rochester Institute

### Borehole Geophysics. Acoustic logging measurements

Acoustic logging measurements - Review of basic physics background - Concept of P- and S-wave measurements and logging tools - Tube waves - Seismic imaging - Synthetic seismograms - Field application examples

### DESIGN AND CONSTRUCTION OF A WATER TUNNEL. Stephen C. Ko

i DESGN AND CONSTRUCTON OF A WATER TUNNEL By Stephen C. Ko This work has been carried out as a part of a grant from the National Science Foundation for the development of fluid mechanics laboratory equipments

### SCHOOL OF COMPUTING, ENGINEERING AND MATHEMATICS SEMESTER 1 EXAMINATIONS 2012/2013 XE121. ENGINEERING CONCEPTS (Test)

s SCHOOL OF COMPUTING, ENGINEERING AND MATHEMATICS SEMESTER EXAMINATIONS 202/203 XE2 ENGINEERING CONCEPTS (Test) Time allowed: TWO hours Answer: Attempt FOUR questions only, a maximum of TWO questions

### Elasticity: Term Paper. Danielle Harper. University of Central Florida

Elasticity: Term Paper Danielle Harper University of Central Florida I. Abstract This research was conducted in order to experimentally test certain components of the theory of elasticity. The theory was

### Lecture 4 Pressure Sensing. ECE 5900/6900 Fundamentals of Sensor Design

EE 4900: Fundamentals of Sensor Design 1 Lecture 4 Pressure Sensing Pressure Sensing Q: What are we measuring? A: elative Pressure or Gauge Pressure. Pressure is Force (F) per Unit Area (A); P=F/A 2 SI

### Dynamic Tests on Ring Shear Apparatus

, July 1-3, 2015, London, U.K. Dynamic Tests on Ring Shear Apparatus G. Di Massa Member IAENG, S. Pagano, M. Ramondini Abstract Ring shear apparatus are used to determine the ultimate shear strength of

### Piezoelectric Composites as Bender Actuators

Integrated Ferroelectrics, 71: 221 232, 2005 Copyright Taylor & Francis Inc. ISSN 1058-4587 print / 1607-8489 online DOI: 10.1080/10584580590964673 Piezoelectric Composites as Bender Actuators Karla Mossi,

### Finite Element Analysis of the Local Effect of a Piezoelectric Patch on an Aluminum Plate

Mechanics and Mechanical Engineering Vol. 21, No. 2 (2017) 233 242 c Lodz University of Technology Finite Element Analysis of the Local Effect of a Piezoelectric Patch on an Aluminum Plate Aziz Lebied

### Instrumentation Techniques for Studying the Horizontal * Behavior of High-Speed Railways

Procedia Engineering Volume 143, 16, Pages 87 879 Advances in Transportation Geotechnics 3. The 3rd International Conference on Transportation Geotechnics (ICTG 16) Instrumentation Techniques for Studying

### 2d-Laser Cantilever Anemometer

2d-Laser Cantilever Anemometer Introduction Measuring principle Calibration Design Comparative measurement Contact: Jaroslaw Puczylowski University of Oldenburg jaroslaw.puczylowski@forwind.de Introduction

### MECHANICS OF MATERIALS. Prepared by Engr. John Paul Timola

MECHANICS OF MATERIALS Prepared by Engr. John Paul Timola Mechanics of materials branch of mechanics that studies the internal effects of stress and strain in a solid body. stress is associated with the

### Introduction to Mechanical Vibration

2103433 Introduction to Mechanical Vibration Nopdanai Ajavakom (NAV) 1 Course Topics Introduction to Vibration What is vibration? Basic concepts of vibration Modeling Linearization Single-Degree-of-Freedom

### HEALTH MONITORING OF PLATE STRUCTURE USING PIEZO ELECTRIC PATCHES AND CURVATURE MODE SHAPE

ISSN (Online) : 2319-8753 ISSN (Print) : 2347-6710 International Journal of Innovative Research in Science, Engineering and Technology An ISO 3297: 2007 Certified Organization, Volume 2, Special Issue

### Mechanics of Solids. Mechanics Of Solids. Suraj kr. Ray Department of Civil Engineering

Mechanics Of Solids Suraj kr. Ray (surajjj2445@gmail.com) Department of Civil Engineering 1 Mechanics of Solids is a branch of applied mechanics that deals with the behaviour of solid bodies subjected

### Accelerometer Theory & Design

11 Chapter 2 Accelerometer Theory & Design 2.1 Introduction An accelerometer is a sensor that measures the physical acceleration experienced by an object due to inertial forces or due to mechanical excitation.

### Unconventional Double-Current Circuit for Deflection and Temperature Simultaneous Measurement

http://dx.doi.org/10.5755/j01.eee.21.1.6834 ELEKTRONIKA IR ELEKTROTECHNIKA ISSN 1392-1215 VOL. 21 NO. 1 2015 Unconventional Double-Current Circuit for Deflection and Temperature Simultaneous Measurement

### TESTING AND ANALYSIS OF COMPOSITE SANDWICH BEAMS

TESTING AND ANALYSIS OF COMPOSITE SANDWICH BEAMS I. M. Daniel, J. L. Abot, and K. A. Wang Walter P. Murphy Professor, Departments of Civil and Mechanical Engineering, Robert R. McCormick School of Engineering

### Lecture 4 Scanning Probe Microscopy (SPM)

Lecture 4 Scanning Probe Microscopy (SPM) General components of SPM; Tip --- the probe; Cantilever --- the indicator of the tip; Tip-sample interaction --- the feedback system; Scanner --- piezoelectric

### UNCERTAINTY IN TORQUE CALIBRATION USING VERTICAL TORQUE AXIS ARRANGEMENT AND SYMMETRICAL TWO FORCE MEASUREMENT

UNCERTAINTY IN TORQUE CALIBRATION USING VERTICAL TORQUE AXIS ARRANGEMENT AND SYMMETRICAL TWO FORCE MEASUREMENT Dieter Kenzler 1, Jörg Oblasser 2, Andreas Subaric-Leitis 2 and Christian Ullner 2 1 ) SCHATZ

### RESILIENT INFRASTRUCTURE June 1 4, 2016

RESILIENT INFRASTRUCTURE June 1 4, 2016 DAMAGE DETECTION OF UHP-FRC PLATES USING RANDOM DECREMENT TECHNIQUE Azita Pourrastegar MASc Student, Ryerson University, azita2.pourrastegar@ryerson.ca, Canada Hesham

### Characterization of Elastomeric Isolators for Shock

ME 4054 Design Projects Characterization of Elastomeric Isolators for Shock - Vol. 1 - May 1, 2009 Team Nick Haupt Matt Hildebrand Jim Holmberg Brian Kornis Sam Newbauer Advisors Jim Wieczorek Ed Alexander

### Piezo materials. Actuators Sensors Generators Transducers. Piezoelectric materials may be used to produce e.g.: Piezo materials Ver1404

Noliac Group develops and manufactures piezoelectric materials based on modified lead zirconate titanate (PZT) of high quality and tailored for custom specifications. Piezoelectric materials may be used

### TRANSDUCERS. An Introduction

TRANSDUCERS An Introduction TRANSDUCERS A transducer is a device that converts energy from one form to another Energy forms can be mechanical, visual, aural, electrical, thermal, chemical, etc. (examples

### two structural analysis (statics & mechanics) APPLIED ACHITECTURAL STRUCTURES: DR. ANNE NICHOLS SPRING 2017 lecture STRUCTURAL ANALYSIS AND SYSTEMS

APPLIED ACHITECTURAL STRUCTURES: STRUCTURAL ANALYSIS AND SYSTEMS DR. ANNE NICHOLS SPRING 2017 lecture two structural analysis (statics & mechanics) Analysis 1 Structural Requirements strength serviceability

### Innovative Systems Design and Engineering ISSN (Paper) ISSN (Online) Vol.4, No.15, 2013

A Study on Piezoelectric Elements and Its Utility in Designing of Electronic Scale Nilotpal Manna Department of Electronics and Instrumentation Engineering JIS College of Engineering, Kalyani, Nadia 741235,

### Recommended Format for Mechanics of Solids Laboratory Reports

Recommended Format for Mechanics of Solids Laboratory Reports A. Title page (a) Lab title (b) Names of participant(s) (c) Date of experiment (d) Date submitted B. Introduction In one short paragraph, summarize

### Chapter 5. Vibration Analysis. Workbench - Mechanical Introduction ANSYS, Inc. Proprietary 2009 ANSYS, Inc. All rights reserved.

Workbench - Mechanical Introduction 12.0 Chapter 5 Vibration Analysis 5-1 Chapter Overview In this chapter, performing free vibration analyses in Simulation will be covered. In Simulation, performing a

### Optimization of MEMS Piezo-Resonators

Copyright 2013 Tech Science Press SL, vol.7, no.2, pp.129-134, 2013 Optimization of MEMS Piezo-Resonators A. Frangi 1, M. Cremonesi 1, A. Jaakkola 2 and K. Bathe 2 Abstract: Single crystal silicon MEMS

### = A x (t) + B utt), by d{ _-=-=-

M.Tech. [ 24 103 ] Degree Examination ndustrial Process nstrumentation First Semester COMPUTER CONTROL OF PROCESSES (Effective from the Admitted Batch of2003-2004) Time: 3 Hours Maximum marks: 100 Answer

### Degradation of Piezoelectric Materials for Energy Harvesting Applications

Degradation of Piezoelectric Materials for Energy Harvesting Applications P Pillatsch 1, N Shashoua 1,ASHolmes 2, E M Yeatman 2,PKWright 1 1 University of California Berkeley, Advanced Manufacturing for

### Sensors & Transducers 2016 by IFSA Publishing, S. L.

Sensors & Transducers, Vol. 96, Issue, January 206, pp. 52-56 Sensors & Transducers 206 by IFSA Publishing, S. L. http://www.sensorsportal.com Collapse Mode Characteristics of Parallel Plate Ultrasonic

### University of California

University of California Peer Reviewed Title: Damage Identification Study of a Seven-story Full-scale Building Slice Tested on the UCSD-NEES Shake Table Author: Moaveni, Babak, Tufts University He, Xianfei,

### Transduction Based on Changes in the Energy Stored in an Electrical Field

Lecture 7-1 Transduction Based on Changes in the Energy Stored in an Electrical Field - Electrostriction The electrostrictive effect is a quadratic dependence of strain or stress on the polarization P

### Deflections and Strains in Cracked Shafts due to Rotating Loads: A Numerical and Experimental Analysis

Rotating Machinery, 10(4): 283 291, 2004 Copyright c Taylor & Francis Inc. ISSN: 1023-621X print / 1542-3034 online DOI: 10.1080/10236210490447728 Deflections and Strains in Cracked Shafts due to Rotating

### Numerical modelling of induced tensile stresses in rock in response to impact loading

Numerical modelling of induced tensile stresses in rock in response to impact loading M.T. Mnisi, D.P. Roberts and J.S. Kuijpers Council for Scientific and Industrial Research (CSIR): Natural Resources

### Design and Simulation of Various Shapes of Cantilever for Piezoelectric Power Generator by Using Comsol

Design and Simulation of Various Shapes of Cantilever for Piezoelectric Power Generator by Using Comsol P. Graak 1, A. Gupta 1, S. Kaur 1, P. Chhabra *1, D. Kumar **1, A. Shetty 2 1 Electronic Science

### Pressure Sensor Evaluation Dr. Lynn Fuller Lianna Dicke Webpage:

ROCHESTER INSTITUTE OF TECHNOLOGY MICROELECTRONIC ENGINEERING Pressure Sensor Evaluation Lianna Dicke Webpage: http://people.rit.edu/lffeee 82 Lomb Memorial Drive Rochester, NY 14623-5604 Email: Lynn.Fuller@rit.edu

### User s Manual. STUDY OF DIELECTRIC CONSTANT Model: DEC-01

User s Manual STUDY OF DIELECTRIC CONSTANT Model: DEC-01 Manufactured by.. Scientific Equipment & Services 358/1, New Adarsh Nagar, Roorkee - 247 667, UA, INDIA Ph.: +91-1332-272852, 277118 Fax: +91-1332-274831

### International Snow Science Workshop, Davos 2009, Proceedings

COMPARISON AND COMPLEMENTARITIES OF AVALANCHE PRESSURE MEASUREMENTS: PIEZO-ELECTRIC LOAD CELLS AND DEFORMATION BASED PRESSURE DECONVOLUTION Djebar BAROUDI 1, Betty SOVILLA 2, Emmanuel THIBERT 1 1 Cemagref,

### Piezo Engineering Primer

. The Direct and Inverse Piezoelectric Effect In 88, while performing experiments with tourmaline, quartz, topaz, cane sugar and Rochelle salt crystals, Pierre and Jacques Curie discovered that when mechanical

### Machine Vibration Measurement

CSU, Chico Spring 2009 Machine Vibration Measurement Hussam AliKhan Mechatronic Engineering Student California State University Chico, CA 95929-0789 April 8, 2009 Machine Vibration Measurement Abstract

### Dynamic Mechanical Analysis of Solid Polymers and Polymer Melts

Polymer Physics 2015 Matilda Larsson Dynamic Mechanical Analysis of Solid Polymers and Polymer Melts Polymer & Materials Chemistry Introduction Two common instruments for dynamic mechanical thermal analysis

### Laser Interferometric Displacement Measurements of Multi-Layer Actuators and PZT Ceramics

Ferroelectrics, 320:161 169, 2005 Copyright Taylor & Francis Inc. ISSN: 0015-0193 print / 1563-5112 online DOI: 10.1080/00150190590967026 Laser Interferometric Displacement Measurements of Multi-Layer

### Piezoline. Introduction Instruction for Using 62 PIEZOLINE. Contents

62 PIEZOLINE Piezoline Contents Introduction Instruction for using 62 Sealings 62 Piezoelectric actuators 63 1 Piezoelectric Effect inverse piezoelectric effect 64 2 Design of piezoactuators 64 2.1 Piezo

### Strain sensors based on bragg gratings

IMEKO 20 th TC3, 3 rd TC16 and 1 st TC22 International Conference Cultivating metrological knowledge 27 th to 30 th November, 2007. Merida, Mexico. Strain sensors based on bragg gratings K. Haase Hottinger